Mechanics of tissue rupture during needle insertion in transverse isotropic soft tissue
- 134 Downloads
During robot-assisted percutaneous puncture surgery, a bevel-tipped needle is inserted into a fiber-structured soft tissue, which has transverse isotropic mechanical properties. Tissue rupture may lead to needle bending and insertion error. Therefore, analysis of the mechanics of rupture events in transverse isotropic space is essential for the precise control of needle motion. We analyzed the needle stiffness force and the cutting force, both related to rupture events, which are caused by contact deformation and tissue cutting, respectively. A modified model based on a homogeneous elastic half-space contact force model was used to describe the variation in the stiffness force of the needle. An energy-based maximum stiffness force model was proposed to predict the surface rupture for a needle that is inserted at different orientations. A model for the cutting force was presented based on a solid mechanics theory of unidirectional fiber-reinforced composite material to analyze the effect of the needle tip pose on the cutting force. Experiments on porcine muscle confirmed that the stiffness force and the cutting force were clearly influenced by the insertion orientation angle. The developed models could predict the stiffness force and the cutting force during needle insertion into the transverse isotropic tissue. This method provides a foundation for predicting needle deflection in the transverse isotropic space.
KeywordsFiber-structured soft tissue Cutting force Stiffness force Insertion orientation Needle tip pose
The authors would like to thank Mr. H Suo for his contribution to the experiment design and data collection. The authors would also like to thank Dr. C Suo and Dr. BB Farouk for their help in proofreading the manuscript.
This work was supported by the National Natural Science Foundation of China (No. 51775368 and No. 51811530310).
- 1.Abayazid M (2015) Robotically steering flexible needles. Ph.D. dissertation, Dept. Faculty of Eng Tech, Twente Univ, Enschede, NetherlandsGoogle Scholar
- 7.Carra A, Avilavilchis JC (2010) Needle insertion modeling through several tissue layers. 2010 2nd International Asia Conference on CAR, pp 237–240Google Scholar
- 9.Feng Y, Lee CH, Sun L, Okamoto RJ, Ji S (2015) Automated estimation of elastic material parameters of a transversely isotropic material using asymmetric indentation and inverse finite element analysis. ASME 2015 International Mechanical Engineering Congress and Exposition, pp V003T03A007Google Scholar
- 10.Hing JT, Brooks AD, Desai JP (2006) Reality-based needle insertion simulation for haptic feedback in prostate brachytherapy. IEEE International Conference on Robotics Automation, pp 619–624Google Scholar
- 12.Khadem M, Fallahi B, Rossa C, Sloboda RS (2015) A mechanics-based model for simulation and control of flexible needle insertion in soft tissue. IEEE International Conference on Robotics Automation, pp 2264–2269Google Scholar
- 14.Lendvay TS, Hsieh FJ, Hannaford B, Rosen J (2008) The biomechanics of percutaneous needle insertion. 16th Conference on Medicine Meets Virtual Reality, pp 245–247Google Scholar
- 16.Mahvash M, Hayward V (2001) Haptic rendering of cutting, a fracture mechanics approach. Haptics-e 2:1–12Google Scholar
- 17.Mahvash M, Dupont PE (2010) Mechanics of dynamic needle insertion into a biological material. IEEE Trans Biomed Eng 57:934–943Google Scholar
- 18.Ní Annaidh A, Bruyère K, Destrade M, Gilchrist MD, Otténio M (2012) Characterization of the anisotropic mechanical properties of excised human skin. J Mech Behav Biomed Mater 28:139–148Google Scholar
- 25.Tai Y, Wei L, Zhou H, Nahavandi S, Shi J (2016) Tissue and force modelling on multi-layered needle puncture for percutaneous surgery training. IEEE International Conf on SMC, pp 002923–002927Google Scholar