A motion constrained cross-wire phantom for tracked 2D ultrasound calibration
- 262 Downloads
Ultrasound-guided 3D interventions require calibration to relate real-time 2D images with the position and orientation of the ultrasound probe. Capturing several images of a single fixed point from different viewpoints is a simple and commonly used approach, but it is cumbersome and tedious. A new phantom for calibration was designed, built and tested to simplify this process.
A mechanical phantom that restricts the motion of the ultrasound probe was designed such that the ultrasound image always captures a designated fixed point. Software was implemented which computes calibration parameters. Although the software provides no scientific novelty, it is required to demonstrate the proof of concept and to assess the accuracy and precision of the calibration phantom. The software also illustrates how the phantom enables the fixed point to be located automatically, both in tracker device coordinates and in image pixel coordinates.
The phantom was used to capture several hundred images of a single fixed point in less than 1 min, with different probe positions and orientations around the fixed point and with the single fixed point located in different parts of the ultrasound image. It would not be feasible to capture the same number of images by manual alignment of the probe with the fixed point.
Images for single fixed point calibration can be captured easily and quickly with a new calibration phantom. Since a larger number of images can be used to compute the required parameters, the calibration robustness is increased.
KeywordsTracked 2D ultrasound Tracked ultrasound Freehand 3D ultrasound Freehand 3D ultrasound calibration Calibration
Unable to display preview. Download preview PDF.
- 3.Blackall JM, Rueckert D, Maurer CR Jr, Penney GP, Hill DLG, Hawkes DJ (2000) An image registration approach to automated calibration for freehand 3d ultrasound. In: MICCAI ’00: proceedings of the third international conference on medical image computing and computer-assisted intervention. Springer, London, pp 462–471Google Scholar
- 4.Boctor EM, Fichtinger G, Taylor RH, Choti MA (2003) Tracked 3D ultrasound in radio-frequency liver ablation. In: Walker WF, Insana MF (eds) Society of photo-optical instrumentation engineers (SPIE) conference series, society of photo-optical instrumentation engineers (SPIE) conference series, vol 5035. pp 174–182. doi: 10.1117/12.479952
- 5.Brendel B, Winter S, Ermert H (2004) A simple and accurate calibration method for 3d freehand ultrasound. Biomed Tech 49: 872–873Google Scholar
- 9.Galassi M, Davies J, Theiler J, Gough B, Jungman G, Booth M, Rossi F (2005) GNU scientific library reference manual, revised 2nd edn, chap 37. Network Theory Limited, UK, pp, pp 487–500Google Scholar
- 13.Hsu PW (2007) Freehand three-dimensional ultrasound calibration. PhD thesis, University of CambridgeGoogle Scholar
- 17.Moré JJ (1978) Numerical analysis, lecture notes in mathematics, vol 630, chap. The Levenberg-Marquardt algorithm: implementation and theory. Springer, Berlin, pp 105–116Google Scholar
- 20.Tehnical specifications of the Aurora tracking system. http://www.ndigital.com/medical/aurora-techspecs-performance.php