Verification of Articulated Arm Coordinate Measuring Machines Accuracy Using LaserTracer System as Standard of Length
- 287 Downloads
This article describes a novel approach to verification of articulated arm coordinate measuring machines (AA CMMs) based on ISO 10360-2 standard. The approach utilizes the LaserTracer (LT) system as standard of length and automation of calibration procedure using an industrial robot. In this method, the robot is programmed to repeatedly move the AA CMM. The retroreflector is integrated with the stylus of AA CMM. Location of the retroreflector is constantly tracked by LaserTracer and, in select positions, the measurements of distance are performed by both the AA CMM and LT system. The verification of AA CMM accuracy is carried out through the comparison of the two measurement values. The developed procedure is recommended mainly for checking AA CMMs with large measuring volume, where the application of common artefacts is usually insufficient, or time-consuming, due to the fact that measurements have to be performed for several subspaces in the AA CMM measuring volume.
KeywordsArticulated arm coordinate measuring machine AACMM LaserTracer Verification test Accuracy
Reported research was realized within confines of project financed by Polish National Centre for Research and Development No. 0869/R/T022010/10.
- K. Takamasu, Present problems in coordinate metrology for nano and micro scale measurements, MAPAN-J. Metrol. Soc. India, 26(1) (2011) 3–14.Google Scholar
- J. Sładek, Accuracy of coordinate measurements. Publishing House of Cracow University of Technology, Cracow, (2011), pp. 1–495.Google Scholar
- ISO/DIS 10360-12:2014 Geometrical product specifications (GPS)—acceptance and reverification tests for coordinate measuring systems (CMS)—part 12: articulated arm coordinate measurement machines (CMM).Google Scholar
- ASME B89.4.22:2004 Methods for performance evaluation of articulated arm coordinate measuring machines (CMM).Google Scholar
- VDI/VDE 2617-9 Accuracy of coordinate measuring machines. Characteristics and their reverification. Acceptance and reverification tests for articulated arm coordinate measuring machines.Google Scholar
- ISO 10360-2:2009 Geometrical product specifications (GPS)—acceptance and reverification tests for coordinate measuring machines (CMM)—part 2: CMMs used for measuring linear dimensions.Google Scholar
- K. Ostrowska, A. Gąska, R. Kupiec, and J. Sładek, Accuracy assessment of coordinate measuring arms using LaserTracer system, 11th IMEKO TC14 international symposium on measurement and quality control, ISMQC 2013, Cracow and Kielce, Poland (2013), pp. 98–101.Google Scholar
- O. Guarneros, J. De Vicente, M. Maya, J.L. Ocana, C. Molpeceres, J.J. Garcia-Ballesteros, S.R. Rodriguez, and H.M. Duran, Uncertainty estimation for performance evaluation of a confocal microscope as metrology equipment, MAPAN-J. Metrol. Soc. India, 29(1) (2014) 29–42.Google Scholar
- G. Moona, R. Sharma, U. Kiran, and K.P. Chaudhary, Evaluation of measurement uncertainty for absolute flatness measurement by using Fizeau Interferometer with phase-shifting capability, MAPAN-J. Metrol. Soc. India, 29(4) (2014) 261–267.Google Scholar
- K. Iimura, E. Kataoka, M. Ozaki, and R. Furutani, The uncertainty of parallel model coordinate measuring machine, MAPAN-J. Metrol. Soc. India, 26(1) (2011) 47–53.Google Scholar
- ISO/TS 23165:2006 Geometrical product specifications (GPS)—guidelines for the evaluation of coordinate measuring machine (CMM) test uncertainty.Google Scholar