Friction and Wear Characteristics of Surface-Modified Titanium Alloy for Metal-on-Metal Hip Joint Bearing
Cobalt-chromium-molybdenum bearing materials have excellent wear resistance, but due to the cytotoxicity issue of chromium ions, they need to be replaced with biocompatible metal materials. Titanium alloy is a biocompatible material, and is used as a stem and acetabular shell of a hip joint. However, titanium alloy is not used as a bearing material because its wear resistance is poor. Therefore, in order to evaluate the applicability of titanium alloys for metal on metal bearing combinations, surface modification was performed in three methods. Surface modification by surface texturing, thermal oxidation, and thermal oxidation - surface texturing was performed on one side of the friction pair. Thermally oxidized-textured titanium surface showed the best friction and wear characteristics as the bearing surface. The friction decreased by 62% and the wear resistance increased by up to 300 times compared with the other combinations. This was due to two causes. First, the formation of an oxide layer onto the surface of the dimpled titanium resulted in reduced occurrence of adhesive wearing, and the plastic deformation of the edge of the dimple is suppressed. Second, dimples that were not deformed by oxidation collected hard oxide wear particles inside, which reduced the occurrence of abrasive wear.
KeywordsSurface texturing Thermal oxidation Titanium alloy Wear resistance MOM hip joint
Unable to display preview. Download preview PDF.
- 6.Cohen, D., “How Safe are Metal-on-Metal Hip Implants?” BMJ, Vol. 344, Paper No. e1410, 2012.Google Scholar
- 11.Pettersson, M., Tkachenko, S., Schmidt, S., Berlind, T., Jacobson, S., et al., “Mechanical and Tribological Behavior of Silicon Nitride and Silicon Carbon Nitride Coatings for Total Joint Replacements,” Journal of the Mechanical Behavior of Biomedical Materials, vol. 25, pp. 41–47, 2013.CrossRefGoogle Scholar
- 13.Head, W. C., Bauk, D. J., and Emerson, J. R., “Titanium as the Material of Choice for Cementless Femoral Components in Total Hip Arthroplasty,” Clinical Orthopaedics and Related Research, vol. 311, pp. 85–90, 1995.Google Scholar
- 14.Kim, H., “Study on the Tribologic Evaluation of TiN and TiAlN Thin Film Coatings for Application of Orthopaedic Medical Devices,” Ph.D. Thesis, Sejong University, 2012.Google Scholar
- 24.ATI Metals, “ATI 6-4 ELMTM,” https://www.atimetals.com/Products/Pages/ati-6-4-eli.aspx (Accssed 13 APR 2018)Google Scholar
- 28.Kim, D. W., Yuhta, T., Lee, K. W., and Yim, S. J., “Improvement of Lubrication Properties for New Surface Structure with Regular Patterning for the Frictional Surfaces of the Artificial Joints: Evaluation of Lubrication Properties by the Pendulum Type Hip Joint Simulator,” Journal of the Korean Orthopaedic Association, vol. 39, no. 7, pp. 745–752, 2004.CrossRefGoogle Scholar
- 29.Kim, D. W. and Yuhta, T., “A Basic Study on Functional Friction Surface of Artificial Joints,” Journal of Biomedical Engineering Research, vol. 22, no. 6, pp. 519–526, 2000.Google Scholar
- 33.Oshida, Y., “Bioscience and Bioengineering of Titanium Materials,” Elsevier, 2nd Ed., 2010.Google Scholar
- 36.Afoke, N. Y., Byers, P. D., and Hutton, W. C., “Contact Pressures in the Human Hip Joint,” Bone & Joint Journal, vol. 69, no. 4, pp. 536–541, 1987.Google Scholar
- 38.Kim, H. J., Shin, D. G., and Kim, D.-E., “Frictional Behavior between Silicon and Steel Coated with Graphene Oxide in Dry Sliding and Water Lubrication Conditions,” International Journal of Precision Engineering and Manufacturing-Green Technology, vol. 3, no. 1, pp. 91–97, 2016.MathSciNetCrossRefGoogle Scholar