Gyroscope-based assessment of temporal gait parameters during treadmill walking and running
- 634 Downloads
Wireless sensing solutions that provide accurate long-term monitoring of walking and running gait characteristics in a real-world environment would be an excellent tool for sport scientist researchers and practitioners. The purpose of this study was to compare the performance of a body-worn wireless gyroscope-based gait analysis application to a marker-based motion capture system for the detection of heel-strike and toe-off and subsequent calculation of gait parameters during walking and running. The gait application consists of a set of wireless inertial sensors and an adaptive algorithm for the calculation of temporal gait parameters. Five healthy subjects were asked to walk and run on a treadmill at two different walking speeds (2 and 4 kph) and at a jogging (8 kph) and running (12 kph) speed. Data were simultaneously acquired from both systems. True error, percentage error and ICC scores indicate that the adaptive algorithm successfully calculated strides times across all speeds. However, results showed poor to moderate agreement for stance and swing times. We conclude that this gait analysis platform is valid for determining stride times in both walking and running. This is a useful application, particularly in the sporting arena, where long-term monitoring of running gait characteristics outside of the laboratory is of interest.
KeywordsAdaptive algorithm Inertial sensor Gait events Heel-strike Toe-off Stride time Stance time Swing time
This research was completed as part of a wider program of research within the TRIL Centre (Technology Research for Independent Living). The TRIL Centre is a multi-disciplinary research centre, bringing together researchers from UCD, TCD and Intel, funded by Intel, GE Healthcare and IDA Ireland (http://www.trilcentre.org). The authors would like to thank Ms. Marie Bay and Mr. Flip van den Berg for their images.
Conflict of interest
Funding and SHIMMER hardware were provided by the Intel Corporation and the TRIL centre. The SHIMMER board design is owned by the Intel Corporation.
- 3.Burns A, McGrath MJ, Delaney J et al (2010b) Open shareable research platform for developing interoperable personal health systems. In: 1st AMA-IEEE medical technology conference on individualized medicine, Washington, DCGoogle Scholar
- 4.Channells J, Purcell B, Barrett R et al (2005) Determination of rotational kinematics of the lower leg during sprint running using accelerometers. In: BioMEMS and Nanotechnology II, BrisbaneGoogle Scholar
- 5.Chung WY, Bhardwaj S, Purwar A et al (2007) A fusion health monitoring using ECG and accelerometer sensors for elderly persons at home. In: International Conference of the IEEE EMBS, LyonGoogle Scholar
- 6.Ferraris F, Grimaldi U, Parvis M (1995) Procedure for effortless in-field calibration of three-axis rate gyros and accelerometers. Sens Mater 7(5):311–330Google Scholar
- 10.Hay JG, Reid JG (1993) Anatomy, mechanics, and human motion. Prentice-Hall, Englewood CliffsGoogle Scholar
- 11.Heiden T, Burnett A (2004) Determination of heel strike and toe-off in the running stride using an accelerometer: application to field based gait studies. In: International Symposium of Biomechanics in Sport, OttawaGoogle Scholar