Aerosol Vacuum-Assisted Plasma Ionization (Aero-VaPI) Coupled to Ion Mobility-Mass Spectrometry
In this communication, we report on the real-time analysis of organic aerosol particles by Vacuum-assisted Plasma Ionization-Mass Spectrometry (Aero-VaPI-MS) using a home-built VaPI ion source coupled to a Synapt G2-S HDMS ion mobility-mass spectrometry (IM-MS) system. Standards of organic molecules of interest in prebiotic chemistry were used to generate aerosols. Monocaprin and decanoic acid aerosol particles were successfully detected in both the positive and negative ion modes, respectively. A complex aerosol mixture of different sizes of polymers of L-malic acid was also examined through ion mobility (IM) separations, resulting in the detection of polymers of up to eight monomeric units. This noncommercial plasma ion source is proposed as a low cost alternative to other plasma ionization platforms used for aerosol analysis, and a higher-performance alternative to more traditional aerosol mass spectrometers. VaPI provides robust online ionization of organics in aerosols without extensive ion activation, with the coupling to IM-MS providing higher peak capacity and excellent mass accuracy.
KeywordsPlasma ionization Aerosol analysis Prebiotic chemistry
Few mass spectrometry (MS) platforms are available for aerosol particle analysis that have both high temporal and high mass resolution . Current online analysis methods suffer from limitations such as insufficient peak capacity, limited mass accuracy, unpredictable adduct formation, extensive ion activation, and slow response [2, 3, 4, 5, 6, 7, 8]. Developing soft aerosol ion sources that can be coupled to most high resolution mass spectrometers would thus enable more robust molecular identification of organics from aerosol particles, yielding more easily interpretable insights into aerosol chemistry mechanisms and composition.
One of the most commonly used mass spectrometers in the aerosol field is the high-resolution time-of-flight aerosol mass spectrometer (aerosol HR-ToF-AMS) . It utilizes electron ionization, yielding complex fragment mass spectra absent of molecular ions, hindering the ability to easily track intact species . Later, a Filter Inlet for Gases and AEROsols (FIGAERO) was developed and coupled to a high-resolution time-of-flight chemical ionization-mass spectrometer (aerosol HRToF-CIMS) , which relies on filter collection of particles followed by thermodesorption prior to ionization. Because of the finite filter collection time, aerosol particle measurements with the FIGAERO-HRToF-CIMS lack high temporal resolution, and are susceptible to on-filter reactions causing artifacts during sampling and measurement . Ambient ionization techniques such as extractive electrospray ionization , direct analysis in real time (DART) , flowing atmospheric-pressure afterglow (FAPA) , and droplet assited inlet ionization  have also been used for improving aerosol analysis. However, these MS techniques can still suffer from extraction efficiency limitations, ion suppression, incomplete aerosol sampling, high background, or oxidation reactions. More recently, an aerosol flowing atmospheric-pressure afterglow (AeroFAPA) ion source was developed to detect organic compounds in aerosols . Also, Zenobi et al. developed a dielectric barrier discharge plasma technique  used for studying offline-vaporized liquid or solid samples , and for real-time breath analysis . Further enhancements to aerosol MS analysis have been achieved through the coupling of MS to ion mobiltity (IM), enabling the separation of isobaric species in aerosol mixtures .
In this work, we present a home-built Vacuum-assisted Plasma Ionization (VaPI) ion source for aerosols (Aero-VaPI) coupled to a Waters Synapt G2-S HDMS Ion Mobility-Mass Spectrometry (IM-MS) system. The Aero-VaPI ion source is an atmospheric pressure glow discharge plasma that is interconnected with the inlet of an IM-mass spectrometer. The vacuum pull from the mass spectrometer induces a Venturi effect at a sampling tee, enabling the capture and entrainment of the sample aerosol into the metastable atom population of the plasma before going into the mass spectrometer. This arrangement offers advantages such as a cost effective and simple assembly, efficient ionization, aerodynamic Venturi sample entrainment, and more straightforward molecular identification [15, 16]. The applicability of the Aero-VaPI-MS system to particle analysis in a prebiotic chemistry context is presented.
Chemical standards of monocaprin (1-monodecanoylglycerol) and decanoic acid (each ≥99%) were purchased from Sigma Aldrich, and L-malic acid (99%) from Chem-Impex International. Aerosol samples of monocaprin and decanoic acid were separately generated from aqueous and methanolic ~2 mM solutions with a TSI 3076 Constant Output Atomizer at 2 standard liters per minute (SLM) and measured with Aero-VaPI-MS in the positive and negative ion modes, respectively. Poly-L-malic acid was synthesized via heated dry-down reactions previously described by Forsythe et al. (2015) , without any purification. An L-malic acid-equivalent 0.50 mM solution of poly-L-malic acid was analyzed with ESI-IM-MS in the negative mode. A nebulizer (SLT8900H, Salter Labs® 8900 Series Small Volume Jet Nebulizer from Medline Inc.) was used to generate aerosols at 6 SLM of the same poly-L-malic acid solution and was measured with Aero-VaPI-IM-MS in the negative ion mode. The nebulizer offerred an air jet of lower pressure (10 psi versus 34 psi of N2), producing a more stable signal for L-malic acid polymers during particle generation. Aerosols were not size-selected prior to mass spectrometry analysis.
Results and Discussion
Our interest in aerosols stems from our research in the prebiotic de novo synthesis of proto-biopolymers in an origins of life scenario. It has been shown that aerosols can act as chemical reactors in the prebiotic formation of such polymeric species [18, 19, 20, 21]. Dobson et al., for example, have proposed that by their ability to concentrate molecules, aerosols could have served as efficient prebiotic chemical reactors for the first protocelullar structures . Condensation reactions are important in prebiotic chemistry, and include formation of lipids  and far-from-equilibrium polymers . These reactions can be favored in aerosols, where water loss is driven by evaporation. Trainer et al. have hypothesized that organic species that have functionalities of interest to prebiotic chemistry are more likely to exist in the aerosol phase . Previous studies have focused on amino acid or tholin formation [24, 25], but only few have investigated the molecular composition of products generated via aerosols .
Aero-VaPI-IM-MS spectral data for poly-L-malic acid aerosols (Figure 3c,d, and Figure S4.) were very similar to that obtained by ESI IM-MS, with the interesting difference that the water loss signals were much less abundant than with ESI. It was necessary to multiply the [Mˊ – H]- ion signals by a factor of 10 in order to better observe their IM distribution (Figure 3d). As expected, IM enabled the separation of some isobaric species, as discussed in the Supporting Information, Figure S5.
Oligomers did not represent the majority of the total ion signal in some drift time regions, such as the region between 1–2 ms and the shoulder at 5.5 ms. The first arrival time distribution region (1–2 ms) was broad and populated mostly by small ions such as water loss ions C4H1O3- and C4H3O4- at m/z 97 and 115, which may be dehydrated forms of malic acid such as maleic anhydride and maleic acid, respectively (Figure S6). Oxidized species possibly arising from these ions, such as C4H5O6-, C4H6O6-, C4H5O7-, and C4H7O8-, were also observed with corresponding m/z values of 149, 150, 165, and 183. The presence of these smaller ionic species suggests that oligomer ion-molecule reactions do occur within the Aero-VaPI source due to the higher reactivity of plasma sources relative to ESI . VaPI plasma ionization is expected to have an abundance of high energy charge transfer species, as there is no grid electrode to filter them out from the plasma, as in DART . Further experiments are required to investigate these and other phenomena related to ion generation mechanisms in VaPI.
The Aero-VaPI ion source is cost effective, has minimal assembly, provides softer ionization than with standard aerosol mass spectrometers, and, when coupled to IM-MS, can rapidly separate species in aerosol mixtures for the molecular detection of organic analytes in particles. Although organics may undergo a small amount of activation in the VaPI source, no sample handling is required with the Aero-VaPI-MS system, avoiding solvent-dependent extraction efficiency limitations. Initial results suggest that this technique can be applied to monitor aerosol chemistry in real-time, including prebiotic condensation reactions. Future studies of new prebioic aerosol chemistry with Aero-VaPI-MS are expected to provide insight into the mechanisms of de novo proto-biolopymer formation.
This work was jointly supported by NSF and the NASA Astrobiology Program, under the NSF Center for Chemical Evolution, CHE-1004570.
- 6.Lopez-Hilfiker, F.D., Mohr, C., Ehn, M., Rubach, F., Kleist, E., Wildt, J., Mentel, T.F., Lutz, A., Hallquist, M., Worsnop, D., Thornton, J.A.: A novel method for online analysis of gas and particle composition: Description and evaluation of a filter inlet for gases and aerosols (FIGAERO). Atmos. Meas. Tech. 7, 983–1001 (2014)CrossRefGoogle Scholar
- 14.Krechmer, J.E., Groessl, M., Zhang, X., Junninen, H., Massoli, P., Lambe, A.T., Kimmel, J.R., Cubison, M.J., Graf, S., Lin, Y.H., Budisulistiorini, S.H., Zhang, H., Surratt, J.D., Knochenmuss, R., Jayne, J.T., Worsnop, D.R., Jimenez, J.L., Canagaratna, M.R.: Ion mobility spectrometry-mass spectrometry (IMS-MS) for on- and offline analysis of atmospheric gas and aerosol species. Atmos. Meas. Tech. 9, 3245–3262 (2016)CrossRefGoogle Scholar
- 16.Keelor J.D., Farnsworth P.B., L. Weber A., Abbott-Lyon H., Fernández F.M.: Multimodal vacuum-assisted plasma ion (VAPI) source with transmission mode and laser ablation sampling capabilities. J. Am. Soc. Mass Spectrom. 27, 897–907 (2016)Google Scholar
- 22.Mamajanov, I., MacDonald, P.J., Ying, J., Duncanson, D.M., Dowdy, G.R., Walker, C.A., Engelhart, A.E., Fernandez, F.M., Grover, M.A., Hud, N.V., Schork, F.J.: Ester formation and hydrolysis during wet-dry cycles: Generation of far-from-equilibrium polymers in a model prebiotic reaction. Macromolecules. 47, 1334–1343 (2014)CrossRefGoogle Scholar
- 27.Forsythe J.G., Yu S.-S., Mamajanov I., Grover M.A., Krishnamurthy R., Fernández F.M., Hud N.V.: Ester-mediated amide bond formation driven by wet–dry cycles: A possible path to polypeptides on the prebiotic earth. Angew. Chem. Int. Edit. 54, 9871–9875 (2015)Google Scholar