EIT image reconstruction with four dimensional regularization
- 167 Downloads
Electrical impedance tomography (EIT) reconstructs internal impedance images of the body from electrical measurements on body surface. The temporal resolution of EIT data can be very high, although the spatial resolution of the images is relatively low. Most EIT reconstruction algorithms calculate images from data frames independently, although data are actually highly correlated especially in high speed EIT systems. This paper proposes a 4-D EIT image reconstruction for functional EIT. The new approach is developed to directly use prior models of the temporal correlations among images and 3-D spatial correlations among image elements. A fast algorithm is also developed to reconstruct the regularized images. Image reconstruction is posed in terms of an augmented image and measurement vector which are concatenated from a specific number of previous and future frames. The reconstruction is then based on an augmented regularization matrix which reflects the a priori constraints on temporal and 3-D spatial correlations of image elements. A temporal factor reflecting the relative strength of the image correlation is objectively calculated from measurement data. Results show that image reconstruction models which account for inter-element correlations, in both space and time, show improved resolution and noise performance, in comparison to simpler image models.
KeywordsElectrical impedance tomography Regularization Spatial and temporal priors Image reconstruction
This work was supported by a grant from NSERC Canada.
- 5.Babaeizadeh S, Brooks DH (2006) Effects of electrode location error on boundary element impedance tomography solutions: CR bounds and simulation results. In: 3rd IEEE ISBI: nano to macro, pp. 1080–1083Google Scholar
- 11.Colton D, Kress R (1992) Inverse acoustic and electromagnetic scattering theory. Springer, Berlin, pp. 121, 289 (new edition: 1998, p. 133, 304)Google Scholar
- 12.Dai T, Soleimani M, Adler A (2007) Four-dimensional regularization for electrical impedance tomography imaging. In: VIII Conference on electrical impedance tomography, Graz, 29 August–2 September 2007Google Scholar
- 18.Hammermeister KE, Brooks RC, Warbasse JR (1974) The rate of change of left ventricular volume in Man: I. Validation and peak systolic ejection rate in health and disease. Circulation 49:729–738Google Scholar
- 19.Harris ND, Brown BH, Barber DC (1992) Continuous monitoring of lung ventilation with electrical impedance tomography. Med Biol Soc :1754–1755Google Scholar
- 22.Lionheart W, Polydorides N, Borsic A (2005) Why is EIT so hard. In: Holder DS (Ed) Electrical impedance tomography: methods, history and applications. IOP, BristolGoogle Scholar
- 24.Ormiston JA, Shah PM, Tei C, Wong M (1981) Size and motion of the mitral valve annulus in man. I. A two-dimensional echocardiographic method and findings in normal subjects. Circulation 64: 113–120Google Scholar