Comprehensive Lipidome Analysis by Shotgun Lipidomics on a Hybrid Quadrupole-Orbitrap-Linear Ion Trap Mass Spectrometer
- 3.7k Downloads
Here we report on the application of a novel shotgun lipidomics platform featuring an Orbitrap Fusion mass spectrometer equipped with an automated nanoelectrospray ion source. To assess the performance of the platform for in-depth lipidome analysis, we evaluated various instrument parameters, including its high resolution power unsurpassed by any other contemporary Orbitrap instrumentation, its dynamic quantification range and its efficacy for in-depth structural characterization of molecular lipid species by quadrupole-based higher-energy collisional dissociation (HCD), and ion trap-based resonant-excitation collision-induced dissociation (CID). This evaluation demonstrated that FTMS analysis with a resolution setting of 450,000 allows distinguishing isotopes from different lipid species and features a linear dynamic quantification range of at least four orders of magnitude. Evaluation of fragmentation analysis demonstrated that combined use of HCD and CID yields complementary fragment ions of molecular lipid species. To support global lipidome analysis, we designed a method, termed MSALL, featuring high resolution FTMS analysis for lipid quantification, and FTMS2 analysis using both HCD and CID and ITMS3 analysis utilizing dual CID for in-depth structural characterization of molecular glycerophospholipid species. The performance of the MSALL method was benchmarked in a comparative analysis of mouse cerebellum and hippocampus. This analysis demonstrated extensive lipidome quantification covering 311 lipid species encompassing 20 lipid classes, and identification of 202 distinct molecular glycerophospholipid species when applying a novel high confidence filtering strategy. The work presented here validates the performance of the Orbitrap Fusion mass spectrometer for in-depth lipidome analysis.
Key wordsShotgun lipidomics Hybrid quadrupole-Orbitrap-linear ion trap mass spectrometry High mass resolution Nanoelectrospray ionization Collision induced dissociation
Lipids comprise a diverse group of molecules that play important roles in membrane dynamics, storage of metabolic energy, and signaling [1, 2]. The complete assortment of molecular lipid species in a biological system is termed a lipidome and is governed by a network of metabolic pathways that synthesizes distinct lipid species by assembling or disassembling a multitude of hydrocarbon residues (e.g., fatty acids) and polar head groups (e.g., choline) . Lipid species can be divided into several lipid categories and lipid classes based on their chemical structures . The most abundant lipid categories in eukaryotic cells include glycerophospholipids, sphingolipids, glycerolipids, and sterol lipids. Dysfunctional regulation of lipid metabolism causes cellular lipotoxicity, impairs cellular processes, and contributes to the pathogenesis of disorders such as obesity, atherosclerosis, and neurodegeneration . To improve the understanding of the regulation of lipidome composition during physiological and pathophysiological processes warrants lipidomics routines supporting quantitative monitoring of molecular lipid species on a global scale.
Lipidomic techniques cover a broad range of mass spectrometry-based workflows affording extensive lipidome coverage with several hundreds of lipid species [3, 6, 7, 8, 9]. These techniques are based on direct infusion MS (i.e., shotgun lipidomics) or liquid chromatography (LC)-MS [10, 11, 12, 13]. A core component of all lipidomic platforms is the type of mass spectrometer employed, which spans sensitive low resolution triple quadrupole and ion trap instruments to high resolution hybrid quadrupole time-of-flight, ion trap-Orbitrap, and ion trap-Fourier transform ion cyclotron resonance (FT-ICR) machines offering high resolving power and high mass accuracy [14, 15, 16, 17]. By shotgun lipidomics, lipid extracts are directly infused into the mass spectrometer, followed by identification and quantification of lipid species based on accurately determined masses and/or specific tandem mass analysis [10, 16]. Identified lipid species should be annotated by a shorthand nomenclature corresponding to the level of detail attainable by the mass analysis [18, 19, 20]. The detection of lipid species by high resolution FTMS analysis or by tandem mass analysis and detection of lipid class-specific fragment ions [e.g., m/z 184.0733 for phosphatidylcholine (PC) species] supports only annotation by “sum composition” (e.g., PC 34:1). In comparison, annotation by more detailed “molecular species composition” (e.g., PC 16:0-18:1) requires tandem mass analysis and detection of molecular structure-specific fragment ions . Further in-depth structural analysis by ozone derivatization and structure-specific fragmentation analysis affords annotation by “defined molecular species composition,” which can denote the double bond position within the fatty acid moieties of lipid species [e.g., PC 16:0-18:1(n-9)]  and the abundance of positional isomers [21, 23, 24].
The Orbitrap Fusion is a new generation of high resolution instrumentation made commercially available in 2013. This hybrid quadrupole-Orbitrap-ion trap mass spectrometer is equipped with three mass analyzers, two detectors, and two collision cells enabling complementary multi-stage fragmentation analysis using both quadrupole-based higher-energy collisional dissociation (HCD) and ion trap-based resonant-excitation collision-induced dissociation (CID). The ultra-high-field mass analyzer of the Orbitrap Fusion affords acquisition of FTMS spectra with resolving power up to 450,000 (full width at half maximum (FWHM) at m/z 200) at a transient duration of 1 s. The high resolving power at this scan speed is thus far unsurpassed by any other commercially available Orbitrap instrument, and at m/z 400 comparable to that of commonly used FT-ICR mass spectrometers operated with a transient duration of 1 s, as typically used for routine analysis [17, 25, 26, 27]. A key benefit of high resolving power for lipidomic analysis is the ability to accurately discriminate lipid species with isobaric nominal masses as well as their isotopes and, hence, support accurate lipid identification [17, 28]. Moreover, the capability of multi-stage fragmentation provides a new avenue for in-depth structural lipid analysis capable of exploiting differential fragmentation patterns obtained by CID and HCD ion activation and, as such, resolve isomeric lipid species . Importantly, the Orbitrap Fusion offers a wide range of operational modes supporting in-depth lipidome analysis, but requires systematic evaluation and optimization of instrument parameters in order to benchmark its performance and identify potential pitfalls.
Here, we report on the development and application of a novel shotgun lipidomics platform using the Orbitrap Fusion mass spectrometer equipped with an automated direct infusion nanoelectrospray ion source. To benchmark the platform for lipidomic analysis, we evaluated the ability to accurately distinguish isobaric lipid species and their isotopes, the isotopic fidelity (i.e., accuracy of relative isotopic abundances), the dynamic quantification range, and parameters for in-depth fragmentation analysis by HCD and CID. This systematic evaluation served as a guideline for developing a comprehensive acquisition method, termed MSALL, featuring high resolution FTMS analysis with a resolution setting of 450,000 for accurate lipid identification and quantification, and high resolution FTMS2 analysis by sequential precursor ion fragmentation with combined HCD and CID and ITMS3 analysis using dual CID for in-depth structural characterization of molecular glycerophospholipid species. The performance of the MSALL method for global lipidome analysis was evaluated by a comparative analysis of mouse cerebellum and hippocampus. This analysis demonstrated reproducible lipidome quantification of 311 lipid species encompassing 20 lipid classes, and the identification of 202 distinct molecular glycerophospholipid species by applying a novel high confidence filtering strategy requiring the detection of multiple molecular structure-specific fragment ions. The systematic evaluation of the performance characteristics of the Orbitrap Fusion-based platform presented herein validates its use for in-depth lipidome analysis.
2 Materials and Methods
2.1 Chemicals and Lipid Standards
Chemicals, solvents, and synthetic lipid standards were purchased from Sigma-Aldrich (St. Louis, MO, USA), Rathburn Chemicals (Walkerburn, Scotland), Avanti Polar Lipids (Alabaster, AL, USA), and Larodan Fine Chemicals (Malmö, Sweden).
2.2 Mouse Brain Tissue Sampling
Animal experiments were conducted in strict accordance with German law (in congruence with 86/609/EEC) for the use of laboratory animals and approved by the local animal welfare committee at the Johannes Gutenberg University Mainz. Male C57Bl/6 wild-type mice were euthanized by an overdose of ketamine via intraperitoneal injection. Subsequently, the mice were perfused intracardially with 4 °C 155 mM ammonium acetate, and the cerebellum and hippocampus were dissected. The tissues were frozen on dry ice and stored at –80 °C until further processing.
2.3 Lipid Extraction
Brain tissues were homogenized in 155 mM ammonium acetate and analyzed for total protein concentration using BCA Protein Assay Kit (Thermo Scientific (Rockford, IL, USA)). Aliquots of tissue homogenates corresponding to 10 μg of total protein were subjected to lipid extraction by a modified Bligh and Dyer protocol executed at 4 °C . Briefly, the tissue homogenates were spiked with internal lipid standard mix containing cholesteryl ester (CE) 19:0, triacylglycerol (TAG) 17:1/17:1/17:1, diacylglycerol (DAG) 19:0/19:0, lysophosphatidic acid (LPA) O-16:0, phosphatidic acid (PA) 17:0/14:1, lysophosphatidylserine (LPS) 17:1, phosphatidylserine (PS) 17:0/20:4, phosphatidylethanolamine (PE) O-20:0/O-20:0, lysophosphatidylcholine (LPC) O-17:0, PC O-18:1/O-18:1, phosphatidylinositol (PI) 17:0/20:4, phosphatidylglycerol (PG) 17:0/14:1, ceramide (Cer) 18:1;2/17:0;0, sphingomyelin (SM) 18:1;2/17:0;0, glucosylceramide (HexCer) 18:1;2/12:0;0, and sulfatide (SHexCer) 18:1;2/12:0;0. Next, the samples were added chloroform/methanol (2:1, v/v) and vigorously mixed for 1 h. Samples were subsequently centrifuged for 1 min at 500 g to facilitate phase separation. The lower organic phase was subsequently transferred to a new tube and vacuum evaporated.
2.4 Mass Spectrometric Lipid Analysis
Lipid extracts and synthetic lipid standards were dissolved in chloroform/methanol (1:2, v/v) and subjected to mass spectrometric analysis using an Orbitrap Fusion Tribrid (Thermo Fisher Scientific, San Jose, CA, USA) equipped with a TriVersa NanoMate (Advion Biosciences, Ithaca, NY, USA). Aliquots of lipid extracts or synthetic lipid standards were loaded in 96-well plates, mixed with 13.3 mM and 1.3 mM ammonium acetate in 2-propanol for positive and negative ion mode analysis, respectively. Samples were infused using a back pressure of 1.25 psi and ionization voltage of ±0.95 kV . All FTMS data were recorded using a max injection time of 100 ms, automated gain control at 2 · 105, two microscans, and a target resolution of 450,000 (FWHM at m/z 200), unless noted otherwise. All FTMS2 data were acquired using max injection time of 100 ms, automated gain control at 5 · 104, one microscan, and a target resolution of 30,000, unless noted otherwise. All ITMS3 data were acquired using max injection time of 200 ms, automated gain control at 1 · 104, and one microscan. All FTMS and ITMS data were acquired in profile mode, and using an ion transfer tube temperature of 80 °C and 125 °C for positive and negative ion mode, respectively. Positive ion mode MSALL analysis (see overview in Figure 4a) was performed using (1) high resolution FTMS analysis of the low m/z range 350–600 [monitoring LPC and lysophosphatidylethanolamine (LPE) species], (2) high resolution FTMS analysis of the high m/z range 500–1200 [monitoring SM, Cer, DAG, PC, ether-linked phosphatidylcholine (PC O-), PE, ether-linked phosphatidylethanolamine (PE O-) and TAG species], (3) sequential FTMS2 analysis in 1.0008 u steps across the m/z range 400.3–1000.8 using a quadrupole ion isolation width of 1.0 u and HCD with normalized collision energy at 25%, (4) sequential FTMS2 analysis in 1.0008 u steps across the m/z range 400.3–1000.8 using a quadrupole ion isolation width of 1.0 u and CID with collision energy at 33%, and (5) targeted ITMS3 analysis for PE, monomethyl phosphatidylethanolamine (MMPE) and dimethyl phosphatidylethanolamine (DMPE) species using a quadrupole ion isolation width of 1.0 u and CID with normalized collision energy at 38% followed by ion trap ion isolation at width of 2.5 and CID with collision energy at 30%. Negative ion mode MSALL analysis was performed using (1) high resolution FTMS analysis of the low m/z range 370–660 [monitoring LPA, LPS, and lysophosphatidylinositol (LPI) species], (2) high resolution FTMS analysis of the high m/z range 550–1200 (monitoring PA, PS, PI, PG, and SHexCer species), (3) sequential FTMS2 analysis in 1.0008 u steps across the m/z range 400.2–1000.7 using a quadrupole ion isolation width of 1.0 u and HCD with collision energy at 30%, (4) sequential FTMS2 analysis in 1.0008 u steps across the m/z range 400.2–1000.7 using a quadrupole ion isolation width of 1.0 u and CID with collision energy at 33%, and (5) targeted ITMS3 analysis for PC, PC, O- and PS species using a quadrupole ion isolation width of 1.0 u and CID with normalized collision energy at 38% followed by using ion trap ion isolation width of 2.5 and CID with collision energy at 30%. All FTMS2 and ITMS3 analyses were performed using data-independent acquisition (DIA) using pre-compiled target lists with precursor m/z values.
2.5 Annotation of Lipid Species
Lipid species were annotated as previously described [3, 6, 30]. For example, glycerophospholipid and glycerolipid species annotated by “sum composition” were denoted as: <lipid class > <total number of C in fatty acid moieties>:<total number of double bonds in fatty acid moieties > (e.g., PI 34:1). Sphingolipid species annotated by “sum composition” were denoted as < lipid class > <total number of C in the long-chain base and fatty acid moiety>:<total number of double bonds in the long-chain base and fatty acid moiety>;<total number of OH groups in the long-chain base and fatty acid moiety > (e.g., SM 35:1;2) . Molecular glycerophospholipid species annotated by “molecular species composition” were denoted as: <lipid class > <number of C in the first fatty acid moiety>:<number of double bonds in the second fatty acid moiety > – < number of C in the second fatty acid moiety>:<number of double bonds in the second fatty acid moiety > (e.g., PS 16:0-22:6).
2.6 Data Processing and Lipid Identification
Lipid species detected by MSALL using high resolution FTMS analysis were identified and quantified using ALEX software as previously described . In short, lipid species detected by FTMS and annotated by sum composition were quantified by normalizing their intensity to the intensity of an internal lipid standard of identical lipid class and multiplying with the spike amount of the internal lipid standard and the type I isotope correction factor [32, 33, 34]. Molecular glycerophospholipid species detected by MSALL using FTMS2 and ITMS3 were identified using a prototype ALEX123 software combined with a prototype lipid fragmentation database. Fragment ion information was compiled by systematic fragmentation analysis of (only) synthetic glycerophospholipid standards in both positive and negative ion mode. This analysis demonstrated that glycerophospholipid species release multiple fragment ions corresponding to both direct release of fatty acid moieties (e.g., acyl anions in negative ion mode) and neutral loss of fatty acid moieties (e.g., as ketenes in both positive and negative ion mode). Using this information, we constructed a library of more than 4500 molecular glycerophospholipid species having fatty acid moieties with 10 to 26 carbon atoms and containing 0 to 6 double bonds. For all molecular glycerophospholipid species, we computed m/z values of expected fragment ions based on the fragmentation patterns observed for the synthetic glycerophospholipid standards. This fragmentation database was used by the prototype ALEX123 software for identifying fragment ions detected by FTMS2 and ITMS3 analysis using m/z tolerance settings of ±0.005 u and ±0.1 u, respectively. The ALEX123 output format was saved in relational database format  and further processed for data visualization, filtering, and molecular glycerophospholipid species identification using Tableau Desktop (Tableau Software). An intensity filter was applied for identification requiring that ions detected by FTMS, FTMS2 and ITMS3 should have intensity higher than 2000, 500 and 1 count, respectively (values determined by manual assessment of spectral quality), and be detected in at least half of all sample injections. The numbers of molecular glycerophospholipid species identified by both positive and negative FTMS2 analysis using HCD, FTMS2 analysis using CID and ITMS3 analysis were compared using Venn Diagram Online (www.bioinformatics.lu/venn.php).
3 Results and Discussion
3.1 A Shotgun Lipidomics Platform Featuring an Orbitrap Fusion Mass Spectrometer
Operational Modes and Data Acquisition Procedures Supported by the Orbitrap Fusion Mass Spectrometer
Data acquisition procedures
DDA of SIMa
based on MS data
DIA of SIM
based on target list
HCD in the quadrupole
CID in the ion trap
DDA of MS2 based on MS data
DIA of MS2 based on target list
HCD in the quadrupole
CID in the ion trap
DDA of MSn
based on MSn-1 data
DIA of MSn
based on target list
3.2 Evaluation of High Resolution FTMS Analysis and Isotope Fidelity
To assess the accuracy of the isotope pattern measured by FTMS analysis (i.e., isotopic fidelity), we next overlaid recorded spectra with simulated isotope profiles (Figure 1). This analysis showed that FTMS data recorded on both the Orbitrap Fusion and the LTQ Orbitrap XL closely matched the simulated isotopic profile of the mono-isotopic peak (M0) and first isotope peak (M1) of Cer 42:2;2, PC 34:2, and TAG 54:2. More pronounced deviations were observed between simulated and measured isotopic profiles for ion clusters containing the second isotope (M2) of Cer 42:2;2, PC 34:2, and TAG 54:2 and the corresponding mono-isotopic (M0) peak of Cer 42:1;2, PC 34:1, and TAG 54:1, respectively. In all instances, we observed that the measured FTMS profiles were underestimated compared with the simulated isotope profiles. Similarly, comparing the spectral profiles of isotope clusters containing the third isotope (M3) of Cer 42:2;2, PC 34:2, and TAG 54:2 and the corresponding first isotope (M1) of Cer 42:1;2, PC 34:1, and TAG 54:1, respectively, we again noticed a systematic underestimation of the measured profiles compared with simulated isotope profiles. Importantly, in all instances, the FTMS data recorded on the Orbitrap Fusion better matched the simulated isotope profiles compared with that of FTMS data obtained using the LTQ Orbitrap XL.
To more systematically evaluate the isotopic fidelity of FTMS analysis on the Orbitrap Fusion and the LTQ Orbitrap XL, we next analyzed a set of samples where the molar ratio between synthetic TAG 54:2 and TAG 54:1 was systematically titrated. For example, we analyzed samples containing only TAG 54:2 or TAG 54:1, or various mixtures including TAG 54:2/TAG 54:1 (5:1, mol/mol), TAG 54:2/TAG 54:1 (1:1, mol/mol), and TAG 54:2/TAG 54:1 (1:5, mol/mol). As a proxy for isotopic fidelity, we determined the isotopic defects for each TAG isotopologue by computing the difference between the measured relative intensity and the simulated relative intensity of a given TAG isotopologue. To assess isotopic fidelity, we plotted the isotope defect as a function of the molar fraction of TAG 54:1 (Figure S1 in Supporting Information). This analysis showed that the isotopic defects by FTMS analysis on the Orbitrap Fusion were within an interval of –9% [for TAG 54:2(M2)] to 4% [for TAG 54:2(M0)] (Figure S1a). Notably, the isotopic defects for mono-isotopic TAG 54:2 and TAG 54:1, which are the ions used for identification and quantification, were minor and ranged between 0% and 4% and –5% and 0%, respectively. In comparison, the isotopic defects observed by FTMS analysis on the LTQ Orbitrap XL were within the interval of –18% to 3% [for TAG 54:2(M2)] (Figure S1b). Based on these results, we conclude that the isotopic fidelity of FTMS analysis using a resolution setting of 450,000 on the Orbitrap Fusion is acceptable and adequate for supporting accurate lipid identification and quantification.
3.3 Evaluation of Dynamic Quantification Range
A similar evaluation was performed using the neutral lipid standards TAG 51:3 (molecular species TAG 17:1-17:1-17:1) and deuterated TAG 51:1 (molecular species TAG 17:0-17:1-17:0 + D5) (Figure 2b). This experiment showed that also the TAG quantification range was linear with a slope value of approximately one across four orders of magnitude and featured a limit of quantification of approximately 1 nM TAG 51:3 (corresponding to molar ratio value of 0.005 in Figure 2b). Notably, the dynamic quantification range and limit of quantification of FTMS analysis on the Orbitrap Fusion are similar to shotgun lipidomic analysis on other FTMS instruments , hybrid quadrupole time-of-flight and triple quadrupole instruments . Based on these results, we conclude that FTMS analysis on the Orbitrap Fusion is applicable for quantitative lipidome analysis.
3.4 Evaluation of HCD and CID for Structural Lipid Characterization
3.5 Evaluation of Ion Isolation Efficiency for Fragmentation Analysis
Ion isolation is a critical determinant of the sensitivity and specificity of fragmentation analysis. Especially the ion isolation width is of importance since it is key for limiting false-positive identification of lipid species caused by potential interference from co-isolation and fragmentation. Reducing the ion isolation width to approximately 1 u, as required for the specificity of shotgun lipidomic analysis, causes a loss of precursor ion intensity and consequently affects the sensitivity of fragmentation analysis. The Orbitrap Fusion affords both quadrupole-based and ion trap-based ion isolation, each with a distinct set of analytical advantages. Quadrupole ion isolation is fast and occurs at the front-end of the instrument, whereas ion trap isolation occurs at the distal region of the instrument and offers, in principle, more efficient trapping of ions but is prone to promote fragmentation and loss of labile lipid ions such as Cer and PC species with formate adducts . To evaluate the performance of quadrupole-based and ion trap-based ion isolation, we performed two experiments. The first experiment was designed to evaluate the impact of quadrupole ion isolation width on the intensity of isolated precursor ion and co-isolation of neighboring ions. To this end, we performed FTMS2 analysis of m/z 778.5, corresponding to mono-isotopic precursor ion of synthetic PC 36:6 (molecular species PC 18:3-18:3), using different quadrupole ion isolation widths and monitored the intensities of the mono-isotopic precursor ion and its first and second isotope (Figure S2a in Supporting Information). This analysis demonstrated that using a quadrupole ion isolation width of 1.0 u yielded a 33% loss of mono-isotopic precursor ion intensity compared with using a quadrupole ion isolation width of 16 u, and a limited co-isolation of the first isotope (0.8% relative to the intensity of the mono-isotopic ion) and no co-isolation of the second isotope (Figure S2b). Notably, using an ion isolation width of 2.0 u yielded a 22% loss of the mono-isotopic precursor ion, but caused a pronounced co-isolation of the first isotope of PC 18:3-18:3 (M1, 24% relative to the intensity of the mono-isotopic ion). Based on this analysis, we conclude that MS2 analysis with high analytical specificity (i.e., limited co-isolation of precursor ions with a nominal mass difference of 1 u) should be performed using a quadrupole ion isolation width of 1.0 u or lower.
In the second experiment, we compared the ion isolation profile of quadrupole-based and ion trap-based ion isolation. To this end, we performed MS2 analyses of synthetic TAG 51:3 (molecular species TAG 17:1-17:1-17:1) using both quadrupole and ion trap ion isolation. We systematically ramped the precursor m/z value used for ion isolation and monitored the intensities of mono-isotopic TAG 51:3 and its first and second isotope (Figure S2c, d in Supporting Information). We observed that using an ion trap isolation width of 1.9 u yielded an effective ion isolation width of approximately 1 u (and not ~2 u as expected), corresponding to that observed when using a quadrupole ion isolation width of 1.0 u (Figure S2c, d). Moreover, we also observed that performing both quadrupole and ion trap ion isolation at m/z 860.8 yielded no significant co-isolation of the first isotope of TAG 51:3 (M1), thus allowing specific lipid precursor ion isolation for structural analysis. We note that similar results were obtained analyzing a synthetic PC standard (data not shown). Based on these results, we conclude that MS2 analysis using both quadrupole and ion trap ion isolation allows effective and specific isolation of lipid precursor ions. In addition, we recommend careful evaluation and optimization of especially ion trap ion isolation parameters when used for structural characterization of lipid species.
3.6 Shotgun Lipidomics by MSALL
3.7 Quantitative Lipidome Analysis
For quantitative lipidome analysis, we processed the high resolution FTMS data using ALEX software . ALEX software affords identification of lipid species detected by high resolution FTMS analysis and compiles output data in a database table format. This allows tracking all experimental data (e.g., originating .RAW file name, lipid species intensity, measured m/z), and implementing workflows for data normalization and visualization using database exploration tools such as open source Orange  and Tableau Desktop, respectively. Using ALEX software, we identified and calculated the molar abundances of 311 lipid species encompassing 20 lipid classes. These lipid species were annotated by sum composition and detected in at least half of all sample injections. Notably, 183 of the quantified lipids were glycerophospholipid species with two hydrocarbon moieties attached either via ester-linkages (diacyl species) or an ester- and an ether-linkage (plasmanyl or plasmenyl species). In comparison, a recent characterization of mouse forebrain by LC and triple quadrupole mass spectrometry quantified 325 lipid species annotated by sum composition, out of which 161 were glycerophospholipid species . This result demonstrates the efficacy of the Orbitrap Fusion-based platform for extensive identification of lipid species by high resolution FTMS analysis.
3.8 Identification of Molecular Glycerophospholipid Species by MS2 and MS3 Analysis
Having benchmarked the MSALL method for quantitative lipidomics, we next evaluated its efficacy for identification of molecular lipid species. While evaluating fragmentation analysis, we noted that FTMS2 afforded detection of several low abundant structure-specific fragment ions with high mass accuracy (Figure 3a, b and Figure 4c, d). For example, by negative ion mode FTMS2 analysis of synthetic PI 17:0-20:4, we detected 10 structure-specific fragment ions providing information about the fatty acid moiety composition and another 5 PI class-specific fragment ions (Figure 3a, b). FTMS2 analysis of the same PI species in positive ion mode allowed detection of four structure-specific fragment ions and one PI class-specific fragment ion (data not shown). Systematic positive and negative ion mode FTMS2 analysis of other (asymmetric) glycerophospholipid species, including PC, PC O-, PE, PE O-, PA, PS, PG, MMPE, and DMPE species, demonstrated detection of at least four structure-specific fragment ions in any ion mode. For example, FTMS2 fragmentation of PC 16:0-18:1 in positive ion mode allowed detection of four low abundant structure-specific fragment ions corresponding to the neutral loss of the fatty acid moieties (Figure 4c). Based on these observations, we rationalized that low abundant fragment ions detected with high mass resolution and mass accuracy can be utilized for higher confidence identification of molecular lipid species compared with only using a limited set of intense fragment ions such as acyl anion . Notably, low abundant fragment ions are typically not used for lipid identification in large-scale lipidomic analyses. Moreover, no software solutions and no lipid fragmentation databases are currently available for supporting identification of molecular lipid species using combinations of low abundant and intense structure-specific fragment ions detected using both HCD and CID. Hence, in order to enable identification of molecular lipid species detected by the MSALL approach, we compiled a prototype lipid fragmentation database and a prototype software tool for processing MSALL fragmentation data. The prototype lipid fragmentation database contains more than 4500 molecular glycerophospholipid species with m/z values for structure-specific fragment ions and lipid class-specific fragment ions predicted based on fragmentation analysis of representative standards. Presently, the prototype database only contains glycerophospholipids with two hydrocarbon moieties, either as two fatty acid moieties or as an ether moiety and a fatty acid moiety. For PE, MMPE, DMPE, PC, PC O-, and PS species, we also added MS3 fragmentation information on structure-specific fragments. The developed prototype software, termed ALEX123, queries MS2 and MS3 data using information in the lipid database for identification of molecular lipid species using structure-specific and lipid class-specific fragment ions. Notably, the ALEX123 software was designed to support independent processing of FTMS2, ITMS2, FTMS3, and ITMS3 data acquired using any combination of HCD and CID. The ALEX123 was designed to compile output data in database table format , which allows tracking all experimental data across large sample sets (e.g., polarity, ion activation, ion detection, measured m/z values of all fragment ion across all samples) and subsequent data filtering, quality control, and visualization using Tableau Desktop (Figure 5 and S3 in Supporting Information).
To further interrogate the efficacy of lipid identification, we applied the criterion that molecular glycerophospholipid species should be detected by at least three structure-specific fragment ions in at least half of all sample injections. We then compared the shortlists of molecular glycerophospholipid species identified by FTMS2 analysis using HCD, FTMS2 analysis using CID, and ITMS3 analysis. This evaluation demonstrated that all modes of negative ion fragmentation analysis collectively identified 194 molecular glycerophospholipid species (Table S1 in Supporting Information). Nineteen lipid species were identified by all modes of analysis. Moreover, FTMS2 with HCD afforded identification of all molecular lipid species identified by FTMS2 analysis using CID, expect for one lipid species. Comparison of negative ion mode FTMS2 with HCD and ITMS3 analysis showed an overlap of 46 lipid species and the unique detection of 30 molecular lipid species identified by ITMS3 (Figure 6c). Importantly, this result demonstrates the efficacy of implementing the complementary ITMS3 routine for in-depth structural analysis. Notably, the majority of unique lipid species identified using ITMS3 were PS and PC species. For example, the species PS 16:0-22:6 was, on average, detected by 5, 3, and 6 structure-specific fragment ions by negative ion mode FTMS2 with HCD, FTMS2 with CID, and ITMS3 analysis, respectively (Figure S3 in Supporting Information). Similar results were obtained by comparison of molecular lipid species identified by positive ion mode FTMS2 and ITMS3 analysis (Figure 6d). Finally, we compared the number of molecular glycerophospholipid species identified by all modes of fragmentation in negative and positive ion mode (Figure 6e). This analysis demonstrated the collective identification of 202 molecular glycerophospholipid species. Thirty-one lipid species were identified by both positive and negative ion mode analysis, whereas eight and 163 lipid species were identified by positive and negative ion mode fragmentation analysis, respectively. The higher number of identified molecular glycerophospholipid species in negative ion mode is explained by (1) the preferential ionization of majority of glycerophospholipids , and (2) the higher relative intensity of structure-specific fragment ions in this mode of analysis. Based on these results, we conclude that the MSALL method successfully affords comprehensive lipidome analysis. We note that the number of identified molecular species can in the future be increased by expanding the lipid fragmentation database with detailed fragmentation patterns of molecular sphingolipid, glycerolipid, lysoglycerophospholipid, and sterol lipid species, and repeating searches and analysis of the acquired data.
Here, we evaluated the performance of a novel lipidomics platform featuring an Orbitrap Fusion equipped with an automated Triversa Nanomate nanoelectrospray ion source. We demonstrate that the platform affords specific detection of isotopologues of distinct lipid species with high isotope fidelity and features a linear dynamic quantification range of at least four orders of magnitude. Furthermore, we show that combinatorial fragmentation analysis by multistage activation using both HCD and CID ion activation supports high confidence identification of molecular lipid species. To validate the performance of the platform for in-depth lipidome analysis, we developed an MSALL method and performed a comparative analysis of mouse cerebellum and hippocampus. This approach afforded absolute quantification of 311 lipid species belonging to 20 lipid classes by high resolution FTMS analysis, and in-depth molecular characterization of 202 distinct molecular glycerophospholipid species using FTMS2 and ITMS3 analysis combined with a new filtering strategy for high fidelity lipid identification. Based on our systematic evaluations, we conclude that the Orbitrap Fusion-based platform and the MSALL approach is a powerful new resource for in-depth lipidome analysis. We note, however, that future improvements are required for unleashing the full potential of the Orbitrap Fusion for comprehensive lipidome analysis. This includes improvements of the acquisition software for supporting DDA-based MS3 analysis triggered by DIA-based MS2 data, and providing control of ion trap activation Q parameter for supporting detection of low range m/z fragment ions by CID (e.g., the PC-specific fragment ion with m/z 184.0733). Furthermore, new and more extensive lipid fragmentation databases should be compiled with information on low abundant molecular structure-specific fragment ions and fragmentation patterns detectable by multidimensional MSn analysis , and as well implementation of new algorithms using combinations of high resolution and all MSALL data for supporting high confidence lipid identification. For example, in-depth structural analysis of lipid species with more than two fatty acid moieties, such as TAG species, requires detailed characterization of MSn fragmentation patterns using CID and HCD as well as compiling a comprehensive lipid database and designing an identification routine that supports high confidence identification. Importantly, our results demonstrate that high confidence identification of molecular lipid species detected by high resolution FTMS should be based on detection of at least three structure-specific fragment ions. Finally, we note that the Orbitrap Fusion also supports high scan speed FTMS acquisitions (up to 30 Hz), which provide an analytical avenue for the development of new high resolution LC-MS-MSn-based lipidomic workflows.
The authors thank Helen Podmore and Mark Baumert for expert advice on Orbitrap Fusion and Triversa NanoMate operation, respectively. The authors thank Nikolai Schmarowski, Johannes Vogt, Jan Baumgart, and Robert Nitsch for mouse brain samples. They thank Albert Casanovas and Martin Hermansson for critical reading of the manuscript and constructive comments. This work was supported by the VILLUM Center for Bioanalytical Sciences, VILLUM FONDEN (VKR023439), Lundbeckfonden (R54-A5858), and the Danish Council for Strategic Research Grant (11-116196).
- 4.Fahy, E., Subramaniam, S., Brown, H.A., Glass, C.K., Merrill Jr., A.H., Murphy, R.C., Raetz, C.R., Russell, D.W., Seyama, Y., Shaw, W., Shimizu, T., Spener, F., van Meer, G., VanNieuwenhze, M.S., White, S.H., Witztum, J.L., Dennis, E.A.: A comprehensive classification system for lipids. J. Lipid Res. 46, 839–861 (2005)CrossRefGoogle Scholar
- 8.Dennis, E.A., Deems, R.A., Harkewicz, R., Quehenberger, O., Brown, H.A., Milne, S.B., Myers, D.S., Glass, C.K., Hardiman, G., Reichart, D., Merrill Jr., A.H., Sullards, M.C., Wang, E., Murphy, R.C., Raetz, C.R., Garrett, T.A., Guan, Z., Ryan, A.C., Russell, D.W., McDonald, J.G., Thompson, B.M., Shaw, W.A., Sud, M., Zhao, Y., Gupta, S., Maurya, M.R., Fahy, E., Subramaniam, S.: A mouse macrophage lipidome. J. Biol. Chem. 285, 39976–39985 (2010)CrossRefGoogle Scholar
- 9.Tarasov, K., Ekroos, K., Suoniemi, M., Kauhanen, D., Sylvanne, T., Hurme, R., Gouni-Berthold, I., Berthold, H.K., Kleber, M.E., Laaksonen, R., Marz, W.: Molecular lipids identify cardiovascular risk and are efficiently lowered by simvastatin and PCSK9 deficiency. J. Clin. Endocrinol. Metab. 99, E45–E52 (2014)CrossRefGoogle Scholar
- 23.Pham, H.T., Maccarone, A.T., Thomas, M.C., Campbell, J.L., Mitchell, T.W., Blanksby, S.J.: Structural characterization of glycerophospholipids by combinations of ozone- and collision-induced dissociation mass spectrometry: the next step towards "top-down" lipidomics. Analyst 139, 204–214 (2014)CrossRefGoogle Scholar
- 24.Maccarone A.T., Duldig J., Mitchell T.W., Blanksby S.J., Duchoslav E., Campbell J.L.: Rapid and unambiguous characterization of acyl chain position in unsaturated phosphatidylcholines using differential mobility and mass spectrometry. J. Lipid Res 55, 1668-1677 (2014)Google Scholar
- 25.Scigelova M., Hornshaw M., Giannakopulos A., Makarov A.: Fourier transform mass spectrometry. Mol. Cell. Proteom 10, M111. 009431 (2011)Google Scholar
- 26.Li, F., Qin, X., Chen, H., Qiu, L., Guo, Y., Liu, H., Chen, G., Song, G., Wang, X., Li, F., Guo, S., Wang, B., Li, Z.: Lipid profiling for early diagnosis and progression of colorectal cancer using direct-infusion electrospray ionization Fourier transform ion cyclotron resonance mass spectrometry. Rapid Commun. Mass Spectrom. 27, 24–34 (2013)CrossRefGoogle Scholar
- 31.Ejsing, C.S., Moehring, T., Bahr, U., Duchoslav, E., Karas, M., Simons, K., Shevchenko, A.: Collision-induced dissociation pathways of yeast sphingolipids and their molecular profiling in total lipid extracts: a study by quadrupole TOF and linear ion trap-Orbitrap mass spectrometry. J. Mass Spectrom. 41, 372–389 (2006)CrossRefGoogle Scholar
- 47.Sokol, E., Almeida, R., Hannibal-Bach, H.K., Kotowska, D., Vogt, J., Baumgart, J., Kristiansen, K., Nitsch, R., Knudsen, J., Ejsing, C.S.: Profiling of lipid species by normal-phase liquid chromatography, nanoelectrospray ionization, and ion trap-orbitrap mass spectrometry. Anal. Biochem. 443, 88–96 (2013)CrossRefGoogle Scholar
- 48.Southam, A.D., Payne, T.G., Cooper, H.J., Arvanitis, T.N., Viant, M.R.: Dynamic range and mass accuracy of wide-scan direct infusion nanoelectrospray Fourier transform ion cyclotron resonance mass spectrometry-based metabolomics increased by the spectral stitching method. Anal. Chem. 79, 4595–4602 (2007)CrossRefGoogle Scholar
- 51.Hsu F.F., Lodhi I.J., Turk J., Semenkovich C.F.: Structural distinction of diacyl-, alkylacyl, and alk-1-enylacyl glycerophosphocholines as [M – 15] ions by multiple-stage linear ion-trap mass spectrometry with electrospray ionization. J. Am. Soc. Mass Spectrom 25, 1412-1420 (2014)Google Scholar