Flexible needle and patient tracking using fractional scanning in interventional CT procedures
- 18 Downloads
We present a new method for flexible needle and patient localization in interventional CT procedures based on fractional CT scanning. Our method accurately localizes the trajectory of a flexible needle to which a spherical marker is attached at a known distance from the tip with respect to a baseline scan of patient in the CT scanner coordinate frame.
The localization is achieved with a significantly lower dose compared to a full scan using sparse view angle sampling and without reconstructing the CT image of the repeat scan. Our method starts by performing rigid registration between the patient and the baseline scan in 3D Radon space computed from the sparse projection data. It then computes 2D projection difference images in which the flexible needle and the spherical marker appear as prominent features. Their 3D spatial locations are then automatically extracted from the projection images to accurately trace the flexible needle trajectory. To validate our method, we conducted registration and needle trajectory localization experiments in seven abdomen phantom scans using a short and a long flexible needle.
Our experimental results yield a mean needle trajectory localization error of 0.7 ± 0.2 mm and a mean tip localization error of 2.4 ± 0.9 mm with a \(\times \)7.5 radiation dose reduction with respect to a full CT scan.
The significant radiation dose reduction enables more frequent needle trajectory localization during the needle insertion for a similar total dose, or a reduced total dose for the same localization frequency.
KeywordsInterventional CT Fractional CT scanning Reduced dose CT scanning Flexible needle tracking Radon space registration
We thank Eyal Lin and Ronen Shter of GE Healthcare Israel for the CT scans and for their valuable assistance.
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
Human and animal rights
This article does not contain any studies with human participants or animals performed by any of the authors.
- 1.Gupta R, Walsh C, Wang IS, Kachelrieß M, Kuntz J, Bartling S (2014) CT-guided interventions: current practice and future directions. In: Jolesz FA (ed) Intraoperative imaging and image-guided therapy. Springer, New York, pp 173–191Google Scholar
- 4.Chodick G, Ronckers CM, Shalev V, Ron E et al (2007) Excess lifetime cancer mortality risk attributable to radiation exposure from computed tomography examinations in children. IMAJ-RAMAT GAN- 9(8):584Google Scholar
- 5.Miller DL, Vañó E, Bartal G, Balter S, Dixon R, Padovani R, Schueler B, Cardella JF, De Baère T (2010) Occupational radiation protection in interventional radiology: a joint guideline of the Cardiovascular and Interventional Radiology Society of Europe and the Society of Interventional Radiology. Cardiovasc Interv Radiol 33(2):230–239CrossRefGoogle Scholar
- 10.Siemens Healthcare (2018) syngo DynaCT. https://www.HrBhealthcare.siemens.com/angio/options-and-upgrades/clinical-softHrBware-applications/syngo-dynact/technical-details. Accessed 24 Oct 2018
- 18.Wu G, Li X, Lehocky CA, Riviere CN (2013) Automatic steering of manually inserted needles. In: IEEE international conference on systems, man, and cybernetics (SMC), 2013. IEEE, pp 1488–1493Google Scholar
- 22.Vrooijink GJ, Abayazid M, Misra S (2013) Real-time three-dimensional flexible needle tracking using two-dimensional ultrasound. In: IEEE international conference on robotics and automation (ICRA), 2013. IEEE, pp 1688–1693Google Scholar
- 23.Marinetto E, Uneri A, De Silva T, Reaungamornrat S, Zbijewski W, Sisniega A, Vogt S, Kleinszig G, Pascau J, Siewerdsen J (2017) Integration of free-hand 3D ultrasound and mobile C-arm cone-beam CT: Feasibility and characterization for real-time guidance of needle insertion. Comput Med Imaging Graph 58:13–22CrossRefGoogle Scholar
- 24.Huo B, Zhao X, Han J, Xu W (2015) Shape reconstruction and attitude estimation of bevel-tip needle via CT-guidance. In: IEEE international conference on robotics and biomimetics (ROBIO), 2015. IEEE, pp 620–625Google Scholar
- 28.Adelman Z (2018) Reduced-dose region-of-interest image reconstruction in repeat ct scanning. Master Thesis, The Hebrew University of Jerusalem, Dec 2018Google Scholar