Robust navigation support in lowest dose image setting
- 47 Downloads
Clinical cardiac electrophysiology (EP) is concerned with diagnosis and treatment of cardiac arrhythmia describing abnormality or perturbation in the normal activation sequence of the myocardium. With the recent introduction of lowest dose X-ray imaging protocol for EP procedures, interventional image enhancement has gained crucial importance for the well-being of patients as well as medical staff.
In this paper, we introduce a novel method to detect and track different EP catheter electrodes in lowest dose fluoroscopic sequences based on \(\ell _1\)-sparse coding and online robust PCA (ORPCA). Besides being able to work on real lowest dose sequences, the underlying methodology achieves simultaneous detection and tracking of three main EP catheters used during ablation procedures.
We have validated our algorithm on 16 lowest dose fluoroscopic sequences acquired during real cardiac ablation procedures. In addition to expert labels for 2 sequences, we have employed a crowdsourcing strategy to obtain ground truth labels for the remaining 14 sequences. In order to validate the effect of different training data, we have employed a leave-one-out cross-validation scheme yielding an average detection rate of \(86.9\%\).
Besides these promising quantitative results, our medical partners also expressed their high satisfaction. Being based on \(\ell _1\)-sparse coding and online robust PCA (ORPCA), our method advances previous approaches by being able to detect and track electrodes attached to multiple different catheters.
KeywordsComputer-assisted electrophysiology Electrode tracking Sparse coding Online robust PCA Online tracking Fluoroscopy
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki Declaration and its later amendments or comparable ethical standards. For this type of study formal consent is not required. This article does not contain any studies with animals performed by any of the authors.
Informed consent was obtained from all individual participants included in the study.
- 1.Ambrosini P, Ruijters D, Niessen WJ, Moelker A, van Walsum T (2017) Fully automatic and real-time catheter segmentation in X-ray fluoroscopy. In: International conference on medical image computing and computer-assisted intervention. Springer, pp 577–585Google Scholar
- 3.Bourier F, Reents T, Ammar-Busch S, Buiatti A, Kottmaier M, Semmler V, Telishevska M, Brkic A, Grebmer C, Lennerz C, Kolb C, Hessling G, Deisenhofer I (2015) Evaluation of a new very low dose imaging protocol: feasibility and impact on X-ray dose levels in electrophysiology procedures. Europace 18(9):1406–1410CrossRefGoogle Scholar
- 9.Frangi AF, Niessen WJ, Vincken KL, Viergever MA (1998) Multiscale vessel enhancement filtering. In: Wells WM, Colchester A, Delp S (eds) Medical image computing and computer-assisted intervention—MICCAI’98. Springer, Berlin, pp 130–137Google Scholar
- 13.Javed S, Bouwmans T, Jung SK (2017) Improving OR-PCA via smoothed spatially-consistent low-rank modeling for background subtraction. In: Proceedings of the symposium on applied computing. ACM, pp 89–94Google Scholar
- 14.Javed S, Oh SH, Sobral A, Bouwmans T, Jung SK. (2014) OR-PCA with MRF for robust foreground detection in highly dynamic backgrounds. In: Asian conference on computer vision. Springer, pp 284–299Google Scholar
- 18.Matviychuk Y, Mailhé B, Chen X, Wang Q, Kiraly A, Strobel N, Nadar M (2016) Learning a multiscale patch-based representation for image denoising in X-ray fluoroscopy. In: 2016 IEEE international conference on image processing (ICIP). IEEE, pp 2330–2334Google Scholar
- 19.Milletari F, Belagiannis V, Navab N, Fallavollita P (2014) Fully automatic catheter localization in c-arm images using \(\ell _1\)-sparse coding. In: Medical image computing and computer-assisted intervention—MICCAI 2014, Part II, pp 570–577Google Scholar
- 20.Otsu N (1975) A threshold selection method from gray-level histograms. Automatica 11(285–296):23–27Google Scholar
- 21.Panayiotou M, Rhode KS, King AP, Ma Y, Cooklin M, Neill OM, Gill J, Housden CA, Housden RJ (2015) Image-based view-angle independent cardiorespiratory motion gating and coronary sinus catheter tracking for X-ray-guided cardiac electrophysiology procedures. Phys Med Biol 60(20):8087CrossRefGoogle Scholar
- 23.Vaswani N, Bouwmans T, Javed S, Narayanamurthy P (2018) Robust PCA, subspace learning, and tracking. In: IEEE signal processing magazine, vol 35, no 4, pp 32–55Google Scholar
- 28.Zhan J, Lois B, Guo H, Vaswani, N (2016) Online (and offline) robust PCA: novel algorithms and performance guarantees. In: Artificial intelligence and statistics, pp 1488–1496Google Scholar