Effect of X-ray beam quality on determination of exposure index
- 704 Downloads
We investigated the effect of X-ray beam qualities RQA3, 5, 7, and 9 on the exposure index (EI) as defined by International Electrotechnical Commission guideline 62494-1. Half-value layers (HVLs) of RQA5 X-rays passing through anti-scatter grids (grid ratios 6:1, 8:1, 10:1, and 12:1) were also evaluated because grids are frequently used in clinical situations. The maximum percent differences in the EIs for RQA3, 7, and 9 with respect to RQA5 were 35.0, 11.6, and 38.7 %, respectively. The range of HVLs for RQA5–7 beams was 7.10–9.10 mm of aluminum (mm Al). This was wider than the range of HVLs when grids were used (6.94–7.29 mm Al). The effect of variations in X-ray beam qualities in the RQA series on the EI was significantly greater than the effect of grids. This study indicated that, in clinical settings, the EI should be used carefully in X-ray examinations with different X-ray beam qualities.
KeywordsExposure index X-ray beam quality Digital radiography Computed radiography Anti-scatter grid
The authors are grateful to the editor in chief of this journal, Kunio Doi, Ph.D., The University of Chicago. We also thank the editors and the editorial assistant of this journal, Mrs. Elisabeth Lanzl, for providing initial and final polishing of the submitted manuscript in improving the readability and English expressions, and the reviewers for giving us useful comments and suggestions for improving our manuscript.
Compliance with ethical standards
Conflict of interest
We have nothing to declare for this study.
- 1.AAPM Report No. 116. An exposure indicator for digital radiography. Report of AAPM Task Group 116. American Association of Physicists in Medicine; 2009.Google Scholar
- 2.ICRP Publication 93. Managing patient dose in digital radiology. International Commission on Radiological Protection; Ann. ICRP 34(1). 2004.Google Scholar
- 5.IEC 62494-1, Ed. 1.0. Medical electrical equipment—Exposure index of digital X-ray imaging systems—Part 1: definition and requirements for general radiography. Geneva: International Electrotechnical Commission; 2008.Google Scholar
- 7.IEC 61267, Ed. 1.0. Medical diagnostic X-ray equipment—Radiation conditions for use in the determination of characteristics. Geneva: International Electrotechnical Commission; 1994.Google Scholar
- 8.IEC 62220-1, Ed. 1.0. Medical electrical equipment—Characteristics of digital X-ray imaging devices—Part 1: Determination of the detective quantum efficiency. Geneva: International Electrotechnical Commission; 2003.Google Scholar
- 11.Tatsumi D, Shiraishi J, Tsushima H, Kusumi K, Utsunomiya A. Relationship between incident dose on the imaging plate and S value associated with computed radiography (CR) systems: quantitative evaluation using uniform exposure. Jpn J Radiol Technol. 1998;54(11):1273–80 (in Japanese).Google Scholar
- 13.Instruction Manual on Image Processing for Fuji Computed Radiography. 2nd edn. FUJIFILM Corporation; 2001 (in Japanese).Google Scholar