An admittance controller based on assistive torque estimation for a rehabilitation leg exoskeleton
- 99 Downloads
A rehabilitation exoskeleton leg was constructed for gait training, and a method for exoskeleton leg swing control based on admittance model was studied. The man–machine interaction torque, the inertia compensator torque, and the exoskeleton’s assistive torque act as the inputs to the admittance control system for realizing the effective assistance to the wearer. The exoskeleton’s assistive torque is generated by estimating net torque exerted by the muscle based on adaptive frequency oscillator. A variable swing frequency experiment for simulating physical rehabilitation exercise was implemented. The results show that the inertia compensator adjusting the compensation varies with the swing frequency. The control system provides effective assistance to the wearer. The coordinated control experiment of hip and knee joints was also implemented. The results show that the interaction forces are all controlled in a reasonable and small range, and there is a good coordination between the hip joint and knee joint during swing motion.
KeywordsRehabilitation exoskeleton leg Admittance control Adaptive frequency oscillator Coordinate motion Man–machine interaction
This work is supported by the National Natural Science Foundation of China (Grant No. 51205182), the Innovation Foundation of NJIT (Grant No. CKJA 201501, JXKJ201510), Six talent peaks project in Jiangsu Province (Grant No. JXQC-015).
- 6.Quinlivan B, Asbeck A, Wagner D et al (2015) Force transfer characterization of a soft exsuit for gait assistance. In: Proceedings of the ASME 2015 international design engineering technical conferences & computers and information in engineering conference, pp 1–7Google Scholar
- 8.Kawamoto H, Sankai Y (2003) Power assist method for HAL-3 using EMG-based feedback controller. In: IEEE international conference on systems, man and cybernetics, 2003, vol 2, pp 1648–1653Google Scholar
- 12.Aguirre-Ollinger G (2013) Learning muscle activation patterns via nonlinear oscillators: application to lower-limb assistance. In: IEEE/RSJ international conference on intelligent robots and systems (IROS), Tokyo, Japan, 2013, November 3–7, pp 1182–1189Google Scholar
- 14.Torabi M, Sharifi M, Vossoughi G (2018) Robust adaptive sliding mode admittance control of exoskeleton rehabilitation robots. Sci Iran B 25(5):2628–2642Google Scholar
- 16.Ronsse R, Vitiello N, Lenzi T, van den Kieboom J, Carrozza M, Ijspeert A (2010) Adaptive oscillators with human-in-the-loop: Proof of concept for assistance and rehabilitation. In: IEEE RAS and EMBS international conference on biomedical robotics and biomechatronics (BioRob) 2010, pp 668–674Google Scholar
- 21.Petric T, Gams A, Ijspeert AJ et al (2011) On-line frequency adaptation and movement imitation for rhythmic robotic tasks. Int J Robot Res 30(4):1755–1788Google Scholar