Several preclinical imaging systems are commercially available, but their purchase and maintenance costs make them unaffordable for the majority of small- and medium-sized groups. Taking into account the needs of average users, we developed “γ-eye”, a mouse-sized, benchtop γ-camera suitable for in vivo scintigraphic imaging.
The γ-eye is based on two position-sensitive photomultiplier tubes, coupled to a CsI(Na) pixelated scintillator and a low-energy lead collimator with parallel hexagonal holes.
The spatial resolution of the system is 2 mm at 0 mm. The energy resolution is 26 % at 140 keV and the maximum recorded sensitivity 210 cps/MBq. The system was evaluated in a proof-of-concept animal study, using three different clinical Tc-99m-labeled radiopharmaceuticals. Phantom and animal studies demonstrate its ability to provide semiquantitative results even for short scans.
Systems’ performance, dimensions, and cost make γ-eye a unique solution for efficient whole-body mouse nuclear imaging.
This is a preview of subscription content, log in to check access.
Buy single article
Instant access to the full article PDF.
Price includes VAT for USA
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
This is the net price. Taxes to be calculated in checkout.
Peterson TE, Shokouhi S (2012) Advances in preclinical SPECT instrumentation. J Nucl Med 53:841–844
Gomes CM, Abrunhosa AJ, Ramos P, Pauwels EKJ (2011) Molecular imaging with SPECT as a tool for drug development. Adv Drug Deliv Rev 63:547–554
Balaban RS, Hampshire VA (2001) Challenges in small animal noninvasive imaging. ILAR J 42:248–262
Franc BL, Acton PD, Mari C, Hasegawa BH (2008) Small-animal SPECT and SPECT/CT: important tools for preclinical investigation. J Nucl Med 49:1651–1663
Peremans K, Cornelissen B, Van Den Bossche B et al (2005) A review of small animal imaging planar and pinhole SPECT γ camera imaging. Vet Radiol Ultrasound 46:162–170
Kagadis GC, Loudos G, Katsanos K et al (2010) In vivo small animal imaging: current status and future prospects. Med Phys 37:6421–6442
Stout DB, Zaidi H (2016) Preclinical multimodality imaging in vivo. PET Clin 3:251–273
Zanzonico P (2012) Principles of nuclear medicine imaging: planar, SPECT, PET, multi-modality, and auto radiography systems. Radiat Res 177:349–364
Zaidi H (2014) Molecular imaging of small animals: instrumentation and applications. Springer
Ntziachristos V, Turner G, Dunham J et al (2005) Planar fluorescence imaging using normalized data. J Biomed Opt. doi:10.1117/1.2136148
Jung JH, Choi Y, Hong KJ, Min BJ, Choi JY, Choe YS, Lee KH, Kim BT (2009) Development of a dual modality imaging system: a combined gamma camera and optical imager. Phys Med Biol 54:4547–4559
Meikle SR, Kench P, Kassiou M, Banati RB (2005) Small animal SPECT and its place in the matrix of molecular imaging technologies. Phys Med Biol 50:R45.5
Goorden MC, van der Have F, Kreuger R et al (2013) VECTor: a preclinical imaging system for simultaneous submillimeter SPECT and PET. J Nucl Med 54:306–312
Loudos G, Majewski S, Wojcik R et al (2007) Performance evaluation of a dedicated camera suitable for dynamic radiopharmaceuticals evaluation in small animals. IEEE Trans Nucl Sci 54:454–460
Xi W, Seidel J, Kakareka JW et al (2010) MONICA: a compact, portable dual gamma camera system for mouse whole-body imaging. Nucl Med Biol 37:245–253
Kim H, Furenlid LR, Crawford MJ et al (2006) SemiSPECT: a small-animal single-photon emission computed tomography (SPECT) imager based on eight cadmium zinc telluride (CZT) detector arrays. Med Phys 33:465–474
Miller BW, Barber HB, Barrett HH, et al (2012) Progress in BazookaSPECT: high-resolution, dynamic scintigraphy with large-area imagers. Proc SPIE-- Int Soc Opt Eng.
Metaxa A-F, Efthimiadou EK, Boukos N et al (2014) Hollow microspheres based on—folic acid modified—hydroxypropyl cellulose and synthetic multi-responsive bio-copolymer for targeted cancer therapy: controlled release of daunorubicin, in vitro and in vivo studies. J Colloid Interface Sci 435:171–181
Psimadas D, Oliveira H, Thevenot J et al (2014) Polymeric micelles and vesicles: biological behavior evaluation using radiolabeling techniques. Pharm Dev Technol 19:189–193
Zoppellaro G, Kolokithas-Ntoukas A, Polakova K et al (2014) Theranostics of Epitaxially condensed colloidal Nanocrystal clusters, through a soft biomineralization route. Chem Mater 26:2062–2074
Fragogeorgi EA, Savina IN, Tsotakos T et al (2014) Comparative in vitro stability and scintigraphic imaging for trafficking and tumor targeting of a directly and a novel 99mTc(I)(CO)3 labeled liposome. Int J Pharm 465:333–346
Psimadas D, Baldi G, Ravagli C et al (2014) Comparison of the magnetic, radiolabeling, hyperthermic and biodistribution properties of hybrid nanoparticles bearing CoFe2O4 and Fe3O4 metal cores. Nanotechnology 25:25101
Popov V (2004) Matrix output device readout system, US6747263B1.
Bhatia BS, Bugby SL, Lees JE, Perkins AC (2015) A scheme for assessing the performance characteristics of small field-of-view gamma cameras. Phys Med 31:98–103
Bugby SL, Lees JE, Bhatia BS, Perkins AC (2014) Characterisation of a high resolution small field of view portable gamma camera. Phys Med 30:331–339
Zhang Q, Lu Y, Yang K, Ren Q (2013) Position mapping and a uniformity correction method for small-animal SPECT based on connected regional recognition. Nucl Instrum Methods Phys Res Sect Accel Spectrometers Detect Assoc Equip 704:1–6
Jeong MH, Choi Y, Chung YH et al (2004) Performance improvement of small gamma camera using NaI(Tl) plate and position sensitive photo-multiplier tubes. Phys Med Biol 49:4961–4970
Knoll GF (2010) Radiation detection and measurement. Wiley
Nock BA, Maina T, Krenning EP, de Jong M (2014) “To serve and protect”: enzyme inhibitors as radiopeptide escorts promote tumor targeting. J Nucl Med 55:121–127
Audenhaege KV, Holen RV, Vandenberghe S et al (2015) Review of SPECT collimator selection, optimization, and fabrication for clinical and preclinical imaging. Med Phys 42:4796–4813
Miller BW, Moore JW, Barrett HH et al (2011) 3D printing in X-ray and gamma-ray imaging: a novel method for fabricating high-density imaging apertures. Nucl Instrum Methods Phys Res Sect Accel Spectrometers Detect Assoc Equip 659:262–268
Georgiou M, Loudos G, Stratos D et al (2012) Optimization of a gamma imaging probe for axillary sentinel lymph mapping. J Instrum 7:P09010
Kato T, Kataoka J, Nakamori T et al (2013) A novel gamma-ray detector with submillimeter resolutions using a monolithic MPPC array with pixelized Ce:LYSO and Ce:GGAG crystals. Nucl Instrum Methods Phys Res Sect Accel Spectrometers Detect Assoc Equip 699:235–241
Bouckaert C, Vandenberghe S, Van Holen R (2014) Evaluation of a compact, high-resolution SPECT detector based on digital silicon photomultipliers. Phys Med Biol 59:7521–7539
Psimadas D, Georgoulias P, Valotassiou V, Loudos G (2012) Molecular nanomedicine towards cancer: 111In-labeled nanoparticles. J Pharm Sci 101:2271–2280
Banerjee S, Pillai MRA, (Russ) Knapp FF (2015) Lutetium-177 therapeutic radiopharmaceuticals: linking chemistry, radiochemistry, and practical applications. Chem Rev 115:2934–2974
Tandon V, Gauthier N, Ruddy TD (2013) SPECT radiopharmaceuticals. In: Heller GV, Hendel RC (eds) Handb. Nucl. Cardiol. Springer, London, pp. 95–101
Banerjee SR, Pomper MG (2013) Clinical applications of gallium-68. Appl Radiat Isot Data Instrum Methods Use Agric Ind Med 0:2–13
Jan S, Benoit D, Becheva E et al (2011) GATE V6: a major enhancement of the GATE simulation platform enabling modelling of CT and radiotherapy. Phys Med Biol 56:881–901
Pizzonia J, Holmberg J, Orton S et al (2012) Multimodality animal rotation imaging system (Mars) for in vivo detection of intraperitoneal tumors. Am J Reprod Immunol N Y N 1989 67:84–90
Loudos GK, Papadimitroulas P, Zotos P et al (2010) Development and evaluation of QSPECT open-source software for the iterative reconstruction of SPECT images. Nucl Med Commun 31:558–566
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki Declaration and its later amendments or comparable ethical standards.
Human and Animal Rights
All applicable institutional and/or national guidelines for the care and use of animals were followed.
Conflict of Interest
The technology similar to the “γ-eye” presented in this manuscript will be commercialized in the near future by BET Solutions, Athens, Greece. At this point, γ-eye system is in prototype version. Eleftherios Fysikopoulos and Konstantinos Mikropoulos are currently coworkers of BET Solutions, George Loudos is consultant for BET Solutions, and Maria Georgiou is a general partner of BET Solutions.
About this article
Cite this article
Georgiou, M., Fysikopoulos, E., Mikropoulos, K. et al. Characterization of “γ-Eye”: a Low-Cost Benchtop Mouse-Sized Gamma Camera for Dynamic and Static Imaging Studies. Mol Imaging Biol 19, 398–407 (2017). https://doi.org/10.1007/s11307-016-1011-4
- Single photon emission
- Scintigraphic small animal imaging
- Position sensitive photomultiplier tube
- Performance evaluation