High Metal Ion Levels After Use of the ASR™ Device Correlate With Development of Pseudotumors and T Cell Activation
- 603 Downloads
Pseudotumors and immunologic alterations are reported in patients with elevated metal ion levels after resurfacing arthroplasty of the hip. A direct association of increased cobalt and chromium concentrations with the development of pseudotumors has not been established.
We hypothesized that (1) patients with higher blood cobalt and chromium concentrations are more likely to have pseudotumors develop, (2) elevated cobalt and chromium concentrations correlate with increased activation of defined T cell populations, and (3) elevated metal ion levels, small implant size, cup inclination angle, and patient age are risk factors for the development of pseudotumors.
A single-surgeon cohort of 78 patients with 84 Articular Surface Replacement® implants was retrospectively investigated. Between 2006 and 2010, we performed 84 THAs using the Articular Surface Replacement® implant; this represented 2% (84/4950) of all primary hip replacements performed during that period. Of the procedures performed using this implant, we screened 77 patients (99%) at a mean of 43 months after surgery (range, 24–60 months). Seventy-one patients were investigated using ultrasound scanning, and cobalt and chromium concentrations in whole blood were determined by high-resolution inductively coupled plasma mass spectrometry. Differential analysis of lymphocyte subsets was performed by flow cytometry in 53 patients. Results of immunologic analyses were investigated separately for patients with and without pseudotumors. Pseudotumors were found in 25 hips (35%) and were more common in women than in men (p = 0.02). Multivariable regression analysis was performed to identify risk factors for the development of pseudotumors.
Cobalt and chromium concentrations were greater in patients with pseudotumors than in those without (cobalt, median 8.3 versus median 1.0 μg/L, p < 0.001; chromium, median 5.9 versus median 1.3 μg/L, p < 0.001). The percentage of HLA-DR+CD4+ T cells was greater in patients with pseudotumors than in those without (p = 0.03), and the proportion of this lymphocyte subtype was positively correlated with cobalt concentrations (r = 0.3, p = 0.02). Multivariable regression analysis indicated that increasing cobalt levels were associated with the development of pseudotumors (p < 0.001), and that patients with larger implants were less likely to have them develop (p = 0.04); age and cup inclination were not risk factors.
We found a distinct association of elevated metal ion concentrations with the presence of pseudotumors and a correlation of increased cobalt concentrations with the proportion of activated T helper/regulator cells. Thus, the development of soft tissue masses after metal-on-metal arthroplasty could be accompanied by activation of T cells, indicating that this complication may be partly immunologically mediated. Further investigations of immunologic parameters in larger cohorts of patients with metal-on-metal arthroplasties are warranted.
Level of Evidence
Level III, therapeutic study. See the Instructions for Authors for a complete description of levels of evidence.
- 2.Almousa SA, Greidanus NV, Masri BA, Duncan CP, Garbuz DS. The natural history of inflammatory pseudotumors in asymptomatic patients after metal-on-metal hip arthroplasty. Clin Orthop Relat Res. 2013 March 28 [Epub ahead of print].Google Scholar
- 3.al Saffar N, Revell PA. Interleukin-1 production by activated macrophages surrounding loosened orthopaedic implants: a potential role in osteolysis. Br J Rheumatol. 1994;33:309–316.Google Scholar
- 10.de Steiger RN, Hang JR, Miller LN, Graves SE, Davidson DC. Five-year results of the ASR XL Acetabular System and the ASR Hip Resurfacing System: an analysis from the Australian Orthopaedic Association National Joint Replacement Registry. J Bone Joint Surg Am. 2011;93:2287–2293.PubMedCrossRefGoogle Scholar
- 14.Hallab NJ, Caicedo M, McAllister K, Skipor A, Amstutz H, Jacobs JJ. Asymptomatic prospective and retrospective cohorts with metal-on-metal hip arthroplasty indicate acquired lymphocyte reactivity varies with metal ion levels on a group basis. J Orthop Res. 2013;31:173–182.PubMedCentralPubMedCrossRefGoogle Scholar
- 16.Hart AJ, Satchithananda K, Liddle AD, Sabah SA, McRobbie D, Henckel J, Cobb JP, Skinner JA, Mitchell AW. Pseudotumors in association with well-functioning metal-on-metal hip prostheses: a case-control study using three-dimensional computed tomography and magnetic resonance imaging. J Bone Joint Surg Am. 2012;94:317–325.PubMedGoogle Scholar
- 20.Langton DJ, Sidaginamale RP, Joyce TJ, Natu S, Blain P, Jefferson RD, Rushton S, Nargol AV. The clinical implications of elevated blood metal ion concentrations in asymptomatic patients with MoM hip resurfacings: a cohort study. BMJ open. 2013;3.Google Scholar
- 21.Langton DJ, Sprowson AP, Joyce TJ, Reed M, Carluke I, Partington P, Nargol AV. Blood metal ion concentrations after hip resurfacing arthroplasty: a comparative study of articular surface replacement and Birmingham Hip Resurfacing arthroplasties. J Bone Joint Surg Br. 2009;91:1287–1295.PubMedCrossRefGoogle Scholar
- 22.Medicines and Healthcare Products Regulatory Agency (MHRA). Medical Device Alert: ASR™ hip replacement implants manufactured by DePuy International Ltd (MDA/2010/069). Available at: http://www.mhra.gov.uk/Publications/Safetywarnings/MedicalDeviceAlerts/CON093789. Accessed September 16, 2013.
- 23.Nawabi DH, Gold S, Lyman S, Fields K, Padgett DE, Potter HG. MRI predicts ALVAL and tissue damage in metal-on-metal hip arthroplasty. Clin Orthop Relat Res. 2013 January 26 [Epub ahead of print].Google Scholar
- 27.R Core Team. R: A Language and Environment for Statistical Computing. Vienna, Austria; R Foundation for Statistical Computing; 2006. Available at: http://www.R-project.org/. Accessed September 17, 2013 (ISBN 3-900051-07-0).
- 29.Swedish Hip and Knee Society. [Monitoring of metal-on-metal prostheses in Sweden][in Swedish]. Available at: http://www.ortopedi.se/pics/6/59/Ytis_riktlinjer_120516001.pdf. Accessed July 1, 2013.