Path generator control system and virtual compliance calculator for maxillofacial massage robots
Oral disorders such as temporomandibular joint disorders (TMD) and dry mouth are common and often require treatment. Maxillofacial massage is used as a complementary and alternative therapy for these disorders. We developed an oral rehabilitation robot that massages the maxillofacial tissues for this purpose. In this paper, we propose a control system for oral rehabilitation robots.
The control system consists of a massage path generator, virtual compliance calculator, and inverse kinematics calculator. The massage path generator computes a target massage path based on a human head model obtained from a reference MRI image of an adult male. The head model includes the shape and elastic modulus of each component, all of which were obtained experimentally. Virtual compliance control is used to control manipulators with position servo actuators. The manipulators, which have a force sensor at their end-effectors, move actively in the direction of the external force applied to their sensors via virtual compliance control. We implemented this control in WAO-1, our first prototype oral rehabilitation robot.
WAO-1 provided massage to three adult male subjects with and without virtual compliance control. One of the subjects was the adult male whose MRI image was used to synthesize the head model in the massage path generator. Without virtual compliance control, the actual massage force was greater than the target massage force, while that with virtual compliance control was less than the target massage force. Furthermore, with virtual compliance control, the massage paths conformed to the head shape of each patient.
Implementation of virtual compliance control in the WAO-1 massage robot is feasible and useful for implementation of safe and potentially effective maxillofacial massage therapy.
KeywordsRehabilitation robot Maxillofacial massage TMD Drymouth Force control Robotic manipulator
- 1.NIH (2006) US Department of Health and Human Services, TMJ Disorders. NIH Publication No. 06-3487Google Scholar
- 2.Haskin CL, Milam SB, Cameron IL (1995) Pathogenesis of degenerative joint disease in the human temporomandibular joint. Critical reviews in oral biology and medicine. an official publication of the American Association of Oral Biologists. Crit Rev Oral Biol Med 6(3): 248–277CrossRefPubMedGoogle Scholar
- 3.Westesson PL (1993) Reliability and validity of imaging diagnosis of temporomandibular joint disorder. Adv Dental Res 7(2): 137–151Google Scholar
- 4.NIH (2006) US Department of Health and Human Services, Dry mouth. NIH Publication No. 06-3174Google Scholar
- 10.Panya M, Miyoshi T, Terashima K, Kitagawa H (2003) Expert massage motion control by multi-fingered robot hand. In: Proceedings of the 2003 IEEE/RSJ international conference on intelligent robots and systems, vol 3. pp 3035–3040Google Scholar
- 11.Chul-goo K, Bong-ju L, Ik-xu S, Ho-yeon K (2007) Design of a percussive massage robot tapping human backs. In: Proceedings of the 16th IEEE international symposium on robot and human interactive communication, pp 962–967Google Scholar
- 12.Koga H, Usuda Y, Matsuno M, Ogura Y, Ishii H, Solis J, Takanishi A, Katsumata A (2007) Development of oral rehabilitation robot for massage therapy. In: Proceedings of the international special topic conference on information technology in biomedicine, pp 111–114Google Scholar
- 13.Koga H, Usuda Y, Matsuno M, Ogura Y, Ishii H, Solis J, Takanishi A, Katsumata A (2008) Development of the oral rehabilitation robot WAO-1. In: Proceedings of the 2nd biennial IEEE/ RAS-EMBS international conference on biomedical robotics and biomechatronicsGoogle Scholar
- 14.Salisbury JK (1980) Active stiffness control of a manipulator in cartesian coordinates. In: Proceedings of the 19th IEEE conference on decision and control, pp 95–100Google Scholar
- 15.Sugahara Y, Hosobata T, Mikuriya Y, Lim H, Takanishi A (2003) Realization of stable dynamic walking by a parallel bipedal locomotor on uneven terrain using a virtual compliance control. In: Proceedings of the 2003 IEEE/RSJ international conference on intelligent robots and systems, pp 595–600Google Scholar