Review of Superelastic Differential Force Archwires for Producing Ideal Orthodontic Forces: an Advanced Technology Potentially Applicable to Orthognathic Surgery and Orthopedics
Purpose of Review
Gentle and continuous loads are preferred for optimum orthodontic tooth movement. Nitinol, an alloy of nickel and titanium developed for the aerospace industry, found its first clinical applications in orthodontics because it has ideal load-deflection behavior. The purpose of this review is to elucidate the criteria for effective orthodontic mechanics relative to emerging Nitinol technology. The specialized materials with variable stiffness that were originally developed for orthodontics are increasingly attractive for in the temporomandibular joint, orthognathic surgery, and orthopedics.
The evolution of orthodontic archwires is driven by a need to achieve low load-deflection characteristics and Nitinol is the alloy of choice. Scientific knowledge of the biological response to orthodontic forces continues to grow, but definitive guidance on optimal force levels for individual teeth is elusive. Finite element models (FEM) that take into account periodontal ligament (PDL) stresses indicate differential force archwires are needed to realize optimal treatment. However, previous wire fabrication methods, including welding of different materials and selective resistive heating, are limited by poor mechanical performance and spatial resolution. Recently, a novel laser processing technique was developed for precisely programing relative levels of stiffness in a single archwire. FEM was used to estimate the optimal force for each tooth by calculating the 3D bone-PDL surface area.
There remains a general consensus that light and continuous forces are desirable for orthodontic treatment. New developments in archwire materials and technology have provided the orthodontist with a complete spectrum of load-deflection rates and differential force options to express these forces with maximized archwire economy. These technologies also appear to have application to orthopedic implant devices.
KeywordsOrthodontic archwires Superelastic Nitinol Shape memory alloy Laser processing
Compliance with Ethical Standards
Conflict of Interest
Ibraheem Khan has a patent (US20170224444A1) pending, and Smarter Alloys Inc. (with which all authors are affiliated) is involved in the design and manufacture of appliances for orthodontic treatment.
Human and Animal Rights and Informed Consent
This article does not contain any studies with human or animal subjects performed by any of the authors.
Papers of particular interest, published recently, have been highlighted as: • Of importance •• Of major Importance
- 1.Storey E, Smith R. Force in orthodontics and its relation to tooth movement. Aust J Dent. 1952;56:291–304.Google Scholar
- 2.Roberts WE, Sarandeep SH. Bone physiology, metabolism, and biomechanics in orthodontic practice. In: Graber LW, Vanarsdall RL, KWL V, Huang GJ, editors. Orthodontics: current principles and techniques. 6th ed. Oxford: Elsevier Health Sciences; 2016. p. 99–152.Google Scholar
- 5.Burstone CJ, Baldwin JJ, Lawless DT. The application of continuous forces to orthodontics. Angle Orthod. 1961;31:1–14.Google Scholar
- 6.Harder OE, Roberts DA. Alloy having high elastic strengths. US Patent 2,524,661. 1950.Google Scholar
- 27.Coro JC, Coro IM. Nonsurgical treatment of class III malocclusions using the multi-loop edgewise archwire appliance. In: OrthodonticProductsOnline.com. 2012. http://op.alliedmedia360.com//launch.aspx?eid=fdd7e997-e095-421e-a3db-f6650d9d1075. Accessed 06 April 2018.
- 28.Nordstrom B, Shoji T, Anderson WC, Fields HW, Beck FM, Kim D-G, et al. Comparison of changes in irregularity and transverse width with nickel-titanium and niobium-titanium-tantalum-zirconium archwires during initial orthodontic alignment in adolescents: a double-blind randomized clinical trial. Angle Orthod. 2018. In press;88:348–54.CrossRefPubMedGoogle Scholar
- 32.Limsiriwong S, Khemaleelakul W, Sirabanchongkran S, Pothacharoen P, Kongtawelert P, Ongchai S, et al. Biochemical and clinical comparisons of segmental maxillary posterior tooth distal movement between two different force magnitudes. Eur J Orthod. 2017:1–8.Google Scholar
- 45.Miura F. Orthodontic archwire. US Patent 5,017,133. 1991.Google Scholar
- 47.Sachdeva R, Farzin-Nia F. Shape memory orthodontic archwire having variable recovery stresses. US Patent 5,683,245. 1997.Google Scholar
- 48.Mehta ASK. Thermomechanical characterization of variable force NiTi orthodontic archwires. Master’s Thesis. Marquette University. 2015. http://epublications.marquette.edu/theses_open/328. Accessed 06 April 2018.
- 60.Aiyer A, Russell NA, Pelletier MH, Myerson M, Walsh WR. The impact of Nitinol staples on the compressive forces, contact area, and mechanical properties in comparison to a claw plate and crossed screws for the first tarsometatarsal arthrodesis. Foot Ankle Int. 2016;9:232–40.Google Scholar