The Influence of Flow Rate on the Aerosol Deposition Profile and Electrostatic Charge of Single and Combination Metered Dose Inhalers
- 405 Downloads
The capability of the electrostatic next generation impactor (eNGI) has been investigated as a tool capable of measuring the electrostatic charge of single (Flixotide™; containing fluticasone propionate (FP)) and combination (Seretide™; FP and salmeterol xinafoate (SX)) pressurised metered dose inhalers (pMDIs) at different flow rates.
Aerosol mass distributions were investigated at 30, 60 and 90 l.min−1 and simultaneous charge measurements recorded.
Analysis of the mass distribution data indicated a flow dependent relationship, where the aerosol performance (aerodynamic diameter <5 μm) of FP significantly increased between 30 l.min−1 and 60 l.min−1 for both formulations. No significant increase in SX was observed for Seretide with increased flow rate. Analysis of the charge distribution indicated both formulations to primarily charge negatively with a concurrent increase in charge with increased flow rate. Interestingly, the charge-tomass ratio remained relatively constant between 30 l.min−1 and 60 l.min−1 and increased at 90 l.min−1, indicating that charging was majorly influenced at the highest flow rate.
This study has shown how the eNGI could be used as a simple Pharmacopeia based methodology for the evaluation of mass and charge profiles of single and combination pMDIs at a series of flow rates.
KEY WORDSelectrostatic NGI electrostatics eNGI pMDI flow rate inhalation
- 3.Chapter < 601 > (2008),. United States Pharmacopeia 31—National Formulary 26, : United States Pharmacopeial Convention Inc.; 2008.Google Scholar
- 4.Section 2.9.18—appendix XII C. Consistency of Formulated Preparations for inhalation. British Pharmacopoeia 2009.Google Scholar
- 14.Ali M, Mazumder MK, Martonen TB. Measurements of electrodynamic effects on the deposition of MDI and DPI aerosols in a replica cast of human oral-pharyngeal-laryngeal airways. J Aerosol Med Pulm D. 2009;22(1):35–44.Google Scholar
- 24.Kotian R, Peart J, Bryner J, Byron PR. Calibration of the modified electrical low-pressure impactor (ELPI) for use with pressurized pharmaceutical aerosols. J Aerosol Med Pulm D. 2009;22(1):55–65.Google Scholar
- 25.Electrical Low Pressure Impactor product specifications. Osuusmyllynkatu 13, FIN-33700 Tampere, Finland : Dekati Ltd.; 2009.Google Scholar
- 33.Hoe S, Traini D, Chan HK, Young PM. The electrical next generation impactor. Inhalation. 2009:in press.Google Scholar
- 34.Murnane D, Martin GP, Marriott C. Validation of a reverse-phase high performance liquid chromatographic method for concurrent assay of a weak base (salmeterol xinafoate) and a pharmacologically active steroid (fluticasone propionate). J Pharm Biomed Anal. 2006;40(5):1149–54.CrossRefPubMedGoogle Scholar
- 35.Young PM, Price R, Jones S, Billings M-P. Investigation into drug and excipient interaction in dry powder inhaler combination products. In: Dalby RN, Byron PR, Peart J, Suman JD, Farr SJ, editors. Respiratory drug delivery IX; California. USA: Davis Healthcare International Publishing; 2004. p. 749–51.Google Scholar
- 38.Bailey AG, Hashish AH, Williams TJ. Drug delivery by inhalation of charged particles. J Electrostat. 1997;44(1–2):3–10.Google Scholar