Image-based electrode array tracking for epicardial electrophysiological mapping in minimally invasive arrhythmia surgery
- 93 Downloads
Electrophysiological mapping is effective in realizing a precise minimally invasive arrhythmia surgery. Recently, an epicardial electrophysiological mapping system for minimally invasive arrhythmia surgery was reported. The system requires a small electrode array, a tracking system and a global mapping algorithm. The optical tracking system employed in the research requires line of sight and complicated configuration. This paper proposes a new tracking method for locating an electrode array.
We developed a small electrode array and optical markers. Center points of respective optical markers and the electrode array are tracked via an endoscopic stream and calculated in image space. The orientation of the electrode array is calculated using the dot product between the vector joining two center points of two upper optical markers and the vector joining two end points of the longest edge of the electrode array.
Mean tracking errors of position and orientation of the electrode array were 0.51 mm and 0.64°, respectively. And the processing time was constant at 46 ms per frame. Our method could successfully track the electrode array on the epicardium during in vivo experiment and a global epicardial electrophysiological map was reconstructed from separately measured epicardial electrograms by the small electrode array.
An image-based tracking method for locating an electrode array was proposed. Tracking accuracy, processing time and applicability to surgical environment of our method proved to be acceptable. Consequently, our method enables the electrode array tracking system to be simplified with no separate tracking system.
KeywordsArrhythmia Minimally invasive Maze procedure Electrophysiological map Cardiac tracking Object tracking system
Unable to display preview. Download preview PDF.
- 6.Paul T, Windhagen-Mahnert B, Kriebel T, Bertram H, Kaulitz R, Korte T, Niehaus M, Tebbenjohanns J (2001) Atrial reentrant tachycardia after surgery of congenital heart disease: endocardial mapping and radiofrequency catheter ablation using a novel non-contact mapping system. Circulation 103: 2266–2271PubMedGoogle Scholar
- 8.Takata Y (2009) Global epicardial electrophysiological mapping with local multi-channel electrode array. JSCAS 11(1):25–38 (in Japanese)Google Scholar
- 12.Harris C (1992) Tracking with rigid models. In: Blake A, Yuille A (eds) Active vision. MIT Press, Cambridge, pp 59–74Google Scholar
- 15.Burschka D, Corso JJ, Dewan M, Hager GD, Lau W, Li M, Lin H Marayong P, Ramey N (2004) Navigating inner space: 3-d assistance for minimally invasive surgery. In: Workshop advances in robot vision, IEEE/RSJ internationl conference on intelligent robots and systems, Sendai, Japan, pp 67–78Google Scholar
- 16.Casals A, Amat J, Laporte E (1996) Automatic guidance of an assistant robot in laparoscopic surgery. In: Proceedings 1996 IEEE international conference on robotics and automation, vol 1, pp 895–900Google Scholar
- 18.Tonet O, Ramesh TU, Megali G, Dario P (2005) Image analysis-based approach for localization of endoscopic tools. In: Proceedings of the 2nd international conference on computer-aided medical interventions: tools and applications, pp 60–65Google Scholar
- 19.Voros S, Orvain E, Cinquin P (2006) Automatic detection of instruments in laparoscopic images: a first step towards high level command of robotized endoscopic holders. In: BIOROB 2006 1st IEEE/RAS-EMBS international conference on biomedical robotics and biomechatronics, PisaGoogle Scholar
- 20.de Lange P (2009) Condensation algorithm based epicardium tracking. J JSCAS 11(3): 312–313Google Scholar
- 24.Pratt W (1991) Digital image processing, 2nd edn. Wiley, New YorkGoogle Scholar
- 25.Nakamura Y, Kishi K, Kawakami H (2001) Heartbeat synchronization for robotic cardiac surgery. In: Proceedings of the 2001 IEEE international conference on robotics & automation, pp 2014–2019Google Scholar