Multileaf-collimator (MLC) defined small fields in radiotherapy are used in high dose, ultra-conformal techniques such as stereotactic radiotherapy and stereotactic radiosurgery. Proximity to critical structures and irreversible damage arising from inaccurate delivery mean that correct positioning of the MLC system is of the utmost importance. Some of the existing techniques for MLC positioning quality assurance make use of electronic portal imaging device (EPID) images. However, conventional collimation verification algorithms based on the full width at half maximum (FWHM) fail when applied to small field images acquired by an EPID due to overlapping aperture penumbrae, lateral electron disequilibrium and radiation source occlusion. The objective of this study was to investigate sub-pixel edge detection and other techniques with the aim of developing an automatic and autonomous EPID-based method suitable for MLC positional verification of small static fields with arbitrary shapes. Methods investigated included derivative interpolation, Laplacian of Gaussian (LoG) and an algorithm based on the partial area effect hypothesis. None of these methods were found to be suitable for MLC positioning verification in small field conditions. A method is proposed which uses a manufacturer-specific empirically modified FWHM algorithm which shows improvement over the conventional techniques in the small field size range. With a measured mean absolute difference from planned position for Varian linacs of 0.01 ± 0.26 mm, compared with the erroneous FWHM value of 0.70 ± 0.51 mm. For Elekta linacs the proposed algorithm returned 0.26 ± 0.25 mm, in contrast to the FWHM result of 1.79 ± 1.07 mm.
This is a preview of subscription content, log in to check access.
Buy single article
Instant access to the full article PDF.
Price includes VAT for USA
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
This is the net price. Taxes to be calculated in checkout.
Yang Y, Xing L (2004) Quantitative measurement of MLC leaf displacements using an electronic portal image device. Phys Med Biol 49(8):1521
Mamalui-Hunter M, Li H, Low DA (2008) MLC quality assurance using EPID: a fitting technique with subpixel precision. Med Phys 35(6):2347–2355
Samant SS, Zheng W, Parra NA, Chandler J, Gopal A, Wu J, Jain J, Zhu Y, Sontag M (2002) Verification of multileaf collimator leaf positions using an electronic portal imaging device. Med Phys 29(12):2900–2912
Eilertsen K (1997) Automatic detection of single MLC leaf positions with corrections for penumbral effects and portal imager dose rate characteristics. Phys Med Biol 42(2):313
Mohammadi M, Bezak E (2006) Evaluation of MLC leaf positioning using a scanning liquid ionization chamber EPID. Phys Med Biol 52(1):N21
Lebron S, Yan G, Li J, Lu B, Liu C (2017) A universal parameterized gradient-based method for photon beam field size determination. Med Phys 44(11):5627–5637
Rowshanfarzad P, Sabet M, Barnes MP, O’Connor DJ, Greer PB (2012) EPID-based verification of the MLC performance for dynamic IMRT and VMAT. Med Phys 39(10):6192–6207
Rowshanfarzad P, Sabet M, O’Connor DJ, Greer PB (2011) Isocenter verification for linac-based stereotactic radiation therapy: review of principles and techniques. J Appl Clin Med Phys 12(4):185–195
Klein EE, Hanley J, Bayouth J, Yin F-F, Simon W, Dresser S, Serago C, Aguirre F, Ma L, Arjomandy B, Liu C, Sandin C, Holmes T (2009) Task Group 142 report: quality assurance of medical accelerators. Med Phys 36(9):4197–4212
Benedict SH, Yenice KM, Followill D, Galvin JM, Hinson W, Kavanagh B, Keall P, Lovelock M, Meeks S, Papiez L (2010) Stereotactic body radiation therapy: the report of AAPM Task Group 101. Med Phys 37(8):4078–4101
Charles P, Cranmer-Sargison G, Thwaites DI, Crowe S, Kairn T, Knight R, Kenny J, Langton CM, Trapp J (2014) A practical and theoretical definition of very small field size for radiotherapy output factor measurements. Med Phys. https://doi.org/10.1118/1.4868461
Kairn T, Asena A, Charles P, Hill B, Langton CM, Middlebrook N, Moylan R, Trapp J (2015) Field size consistency of nominally matched linacs. Australas Phys Eng Sci Med 38(2):289–297
Alber M, Broggi S, De Wagter C, Eichwurzel I, Engstrom P, Fiorino C, Georg D, Hartmann G, Knoos T, Leal A et al (2008) Guidelines for the verification of IMRT. In: ESTRO booklet 9. Brussels
Chui C-S, Spirou S, LoSasso T (1996) Testing of dynamic multileaf collimation. Med Phys 23(5):635–641
Neal B, Ahmed M, Kathuria K, Watkins T, Wijesooriya K, Siebers J (2016) A clinically observed discrepancy between image-based and log-based MLC positions. Med Phys 43(6 Part 1):2933–2935
Aspradakis MM, Byrne JP, Palmans H, Duane S, Conway J, Warrington AP, Rosser K (2010) Small field MV photon dosimetry. IPEM report 103, IPEM
IAEA (2017) Dosimetry of small static fields used in external beam radiotherapy. Technical reports series. International Atomic Energy Agency, Vienna
Andreo P (2017) The physics of small megavoltage photon beam dosimetry. Radiother Oncol 126(2):205–213
Das IJ, Ding GX, Ahnesjo A (2008) Small fields: Nonequilibrium radiation dosimetry. Med Phys 35(1):206–215
LoSasso T (2008) IMRT delivery performance with a varian multileaf collimator. Int J Radiat Oncol Biol Phys 71(1):S85–S88
Clews L, Greer PB (2009) An EPID based method for efficient and precise asymmetric jaw alignment quality assurance. Med Phys 36(12):5488–5496
Greer PB, Cadman P, Lee C, Bzdusek K (2009) An energy fluence-convolution model for amorphous silicon EPID dose prediction. Med Phys 36(2):547–555
Du W, Gao S, Wang X, Kudchadker RJ (2012) Quantifying the gantry sag on linear accelerators and introducing an MLC-based compensation strategy. Med Phys 39(4):2156–2162
Rowshanfarzad P, McGarry CK, Barnes MP, Sabet M, Ebert MA (2014) An EPID-based method for comprehensive verification of gantry, EPID and the MLC carriage positional accuracy in Varian linacs during arc treatments. Radiat Oncol 9(1):249
Rowshanfarzad P, Sabet M, O’Connor DJ, Greer PB (2011) Verification of the linac isocenter for stereotactic radiosurgery using cine-EPID imaging and arc delivery. Med Phys 38(7):3963–3970
Fuangrod T, Woodruff HC, Rowshanfarzad P, O’Connor DJ, Middleton RH, Greer PB (2014) An independent system for real-time dynamic multileaf collimation trajectory verification using EPID. Phys Med Biol 59(1):61–81
Gao Z, Szanto J, Gerig L (2007) Using multileaf collimator interleaf leakage to extract absolute spatial information from electronic portal imaging device images. J Appl Clin Med Phys 8(1):1–9
Prince SJD (2012) Computer vision: models, learning, and inference. Cambridge University Press, Cambridge
Ding L, Goshtasby A (2001) On the Canny edge detector. Pattern Recognit 34(3):721–725
Canny J (1986) A computational approach to edge detection. IEEE Trans Pattern Anal Mach Intell 6:679–698
Bijhold J, Gilhuijs KGA, Van Herk M, Meertens H (1991) Radiation field edge detection in portal images. Phys Med Biol 36(12):1705
Trujillo-Pino A, Krissian K, Alemán-Flores M, Santana-Cedrés D (2013) Accurate subpixel edge location based on partial area effect. Image Vis Comput 31(1):72–90
Szeliski R (2010) Computer vision: algorithms and applications. Springer, New York
Bushberg JT, Boone JM (2011) The essential physics of medical imaging. Lippincott Williams & Wilkins, Philadelphia
Vial P, Oliver L, Greer PB, Baldock C (2006) An experimental investigation into the radiation field offset of a dynamic multileaf collimator. Phys Med Biol 51(21):5517
Bergman AM, Gete E, Duzenli C, Teke T (2014) Monte Carlo modeling of HD120 multileaf collimator on Varian TrueBeam linear accelerator for verification of 6X and 6X FFF VMAT SABR treatment plans. J Appl Clin Med Phys 15(3):148–163
Bakhtiari M, Kumaraswamy L, Bailey D, De Boer S, Malhotra H, Podgorsak M (2011) Using an EPID for patient-specific VMAT quality assurance. Med Phys 38(3):1366–1373
Ding GX, Duggan DM, Coffey CW (2006) Commissioning stereotactic radiosurgery beams using both experimental and theoretical methods. Phys Med Biol 51(10):2549
The authors wish to gratefully acknowledge Prof. Martin Ebert for his helpful comments during the research meetings.
Conflict of interest
The authors declare that they have no conflict of interest.
This article does not contain any studies with human participants or animals performed by any of the authors.
About this article
Cite this article
Hiatt, J., Mukwada, G., Barnes, M. et al. MLC positioning verification for small fields: a new investigation into automatic EPID-based verification methods. Australas Phys Eng Sci Med 41, 945–955 (2018). https://doi.org/10.1007/s13246-018-0690-3
- Small field MLC EPID-based verification
- SRT quality assurance
- Sub-pixel edge detection