Intraoperative forces and moments analysis on patient head clamp during awake brain surgery
- 454 Downloads
In brain surgery procedures, such as deep brain stimulation, drug-resistant epilepsy and tumour surgery, the patient is intentionally awakened to map functional neural bases via electrophysiological assessment. This assessment can involve patient’s body movements; thus, increasing the mechanical load on the head-restraint systems used for keeping the skull still during the surgery. The loads exchanged between the head and the restraining device can potentially result into skin and bone damage. The aim of this work is to assess such loads for laying down the requirements of a surgical robotics system for dynamic head movements compensation by fast moving arms and by an active restraint able to damp such actions. A Mayfield® head clamp was tracked and instrumented with strain gages (SGs). SG locations were chosen according to finite element analyses. During an actual brain surgery, displacements and strains were measured and clustered according to events that generated them. Loads were inferred from strain data. The greatest force components were exerted vertically (median 5.5 N, maximum 151.87 N) with frequencies up to 1.5 Hz. Maximum measured displacement and velocity were 9 mm and 60 mm/s, with frequencies up to 2.8 Hz. The analysis of loads and displacements allowed to identify the surgery steps causing maximal loads on the head-restraint device.
KeywordsAwake brain surgery Force sensors Intraoperative force/moments measurement Head clamp Head rest Motion compensation
This work was supported by the EU Project Grant ACTIVE FP7-ICT-2009-6-270460. Authors would like to thank Elisa Beretta for her precious help during the experimental setup preparation.
- 3.De Lorenzo D, De Momi E, Dyagilev I, Manganelli R, Formaglio A, Prattichizzo D, Shoham M, Ferrigno G (2011) Force feedback in a piezoelectric linear actuator for neurosurgery. Int J Med Robot Comput Assisted Surg 7:268–275Google Scholar
- 5.Dreier JD, Williams B, Mangar D, Camporesi EM (2009) Patients selection in awake neurosurgery. HSR Proc Intensive Care Cardiovasc Anesth 30:19–27Google Scholar
- 9.Lee M, Rezai AR, Chou J (1994) Depressed skull fractures in children secondary to skull clamp fixation devices. Pediatr Neurosurg 21:174–7; discussion 178Google Scholar
- 13.Malosio, M, Negri SP, Pedrocchi N, Vicentini F, Cardinale F, Tosatti LM (2012) The kinematic architecture of the active headframe: a new head support for awake brain surgery. In: Proceedings of the 34th annual international conference of the engineering in medicine and biology society, San Diego, USAGoogle Scholar
- 19.Seng KY, Lee Peter VS, Lam PM (2002) Neck muscle strength across the sagittal and coronal planes: an isometric study. Clin Biomech (Bristol, Avon) 17:545–547Google Scholar
- 20.Serramito-Garcia R, Arcos-Algaba A, Santin-Amo JM, Garcia-Allut A, Bandin-Dieguez FJ, Gelabert-Gonzalez M (2009) Epidural haematoma due to an headrest in an adult. Neurocirugia (Astur) 20:567–570Google Scholar
- 21.Szelenyi A, Bello L, Duffau H, Fava E, Feigl GC, Galanda M, Neuloh G, Signorelli F, Sala F, Workgroup for Intraoperative Management in Low-Grade Glioma Surgery within the European Low-Grade Glioma Network (2010) Intraoperative electrical stimulation in awake craniotomy: methodological aspects of current practice. Neurosurg Focus 28:E7Google Scholar
- 23.Vasavada AN, Li S, Delp SL (2001) Three-dimensional isometric strength of neck muscles in humans. Spine (Phila Pa 1976) 26:1904–1909Google Scholar