Calibration of the motor-assisted robotic stereotaxy system: MARS
- 285 Downloads
The motor-assisted robotic stereotaxy system presents a compact and light-weight robotic system for stereotactic neurosurgery. Our system is designed to position probes in the human brain for various applications, for example, deep brain stimulation. It features five fully automated axes. High positioning accuracy is of utmost importance in robotic neurosurgery.
First, the key parameters of the robot’s kinematics are determined using an optical tracking system. Next, the positioning errors at the center of the arc—which is equivalent to the target position in stereotactic interventions—are investigated using a set of perpendicular cameras. A modeless robot calibration method is introduced and evaluated. To conclude, the application accuracy of the robot is studied in a phantom trial.
We identified the bending of the arc under load as the robot’s main error source. A calibration algorithm was implemented to compensate for the deflection of the robot’s arc. The mean error after the calibration was 0.26 mm, the 68.27th percentile was 0.32 mm, and the 95.45th was 0.50 mm.
The kinematic properties of the robot were measured, and based on the results an appropriate calibration method was derived. With mean errors smaller than currently used mechanical systems, our results show that the robot’s accuracy is appropriate for stereotactic interventions.
KeywordsStereotactic neurosurgery Robot calibration Isocenter Robotic surgery Deep brain stimulation (DBS)
Unable to display preview. Download preview PDF.
- 8.Bronstein J, Tagliati M, Alterman R, Lozano A, Volkmann J, Stefani A, Horak F, Okun M, Foote K, Krack P et al (2010) Deep brain stimulation for Parkinson disease: an expert consensus and review of key issues. Arch Neurol (online)Google Scholar
- 9.Schulder M (ed) (2003) Handbook of stereotactic and functional neurosurgery. Informa HealthcareGoogle Scholar
- 10.Medtronic: Nexframe stereotactic system manual. Tech. rep., Medtronic Inc Minneapolis USA (2010). URL http://professional.medtronic.com/
- 16.Morgan PS, Carter T, Davis S, Sepehri A, Punt J, Byrne P, Moody A, Finlay P (2003) The application accuracy of the pathfinder neurosurgical robot. In: International congress series CARS 2003. Computer assisted radiology and surgery. Proceedings of the 17th international congress and exhibition, pp 561–567Google Scholar
- 18.Medtech: Rosa neurosurgery robot. Tech. rep., Medtech SAS Montpellier France (2011). URL http://www.medtechsurgical.com/Products/ROSA
- 20.Craig JJ (2005) Introduction to robotics, mechanics and control. Pearson Prentice Hall, NJ, p 07458Google Scholar
- 21.Heinig M, Govela MF, Gasca F, Dold C, Hofmann UG, Tronnier V, Schlaefer A, Schweikard A (2011) Mars—motor assisted robotic stereotaxy system. In: Proceedings of the 5th international IEEE EMBS conference on neural engineering, pp 334–337Google Scholar
- 23.Neitzel M, Mitschang P, Beresheim G (2004) Handbuch Verbundwerkstoffe: Werkstoffe, Verarbeitung, Anwendung. Hanser, MunichGoogle Scholar
- 24.atracsys: Accutrack 250. Accessed 29 Aug 2011 (2011). URL http://atracsys.com/products/trackingsystems.php
- 25.Mooring B, Driels M, Roth Z (1991) Fundamentals of manipulator calibration. Wiley, New YorkGoogle Scholar
- 29.Rachinger J, Bumm K, Wurm J, Bohr C, Nissen U, Dannenmann T, Buchfelder M, Iro H, Nimsky C (2007) A new mechatronic assistance system for the neurosurgical operating theatre: Implementation, assessment of accuracy and application concepts. Stereotact Funct Neurosurg 85(5): 249–255PubMedCrossRefGoogle Scholar