Influence of Air Flow on the Performance of a Dry Powder Inhaler Using Computational and Experimental Analyses
- 817 Downloads
The aims of the study are to analyze the influence of air flow on the overall performance of a dry powder inhaler (Aerolizer®) and to provide an initial quantification of the flow turbulence levels and particle impaction velocities that maximized the inhaler dispersion performance.
Computational fluid dynamics (CFD) analysis of the flowfield in the Aerolizer®, in conjunction with experimental dispersions of mannitol powder using a multistage liquid impinger, was used to determine how the inhaler dispersion performance varied as the device flow rate was increased.
Both the powder dispersion and throat deposition were increased with air flow. The capsule retention was decreased with flow, whereas the device retention first increased then decreased with flow. The optimal inhaler performance was found at 65 l min−1 showing a high fine particle fraction (FPF) of 63 wt.% with low throat deposition (9.0 wt.%) and capsule retention (4.3 wt.%). Computational fluid dynamics analysis showed that at the critical flow rate of 65 l min−1, the volume-averaged integral scale strain rate (ISSR) was 5,400 s−1, and the average particle impaction velocities were 12.7 and 19.0 m s−1 at the inhaler base and grid, respectively. Correlations between the device flow rate and (a) the amount of throat deposition and (b) the capsule emptying times were also developed.
The use of CFD has provided further insight into the effect of air flow on the performance of the Aerolizer®. The approach of using CFD coupled with powder dispersion is readily applicable to other dry powder inhalers (DPIs) to help better understand their performance optimization.
Key WordsCFD, computational fluid dynamics DPI dry powder aerosols inhalation drug delivery
This work is funded by a grant from the Australian Research Council. Matthew S. Coates is a recipient of an International Postgraduate Research Scholarship. The authors would like to thank Plastiape S.p.A. for the supply of the inhalers.
- 4.Borgström, L., Bisgaard, H., O’Callaghan, C., Pedersen, S. 2002Dry-powder inhalersBisgaard, H.O’Callaghan, C.Smaldone, and G. C. eds. Drug Delivery to the LungMarcel Dekker, Inc.New York421448Google Scholar
- 5.Boer, A. H., Bolhuis, G. K., Gjaltema, D., Hagedoorn, P. 1997Inhalation characteristics and their effects on in vitro drug delivery from dry powder inhalers. Part 3: the effect of flow increase rate (FIR) on the in vitro drug release from the Pulmicort 200 TurbuhalerInt. J. Pharm.1536777CrossRefGoogle Scholar
- 6.V. Chavan and R. Dalby. Novel system to investigate the effects of inhaled volume and rates of rise in simulated inspiratory air flow on fine particle output from a dry powder inhaler. AAPS Pharm. Sci. 4(2):Article 6 (2002).Google Scholar
- 7.ANSYS CFX (2003). www.ansys.com/cfx (accessed 08/01/04).Google Scholar
- 13.Menter, F. R. 1994Two-equation eddy-viscosity models for engineering applicationsAIAA J.32269289Google Scholar
- 16.P. G. Stecher, M. Windholz, and D. S. Leahy (eds.). The Merck Index: An Encyclopedia of Chemicals and Drugs, 8th edn., Merck & Co., Inc., Rahway, 1968.Google Scholar
- 20.British Pharmacopoeia, Appendix XII, Aerodynamic assessment of fine particles—Fine particle dose and particle size distribution, Apparatus C (2001).Google Scholar
- 21.Asking, L., Olsson, B. 1997Calibration at different flow rates of a multistage liquid impingerAerosol Sci. Tech.273949Google Scholar
- 22.Finlay, W. H. 2001The Mechanics of Inhaled Pharmaceutical AerosolsAcademic Press Inc. Ltd.LondonGoogle Scholar
- 30.Stahlhofen, W., Rudolf, G., James, A. C. 1989Intercomparison of experimental regional aerosol deposition dataJ. Aerosol Med.2285308Google Scholar
- 31.Finlay, W. H., Zhang, Y., Grgic, B., Heenan, A., Burnell, P., Matida, E. A., Pollard, A., Lange, C. F. 2004Solving a major in vitro–in vivo correlation problem: impactor induction portsDalby, R. N.Byron, P. R.Peart, J.Peart, J. D.Farr, S. J. eds. Respiratory Drug Delivery IXDavis Healthcare Int. PublishingRiver Grove, IL203209Google Scholar
- 33.Clark, A. R., Bailey, R. 1996Inspiratory flow profiles in disease and their effects on the delivery characteristics of dry powder inhalersDalby, R. N.Byron, P. R.Farr, S. J. eds. Respiratory Drug Delivery VInterpharm PressBuffalo Grove, IL221230Google Scholar