User-friendly freehand ultrasound calibration using Lego bricks and automatic registration
- 261 Downloads
As an inexpensive, noninvasive, and portable clinical imaging modality, ultrasound (US) has been widely employed in many interventional procedures for monitoring potential tissue deformation, surgical tool placement, and locating surgical targets. The application requires the spatial mapping between 2D US images and 3D coordinates of the patient. Although positions of the devices (i.e., ultrasound transducer) and the patient can be easily recorded by a motion tracking system, the spatial relationship between the US image and the tracker attached to the US transducer needs to be estimated through an US calibration procedure. Previously, various calibration techniques have been proposed, where a spatial transformation is computed to match the coordinates of corresponding features in a physical phantom and those seen in the US scans. However, most of these methods are difficult to use for novel users.
We proposed an ultrasound calibration method by constructing a phantom from simple Lego bricks and applying an automated multi-slice 2D–3D registration scheme without volumetric reconstruction. The method was validated for its calibration accuracy and reproducibility.
Our method yields a calibration accuracy of \(1.23\pm 0.26\) mm and a calibration reproducibility of 1.29 mm.
We have proposed a robust, inexpensive, and easy-to-use ultrasound calibration method.
KeywordsUltrasound Calibration Phantom Image registration Lego 2D–3D
Compliance with ethical standards
Conflict of interest
Yiming Xiao, Charles Xiao Bo Yan, Simon Drouin, Dante De Nigris, Anna Kochanowska, and D. Louis Collins declare that they have no conflict of interest.
- 14.Comeau RM, Fenster A, Peters TM, (1998) Integrated MR and ultrasound imaging for improved image guidance in neurosurgery. In: Proceedings of SPIE, The International Society for Optical Engineering, pp 747–754Google Scholar
- 17.Boctor E, Viswanathan A, Choti M, Taylor RH, Fichtinger G, Hager G (2004) A novel closed form solution for ultrasound calibration. In: 2nd IEEE international symposium on biomedical imaging: macro to nano, Vols 1 and 2. pp 527–530, 2004Google Scholar
- 23.Blackall JM, Rueckert D, Maurer CR, Penney GP, Hill DLG, Hawkes DJ (2000) An image registration approach to automated calibration for freehand 3D ultrasound. Med Image Comput Comput-Assist Interv—Miccai 1935:462–471Google Scholar
- 24.Mercier L, Del Maestro RF, Petrecca K, Kochanowska A, Drouin S, Yan CX, Janke AL, Chen SJ, Collins DL (2011) New prototype neuronavigation system based on preoperative imaging and intraoperative freehand ultrasound: system description and validation. Int J Comput Assist Radiol Surg 6:507–522CrossRefPubMedGoogle Scholar
- 27.Soehl M, Walsh R, Rankin A, Lasso A, Fichtinger G (2014) Tracked ultrasound calibration studies with a phantom made of LEGO bricks. Med Imaging, Image-Guided Proced, Robot Interv, Modeling 9036:2014Google Scholar
- 29.Cleary K, Anderson J, Brazaitis M, Devey G, DiGioia A, Freedman M, Gronemeyer D, Lathan C, Lemke H, Long D, Mun SK, Taylor R (2000) Final report of the technical requirements for image-guided spine procedures workshop, April 17–20, 1999, Ellicott City, MA, USA. Comput Aided Surg 5:180–215PubMedGoogle Scholar