Metallurgical failure analysis of various implant materials used in orthopedic applications
- 710 Downloads
• In the investigation of a total hip prosthesis, the fracture was multiorigined, probably due to the presence of manufacturing defects in both surfaces of the device; was associated with poor design of the component; and introduced points of high stress concentration.
• The study revealed that the failure of the stainless steel femoral compression plates occurred by a fretting-fatigue mechanism under the action of low nominal stresses and unidirectional bending.
• The failure of femoral compression titanium plates occurred by corrosion fatigue promoted by the presence of intense localized corrosion and intergranular cracking.
• The failure of hollow spinal rods occurred by overload caused by the intense formation of pitting corrosion during service. The pitting was associated with the presence of superficial manufacturing routes.
As a result of the analysis of various implant materials, the main reasons for failure are corrosive attack, manufacturing defects, and nonstandard-caused failures. Biocompatible inorganic materials, such as hydroxy-apatite coating with some binders, would make the dissolution of metal ions difficult and so may delay corrosion and wear and also minimize the loosening of implants from bone.
KeywordsFatigue Failure Analysis Scanning Electron Micro Intergranular Crack Hydroxyapatite Coating
Unable to display preview. Download preview PDF.
- 1.G.K. Smith and J. Black: “Models for Systemic Effects of Metallic Implants,” Proc. Int. Symp. on Retrieval and Analysis of Orthopaedic Implants, NBS special publ. 472, National Bureau of Standards, MD, 1976, pp. 23–30.Google Scholar
- 3.C.R.F. Azevedo: “Failure Analysis of a Commercially Pure Titanium Plate for Osteosynthesis,” J. Eng. Failure Analys., 2003, 10, in press.Google Scholar
- 4.G. Pitrowski: “Clinical Biomedics,” Proc. Int. Symp. on Retrieval and Analysis of Orthopaedic Implants, NBS special publ. 472, National Bureau of Standards, MD, 1976, pp. 41–49.Google Scholar
- 5.D.F. Williams: “A Review of Metallurgical Failure Modes in Orthopaedic Implants,” Proc. Int. Symp. on Retrieval and Analysis of Orthopaedic Implants, NBS special publ. 472, National Bureau of Standards, MD, 1976, pp. 11–21.Google Scholar
- 12.K. Ishikawa, M. Kon, S. Tenshin, and N. Kuwayama: “Effect of Preparation Conditions in Aqueous Solution on Properties of Hydroxyapatite,” Dent. Mater., 1990, 9, pp. 58–69.Google Scholar
- 14.J.L. Gilbert, C.A. Buckley, et al.: “Intergranular Corrosion-Fatigue Failure of Cobalt-Alloy Femoral Stems,” J. Bone Joint Surg. Am., 1994, 76(1), pp.110–15.Google Scholar
- 16.W.L. Jaffe and D.F. Scott: “Total Hip Arthroplasty with Hydroxyapatite-Coated Prosthesis,” J. Bone Joint Surg., 1996, 78A(12), pp. 1919–34.Google Scholar
- 18.D.W. Howie, S.D. Rogers, et al.: “Biological Effect of Cobalt Chrome in Cell and Animal Models,” Clin. Orthop., 1996, pp. 217–32.Google Scholar
- 19.M.A. Akoy, M. Baydoğan, et al.: “Failure Analysis on a Femur Prosthesis,” The Fifth Int. Fracture Symp. (Elaziğ, Turkey), Sept. 2001.Google Scholar
- 20.G. Christian, S. Stefanie, et al.: “Implant Failure of the Gamma Nail,” Injury, Int. J. Care Injured, 1999, 30, pp. 91–99.Google Scholar
- 21.M.M. Edwin, V.A. Edward, J.S. Peter, and J.S. Neal: “Exchange Nailing for Failure of Initially Rodded Tibial Shaft Fractures,” J. Orthop., 2001, 24(8), pp. 757–62.Google Scholar
- 22.H.G. Willer, L.G. Broback, and G.H. Buchorn: “Crevice Corrosion of Cemented Titanium Hip Implants,” Clin. Orthop., 1996, 333, pp. 51–57.Google Scholar
- 23.B.F. Shagaldi and J. Compson: “Wear and Corrosion of Sliding Counterparts of Stainless-Steel Hip Screw-Plates,” Injury, Int. J. Care Injured, 2000, 31, pp. 85–92.Google Scholar
- 29.G. Berger, U. Ploska, and G. Willmann: “Hydroxyapatite’s Solubility May Cause Loosening of Coated Implants,” Proc. 13th Int. Symp. on Ceramics in Medicine (Bologna, Italy), 2001, pp. 192–95.Google Scholar
- 30.O.E.M. Pohler: “Failure of Metallic Orthopaedic Implants,” Materials Characterization, Vol 10, Metals Handbook, 9th ed., American Society for Metals, Metals Park, OH, 1986, pp. 668–94.Google Scholar
- 31.J.H. Dumplenton and E.H. Miller: “Failure of Metallic Orthopaedic Implants,” in Metals Handbook, 8th ed., ASM International, Materials Park, OH, 1993, pp. 571–80.Google Scholar
- 32.K.R. Wheller and L.A. James: “Fatigue Behaviour of Type 316 Stainless Steel under Simulated Body Conditions,” J. Biomed. Mater. Res., 1971, pp. 267–81.Google Scholar
- 33.S. Suresh: Fatigue of Materials, Cambridge University Press, New York, 1998.Google Scholar
- 34.S.S. Manson: ASTM STP 495, Avoidance, Control, and Repair of Fatigue Damage, ASTM, Philadelphia, 1971, pp. 254–346.Google Scholar
- 35.R.O. Ritchie, R.H. Dauskardt, and B.N. Cox: “Fatigue of Advanced Materials: Summary and Future Trends,” Proc. of the Eng. Foundation Int. Conf. (Santa Barbara, CA), 1991, pp. 485–93.Google Scholar
- 36.G. Haicheng, G. Huifang, C. Shufen, and C. Laird: “Orientation Dependence of Cyclic Deformation in High Purity Titanium Single Crystals,” Mater. Sci. Eng. A, 1994, 118, pp. 23–36.Google Scholar
- 37.I. Aitchison and B. Cox: “Interpretation of Fractographs of SSC in Hexagonal Metals,” Corrosion, 1972, 28(3), pp. 83–87.Google Scholar