Development of a technical approach to modify the internal surface of biomedical tubes and other elongated small lumen macrodevices with parylene coating
A multitude of differently composed biomedical polymers has been researched for many years for their distinctive ease of production and wide range of applications. New technologies and new material characteristics have always evolved accordingly. The lumina of biomedical polymer tubes such as catheters, intravenous tubes, and biomedical microfluidic channels do not necessarily show the required biocompatibility and desired functionality. In such cases, the products are provided with additional inner liner or coatings to achieve the desired specific properties. Specific adjustments, for example, low friction coefficient, low gas diffusion resistance, wear resistance, and hydrophobicity, are key properties which are in focus for the improvement of biomedical surfaces. In this pilot study, a technical method was developed to deposit parylene-AF4 on the inner surface of silicone tubes with aspect ratios exceeding 78:1. Uncoated and parylene-AF4-coated silicone tubes were investigated in respect to the aforementioned physical properties. Compared to the uncoated tubes, the parylene-coated tubes showed superior quality with respect to friction coefficient, gas diffusivity as well as wear resistance. It could be demonstrated that the new technical approach is suitable to parylene-coat the inner surfaces of tubes with high aspect ratios thereby achieving conformal coatings.
KeywordsInternal coating Wear resistance Friction coefficient Gas diffusion Parylene
Compliance with ethical standards
Conflict of interest
Chintan Desai and Dr. Norbert Laube were employed by the NTTF Coatings GmbH until two years ago, a company developing, among other coatings, coating for urological implants.
The work was conducted according to the general rules of “Good Laboratory Practice”; no experiments with human or other living materials were performed.
- 2.Kulshrestha, AS, Mahapatro, A, Polymers for Biomedical Applications. ACS Symposium Series. American Chemical Society, Washington (2008)Google Scholar
- 8.Wang, L, Wang, Y, Yang, D, United States Patent. “Method of Coating Polymeric Tubes.” US Patent 6,447,835 B1, 2002Google Scholar
- 9.Jaganathan SK, Supriyanto E, Murugesan S, Balaji A, Asokan MK, “Biomaterials in Cardiovascular Research: Applications and Clinical Implications.” BioMed. Res. Int., 2014, Article ID 459465Google Scholar
- 10.Prashant, PU, “Current and Emerging Catheter Technologies for Percutaneous Transluminal Coronary Angioplasty.” Res. Rep. Clin. Cardiol., 5 213–226 (2014)Google Scholar
- 16.Charmet, J, Bitterli, J, Sereda, O, Liley, M, Renaud, P, Keppner, H, “Optimizing Parylene C Adhesion for MEMS Processes: Potassium Hydroxide Wet Etching.” JMEMS, 22 (4) 855–864 (2013)Google Scholar
- 18.Bruus, H, Theoretical Microfluidics. Oxford Master Series in Condensed Matter Physics. Oxford University Press, Oxford (2008)Google Scholar
- 19.Humphrey, BJ, “Vapor Phase Consolidation of Books with the Parylene Polymers.” JAIC, 25 (1) 15–29 (1986)Google Scholar
- 20.Kramer, P, Sharma, A, Hennecke, E, Yasuda, H, “Polymerization of Para-Xylylene Derivatives (Parylene Polymerization). I. Deposition Kinetics for Parylene N and Parylene C.” J. Polym. Sci.: Polymer Chemistry Edition, 22 (2) 475–491 (1984)Google Scholar
- 22.Gill, WN, Rogojevic, S, Lu, T, “Vapor Deposition of Low-k Polymeric Dielectrics.” In: Ho, PS, Leu, J, Lee, WW (eds.) Low Dielectric Constant Materials for IC Applications. Springer Series in Advanced Microelectronics, Vol. 9, XIX, pp. 95–117 (2003)Google Scholar
- 25.Lam, CNC, Lu, JJ, Neumann, AW, “Measuring Contact Angle.” In: Holmberg, K, Shah, DO, Schwuger, MJ (eds.) Handbook of Applied Surface and Colloid Chemistry, pp. 251–277. John Wiley & Sons, Chichester (2001)Google Scholar
- 26.American Society for Testing and Materials, “ASTM D523, Standard Test Method for Specular Gloss.” ASTM, 6.01 35–39 (2001)Google Scholar
- 28.Bernsmann, F, Laube, N, Baldsiefen, G, Castellucci, M, “Hydrogenated Amorphous Carbon Coatings on Implants Drastically Reduce Biofilm Formation and Water Permeation.” J. Phys. Conference Series, 564 (012001) 1–11 (2014)Google Scholar