Using a low-amplitude RF pulse at echo time (LARFET) for device localization in MRI
We describe a new method for frequency down-conversion of MR signals acquired with the radio-frequency projections method for device localization. A low-amplitude, off-center RF pulse applied simultaneously with the echo signal is utilized as the reference for frequency down-conversion. Because of the low-amplitude and large offset from the Larmor frequency, the RF pulse minimally interfered with magnetic resonance of protons. We conducted an experiment with the coil placed at different positions to verify this concept. The down-converted signal was transformed into optical signal and transmitted via fiber-optic cable to a receiver unit placed outside the scanner room. The position of the coil could then be determined by the frequency analysis of this down-converted signal and superimposed on previously acquired MR images for comparison. Because of minimal positional errors (≤0.8 mm), this new device localization method may be adequate for most interventional MRI applications.
KeywordsInterventional MRI Device localization Catheter tracking Frequency down-conversion Self-mixing
The authors thank Umut Cindemir and Berk Camli for their efforts during experiments, Mr. Francis Payne and Dr. Can Akgün for their valuable editorial support. The experiments were conducted at National Magnetic Resonance Research Center (UMRAM) at Bilkent University, Ankara and Acıbadem Kozyatağı Hospital, Istanbul. This study was supported by The Scientific and Technological Research Council Of Turkey (TUBITAK, Project 111E197) and Boğaziçi University LifeSci Center (Ministry of Development, 2009K1200520), and EU Marie Curie Actions IRSES Project 269300 (TAHITI, Improving Therapy and Intervention through Imaging).
- 2.Bakker CJ, Hoogeveen RM, Weber J, van Vaals JJ, Viergever MA, Mali WP (1996) Visualization of dedicated catheters using fast scanning techniques with potential for MR-guided vascular interventions. Magn Reson Med 36(6):816Google Scholar
- 3.Bakker CJ, Smits HF, Bos C, van der Weide R, Zuiderveld KJ, van Vaals JJ, Hurtak WF, Viergever MA, Mali WP (1998) MR-guided balloon angioplasty: in vitro demonstration of the potential of MRI for guiding, monitoring, and evaluating endovascular interventions. J Magn Reson Imaging 8(1):245Google Scholar
- 8.Busse H, Thormer G, Garnov N, Haase J, Kahn T, Moche M (2010) Technique for wireless position tracking of intravascular catheters: Performance evaluation in a vessel phantom. Proc Intl Soc Mag Reson Med 18:4163Google Scholar
- 16.Guttman MA, Ozturk C, Raval AN, Raman VK, Dick AJ, DeSilva R, Karmarkar P, Lederman RJ, McVeigh ER (2007) Interventional cardiovascular procedures guided by real-time MR imaging: an interactive interface using multiple slices, adaptive projection modes and live 3D renderings. J Magn Reson Imaging 26(6):1429PubMedCrossRefPubMedCentralGoogle Scholar
- 19.Kocaturk O, Saikus CE, Guttman MA, Faranesh AZ, Ratnayaka K, Ozturk C, McVeigh ER, Lederman RJ (2009) Whole shaft visibility and mechanical performance for active MR catheters using copper-nitinol braided polymer tubes. J Cardiovasc Magn Reson 11:29Google Scholar
- 27.Pitsaer C, Umathum R, Homagk AK, Ozturk C, Bock M (2010) Three concepts for tuning and matching intravascular catheter coils. Proc Intl Soc Mag Reson Med 8:1852Google Scholar
- 31.Ratnayaka K, Faranesh AZ, Guttman MA, Kocaturk O, Saikus CE, Lederman RJ (2008) Interventional cardiovascular magnetic resonance: still tantalizing. J Cardiovasc Magn Reson 10:62Google Scholar
- 33.Sarioglu B, Aktan O, Oncu A, Mutlu S, Dundar G, Yalcinkaya AD (2012) An optically powered CMOS receiver system for intravascular magnetic resonance applications. IEEE J Emerg Sel Topics Power Electron 2(4):683Google Scholar
- 35.Sun N, Liu Y, Lee H, Weissleder R, Ham D (2009) CMOS RF Biosensor Utilizing Nuclear Magnetic Resonance. IEEE J Solid-State Circ 44(5):1629Google Scholar
- 39.Weiss S, Schaeffter T, Luedeke K, Leussler C, Holz D, Nehrke K, Rasche V, Sinkus R (1999) Catheter localization using a resonant fiducial marker during interactive MR fluoroscopy. Proc Intl Soc Mag Reson Med 12:1954Google Scholar