Investigation of X-ray focal spot alignment using a jig of novel design
- 154 Downloads
The X-ray focal spot and the centre of the flattening filter should be placed as close as possible to the central axis (CAX) on a linear accelerator (linac) to produce a radially symmetric beam. The aim of this study is to devise a method to easily optimise the focal spot position of Varian linac photon beams generated with a flattening filter. A simple and robust jig was designed and built to be inserted into the largest electron applicator. Accessory and jaw position interlocks were overridden to enable photon beam operation. The jig was made from aluminium and consists of a plate permanently fixed inside a low melting point alloy (LMPA) insert frame and a block machined to suspend an ion chamber below the plate, such that the axis of the chamber is at the linac isocentre. The jig was used to optimise the position of the X-ray beam focal spot with respect to the device central axis (CAX). This was achieved by minimising the percentage change in ionisation chamber signal between collimator rotations from 90° to 270° as position steering was changed. As part of the investigation, in-plane (radial) and cross-plane (transverse) profiles obtained from water phantom scans were used to quantify how large the percentage ionisation change can be before profiles are distorted in the region of the CAX.
KeywordsFocal spot Quality assurance Alignment Commissioning
The authors would like to acknowledge Mr Steve Cripps from Varian Medical Systems Australasia for his valuable time and contribution to this investigation.
Compliance with ethical standards
Conflicts of interest
The authors declare that they have no conflict of interest.
This article does not contain any studies with human participants or animals performed by any of the authors.
- 1.Podgorsak EB (2005) Radiation oncology physics: a handbook for teachers and students. IAEA, ViennaGoogle Scholar
- 2.Greene D, Williams PC (1997) Linear accelerators for radiation therapy. 2nd edn. Institute of Physics Pub, BristolGoogle Scholar
- 3.Kirkness B (2006) C-series Clinac® accelerator system basics. Revision AG. Varian Medical Systems, Palo AltoGoogle Scholar
- 4.ICRU (1976) ICRU Report 24. Determination of Absorbed dose in a Patient Irradiated by beams of X or Gamma Rays in Radiotherapy Procedures International Commission on Radiation Units and Measurements, Washington, D.CGoogle Scholar
- 5.Van Dyk J (2013) The modern technology of radiation oncology. A compendium for medical physicists and radiation oncologists. Volume 3 Medical Physics Publishing, MadisonGoogle Scholar
- 7.International Electrotechnical Commission (2014) IEC 60601-2-1:2009+AMD1:2014 Medical electrical equipment- Part 2-1: Particular requirements for the basic safety and essential performance of electron accelerators in the range 1 MeV to 50 MeVGoogle Scholar
- 11.International Electrotechnical Commission (2011) IEC 61217:2011 Radiotherapy equipment- Coordinates, movements and scalesGoogle Scholar
- 13.Rohrig N (2006) Structural shielding design and evaluation for megavoltage x- and gamma-ray radiotherapy facilities, NCRP Report No. 151. Health Phys 91:270. doi: 10.1097/01.hp.0000205229.89321.ce
- 16.International Organization for Standardization (1995), ISO/ IEC Guide 98–3: 2008 Uncertainty of measurement—Part 3: guide to the expression of uncertainty in measurement (GUM), ISO, GenevaGoogle Scholar