Autonomous neuro-registration for robot-based neurosurgery
- 65 Downloads
Neuro-registration is of primary importance as it has a bearing on the accuracy of neurosurgery. Although the accuracy of surgical robots is within the acceptable medical standards, the overall surgical accuracy is dictated by the errors in the neuro-registration process. The purpose of this work is to automate the neuro-registration process to improve the overall accuracy of the robot-based neurosurgery.
A highly accurate 6-degree-of-freedom Parallel Kinematic Mechanism (6D-PKM) robot is used for both neuro-registration and neurosurgery. In neuro-registration, after measurement of points in the medical image space, the end-platform of the 6D-PKM surgical robot carrying the camera will autonomously navigate towards the fiducial markers to measure its coordinates in the real patient space. An accurate relationship between the medical image space and the real patient space is established, and the same robot will navigate the surgical tool to the target.
In order to validate the proposed method for autonomous neuro-registration, experiments are performed using four phantoms. The four phantoms are as follows: PVC skull model, two acrylic blocks and a glass jar with coaxial shells. These phantoms are specifically designed to simulate the neurosurgical process. All the phantoms are registered successfully using the above-stated method. After autonomous neuro-registration, the coordinates of the target point are determined. Neurosurgery validation is carried out by attaching a 1-mm-diameter needle to the robot platform, which is autonomously traversed to reach the target point passing through the two 2-mm-diameter coaxial holes. The experiments are repeated, and the results reveal very good repeatability.
A method for autonomous neuro-registration has been developed. The robot has been successfully registered using the above method. After successful neuro-registration the overall accuracy of the robot-based neurosurgery is considerably improved. The other benefits of the above method are as follows: elimination of line-of-sight problem, no need of extra unit for neuro-registration, less time for registration, intraoperative registration, human error reduction and low cost.
KeywordsAutonomous neuro-registration Robot-based neurosurgery Neuronavigation Image-guided surgery Application of parallel manipulator in surgery
Funding has been provided by Bhabha Atomic Research Centre, Mumbai, India.
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
This article does not contain any studies with animals performed by any of the authors.
- 1.Paraskevopoulos D, Unterberg A, Metzner R, Dreyhaupt J, Eggere G, Wirtz C (2011) Comparative study of application accuracy of two frameless neuronavigation systems: experimental error assessment quantifying registration methods and clinically influencing factors. Neurosurg Rev 34:217–228. https://doi.org/10.1007/s10143-010-0302-5 CrossRefGoogle Scholar
- 4.Bhutani G (2015) Modeling design and development of frameless stereotaxy in robot assisted neurosurgery. HBNI, MumbaiGoogle Scholar
- 6.Kaushik A, Dwarakanath TA, Bhutani G, Venkat PPK (2017) Image based data preparation for robot based neurosurgery. In: Machines, Mechanism and Robotics, Proceedings of iNaCoMM 2017. Springer, BerlinGoogle Scholar
- 7.Surgical Navigation Surgical Navigation Systems : Stryker. http://www.stryker.com/latm/products/OREquipmentConnectivity/SurgicalNavigation/SurgicalNavigationSystems/index.htm. Accessed 26 Mar 2018
- 8.Medical Software and Hardware Innovators - Brainlab. https://www.brainlab.com/en/. Accessed 26 Mar 2018
- 9.Pirotte B, Voordecker P, Joffroy F, Massager N, Wikler D, Baleriaux D, Levivier M, Brotchi J (2001) The Zeiss-MKM system for frameless image-guided approach in epidural motor cortex stimulation for central neuropathic pain. Neurosurg Focus 11:E3. https://doi.org/10.3171/foc.2001.11.3.4 CrossRefPubMedGoogle Scholar
- 13.Grillone GA, Jalisi S (2014) Robotic surgery of the head and neck: a comprehensive guide. Springer, BerlinGoogle Scholar
- 19.Brainlab Automatic Image Registration for Brainlab Navigation (GE). https://www.brainlab.com/en/surgery-products/overview-neurosurgery-products/intraoperative-mr/. Accessed 18 Mar 2018
- 20.Dwarakanath S, Bhutani G, Venkata PPK, Kaushik A (2017) Generalized point correspondence algorithm for neuro-registration. In: Machines, Mechanism and Robotics, Proceedings of iNaCoMM 2017. Springer, BerlinGoogle Scholar
- 23.(2011) PS 3.3-2011 Digital Imaging and Communications in Medicine (DICOM) Part 3: Information Object Definitions. NEMAGoogle Scholar
- 24.Pianykh OS (2011) Digital imaging and communications in medicine (DICOM)Google Scholar
- 25.Hartley RZA (2004) Multiple view geometry. CambridgeGoogle Scholar
- 26.opencv dev team Camera calibration With OpenCV—OpenCV 18.104.22.168 documentation. https://docs.opencv.org/2.4/doc/tutorials/calib3d/camera_calibration/camera_calibration.html#theory. Accessed 19 Mar 2018
- 27.Alexander Mordvintsev & Abid K (2013) Pose Estimation—OpenCV-Python Tutorials 1 documentation. In: OpenCV. http://opencvpythontutroals.readthedocs.io/en/latest/py_tutorials/py_calib3d/py_pose/py_pose.html. Accessed 19 Mar 2018
- 28.Sinha SK, Dwarakanath TA, Jaju A (2017) Six DOF mirror alignment system for beam line applications. In: Machines, Mechanism and Robotics, Proceedings of iNaCoMM 2017. Springer, BerlinGoogle Scholar