Repeatability and reproducibility of measurements of low dissolved radiocesium concentrations in freshwater using different pre-concentration methods
- 933 Downloads
We assessed the repeatability and reproducibility of methods for determining low dissolved radiocesium concentrations in freshwater in Fukushima. Twenty-one laboratories pre-concentrated three of 10 L samples by five different pre-concentration methods (Prussian-blue-impregnated filter cartridges, coprecipitation with ammonium phosphomolybdate, evaporation, solid-phase extraction disks, and ion-exchange resin columns), and activity of radiocesium was measured. The z-scores for all of the 137Cs results were within ± 2, indicating that the methods were a good degree of precision. The relative standard deviations (RSDs) indicating the reproducibility among different laboratories were larger than the RSDs indicating the repeatability in each separate laboratory.
KeywordsMeasurement precision Fukushima River water 137Cs 134Cs Ge-detector
It has been necessary to measure dissolved radiocesium (134Cs and 137Cs) concentrations in freshwater near the Fukushima Daiichi Nuclear Power Plant (FDNPP) since the FDNPP accident to provide information to allow root uptake by crops and accumulation in aquatic organisms to be assessed. Many institutions have therefore monitored dissolved radiocesium concentrations in freshwater in northeastern Japan since the FDNPP accident [1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12].
Dissolved and particulate-bound radiocesium concentrations in freshwater in the northeastern Japan are mostly < 0.1 Bq L−1, and dissolved radiocesium concentrations of 0.001–0.1 Bq L−1 have been found at most monitoring points in the northeastern Japan [9, 12, 13, 14, 15]. The low concentrations mean that effective quantitative measurements of dissolved radiocesium in freshwater require samples to be pre-concentrated before analysis. Environmental radioactivity measurement manuals published by the Japanese Ministry of Education, Culture, Sports, Science and Technology contain methods for measuring dissolved radiocesium in water involving either evaporating samples to dryness or ion exchange . Other methods involve ammonium phosphomolybdate (AMP) coprecipitation, particularly for very saline samples, and instrumental analysis using a germanium semiconductor detector (Ge-detector) . These methods have been used to quantitatively determine dissolved radiocesium in freshwater for a long time (since before the FDNPP accident). New pre-concentration methods were developed after the FDNPP accident, including using cartridges impregnated with Prussian blue (PB) [18, 19, 20, 21] and solid-phase extraction disks [22, 23], to decrease the time required for pre-concentration and solid–liquid separation. These new concentration methods dramatically decreased the time required for analyzing samples, so have been used widely.
The precision achieved when analyzing radiocesium using different methods has been assessed previously. Five methods (involving evaporation to dryness, AMP coprecipitation, PB cartridges, solid-phase disks, and ion exchange resin) were used to analyze three water samples (two ion-exchanged water samples spiked with radiocesium and one freshwater sample) containing dissolved radiocesium at concentrations of 0.01–1.0 Bq L−1 by 18 laboratories . The relative standard deviations (RSDs) were 0.08–0.119, and the z-scores for > 80% of the results were within ± 2. The results indicated that the pre-concentration methods (including the new methods) are good degree of precision. However, only one sample was analyzed repeatedly, so the results did not indicate whether the slight variations in the concentrations were caused predominantly by variability in the results from each laboratory or by variability between the results from different laboratories. The results also did not indicate whether variability between the results from different laboratories may have been caused by different pre-concentration methods being used by the different laboratories.
In the study presented here, we evaluated the variability in dissolved 134Cs and 137Cs concentrations in freshwater samples determined by each participating laboratory (repeatability) and variability in the concentrations determined by the different laboratories (reproducibility). Each participating laboratory analyzed three samples under the same conditions, then all the samples were analyzed in one laboratory to remove the effect of using different instruments at different laboratories on the reproducibility of the results. We assessed the repeatability and reproducibility and the factors affecting the reproducibility of the results from different laboratories.
Materials and methods
Water sample collection
River water was collected from the Abukuma Kogen region in Japan on 25 October 2017 under base flow condition. Most radiocesium in the sample was supplied by the FDNPP accident in March 2011. Approximately 1200 L of river water was collected in six 200 L tanks. The water was pumped from the river at the center of a bridge. The water was immediately transported to the laboratory, where it was passed through A045A 0.45 μm membrane filters (Advantec MFS, Dublin, CA, USA) on 26 October 2017. The filtered water in a 1200 L tank was stirred by passing it through a pump at a flow rate of 3 L s−1 for > 12 h. On 27 October 2017, 50 L of well-mixed water was removed using a washed plastic bucket, and three 10 L polyethylene flexible tanks were filled with the water. The flexible tanks were then refrigerated until they were sent to a participating laboratory. This was repeated for each participating laboratory. The samples were sent to the participating laboratories on 7 November 2017. Each partitioning laboratory received its allocation of three 10 L samples on 8 November 2017 and immediately stored the samples in a refrigerator. We indicated the procedure for sample storage keeping in a refrigerator but did not indicate the acidic treatment because of limiting condition of pre-treatment method. Notable procedures for sample storage for each laboratory are shown in Table S1.
The mean ± standard deviation Na+ concentrations, Ca2+ concentrations, pH values, and electrical conductivities for the 10 L samples were 3.5 ± 0.06 mg L−1, 2.3 ± 0.06 mg L−1, 7.2 ± 0.05, and 3.7 ± 0.05 mS m−1, respectively. We concluded that all the samples sent to the participating laboratories were well mixed because similar Na+ concentrations, Ca2+ concentrations, pH values, and electrical conductivities were found by the different laboratories.
Pre-concentration methods and radioactivity measurements
Number of laboratories using each pre-concentration method and the number of samples analyzed using each pre-concentration method
No. of laboratoriesa
No. of samples
Prussian blue filter cartridge
Ammonium phosphomolybdate coprecipitation
Solid-phase extraction disk
Ion exchange resin
The five pre-concentration methods are described below. The PB filter cartridge method takes advantage of PB (KFe[Fe(CN)6]3·xH2O) being able to adsorb dissolved radiocesium with a good degree of specificity [18, 19, 20, 21]. The PB filter cartridge method involved removing dissolved radiocesium from water passed through a nonwoven fabric impregnated with PB in a cartridge. A water sample was passed through a cartridge at a flow rate of 2.5 L min−1 by pumping, and 134Cs and 137Cs radioactivity in each cartridge was then measured using a Ge-detector with a special cartridge container. As distribution of radiocesium in the cartridge is not uniform, geometric correction was performed to evaluate concentration . Seven laboratories used potassium zinc ferrocyanide PB cartridges , and one laboratory used copper-substituted PB cartridges .
The AMP coprecipitation method takes advantage of the ability of AMP to adsorb dissolved radiocesium with a good degree of specificity . Each water sample was adjusted to ~ pH 1.5 before analysis. The radiocesium in the sample was quantitatively removed from the water sample by coprecipitation as AMP/Cs compounds once AMP had been added. The AMP/Cs compounds were removed by filtering the sample, then dried and transferred to a small container suitable for radioactivity measurement (e.g. 100 ml with 5-cm diameter plastic container) before the radioactivity was measured using a Ge-detector.
The evaporation method is to measure concentrated radiocesium in a water sample by evaporation . In the evaporation method in the present test, a water sample was evaporated in a glass beaker on a hot plate but not boiled excessively. Each sample was evaporated to a volume of between 50 mL and 2 L, then transferred to a 100 mL container or a 2 L Marinelli beaker before the radioactivity was measured using a Ge-detector. One laboratory acidified the samples before the evaporation method was performed.
In the solid-phase extraction disk method, dissolved radiocesium was removed from a water sample passed through a solid-phase extraction disk containing ion-selective resin on a Teflon support [26, 27]. A water sample was passed through a disk using a pressure filtration system, and the 134Cs and 137Cs radioactivity in the disk was measured using a Ge-detector.
In the ion exchange resin method, dissolved radiocesium is collected to a cation exchange resin from a water sample . The ion exchange resin method in the present test was performed by passing a water sample through a 100 mL column containing a cation exchange resin in the Na+ form. The resin containing the radiocesium was then transferred to a 100 mL container and the radioactivity was measured using a Ge-detector.
The more detailed information of each pre-concentration methods have been described previously [21, 22, 24, 28, 29]. The specific differences in procedures among laboratories are summarized in Table S1.
The laboratories pre-concentrated the samples between 8 and 25 November 2017 and determined the radioactivity of 134Cs and 137Cs in the samples by 10 January 2018. The radioactivity measurements were made in the laboratories by using their own Ge-detectors. Each laboratory followed its usual operating procedures for determination of low dissolved radiocesium concentrations in freshwater (e.g., sample storage procedure and measurement time). Radioactivity of 134Cs and 137Cs were decay corrected to 12:00 JST 11 November 2017 to allow the results to be compared. The different pre-concentration methods gave samples with different shapes, so each laboratory calibrated its detector(s) using a standard radioactive source (a special customized source was used for each sample of the same type). Finally, in order to discriminate effects of radioactivity measurement by different detectors from different procedures for sample treatment among laboratories, the samples were sent to one laboratory (Tokyo Power Technology Ltd., Chiba, Japan), where the 137Cs radioactivity in each sample was measured using a single Ge-detector (using a 43,000 s measurement period).
Results and discussion
Assessment of the precision of the results
The ranges and median repeatability RSDs for the participating laboratories are shown in bar charts in Fig. 4c, d. The repeatability for the laboratories can be compared from the figure. For all the results, the 134Cs repeatability RSDs were 0–0.226 (median 0.059), and the 137Cs repeatability RSDs were 0–0.070 (median 0.013). The repeatability RSDs were different for the different laboratories, suggesting that repeatability was different for different laboratories. Like for the concentrations, the repeatability RSDs for four of the pre-concentration methods did not show significantly statistically difference (one-way ANOVA, p < 0.005, the concentrations determined using the ion-exchange resin method was not included in the analysis because only one laboratory used for the determination). The repeatability RSDs for the laboratories using the AMP method (median 0.008 for 134Cs and median 0.001 for 137Cs) were smaller (but not statistically significantly smaller) than the repeatability RSDs for the laboratories using the other pre-concentration methods. This may have been because the laboratories using the AMP method have used it for a long time (since before the FDNPP accident), meaning they have been able to minimize variability between repeat analyses .
Reproducibility RSD and analyses of all the samples by one laboratory
The reproducibility RSDs for each pre-concentration method and for the results for all the pre-concentration methods combined are shown in solid square symbols in Fig. 4c, d. The reproducibility RSDs indicate variability between the results produced by the different laboratories. The reproducibility RSDs for all the 134Cs and 137Cs results were 0.108 and 0.088, respectively. The median 134Cs and 137Cs reproducibility RSDs were 0.059 and 0.013, respectively, indicating that variability between laboratories was generally larger than variability within a laboratory. The reproducibility RSDs for the different pre-concentration methods were not markedly different, but the reproducibility RSDs were lower for the AMP method than for the other methods. This indicated that the methods offered good reproducibility between laboratories as well as within each laboratory.
The results shown above do not describe the separate effects of different pre-concentration methods, Ge-detectors, and other factors on reproducibility. Comparisons of 137Cs concentration and RSD between from each laboratory and at one laboratory are shown in Figs. 3c and 4b, d. The robust average 137Cs concentration ± the standard deviation was 0.145 ± 0.011 Bq L−1 and the reproducibility RSD was 0.073. The average was not different from the averages for the different laboratories. The 137Cs concentrations for the PB cartridge, AMP coprecipitation, evaporation, and solid-phase disk methods were not statistically significantly different (one-way ANOVA, p < 0.005).
The robust averages and standard deviations for the 137Cs concentrations measured at a single laboratory are shown in Fig. 2c. The average concentrations were similar to the concentrations by each laboratory shown in Fig. 2b except for laboratory 2. The reproducibility RSDs measured at a single laboratory were almost the same as the reproducibility RSDs for the different laboratories (Fig. 3d). This suggests that the differences in the results from the different laboratories were not predominantly caused by the different pre-concentration methods and different Ge-detectors used but rather by other factors. The z-scores for > 90% of the 137Cs results were within ± 2, suggesting that the pre-concentration methods used by the laboratories all offered a good degree of precision.
The measurement precision with which low dissolved radiocesium concentrations in freshwater samples were determined by 21 Japanese laboratories was assessed. The repeatability and reproducibility of the results were assessed. All the pre-concentration methods used by the different laboratories gave a good degree of precision, indicating that even techniques used by Japanese laboratories to quantitatively determine low dissolved radiocesium concentrations only since the FDNPP accident in 2011 were high precision. Variability in the concentrations found by different laboratories was larger than variability in the concentrations found by each laboratory. Variability between laboratories was probably not caused by different pre-concentration methods or Ge-detectors being used but by other differences in the sample treatment procedures. This should be investigated further. Few evaluations of the measurement precision of methods used to determine low dissolved radiocesium concentrations in freshwater focused on repeatability and reproducibility have been performed. The results presented here therefore improve our understanding of the precision of such measurements.
This study was supported partly by JSPS KAKENHI Grant Number 16H01791, a grant-in-aid from the Kurita Water and Environment Foundation (No. 16K004). We thank Gareth Thomas, Ph.D., from Edanz Group (www.edanzediting.com/ac) for editing a draft of this manuscript.
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
- 1.Ikeuchi Y (2015). In: Result report of the project in FY2014 for collecting data and developing migration models of the radioactive material distribution caused by Tokyo Electric Power Company Fukushima Daiichi Nuclear Power Station accident, part 1. Japan Atomic Energy Agency (in Japanese) Google Scholar
- 10.Yoshimura K, Onda Y, Sakaguchi A, Yamamoto M, Matsuura Y (2015) An extensive study of the concentrations of particulate/dissolved radiocaesium derived from the Fukushima Dai-ichi Nuclear Power Plant accident in various river systems and their relationship with catchment inventory. J Environ Radioact 139:370–378CrossRefGoogle Scholar
- 11.Kameda Y, Yamaguchi H, Tamada M, Ohta S (2014) Establishment and evaluation of a novel rapid analysis method to measure trace levels of radioactive caesium in surface water by using absorbent disks for radioactive caesium with NaI scintillation counter (ADiCS). J Jpn Soc Water Environ 37:211–218 (in Japanese) CrossRefGoogle Scholar
- 16.Ministry of Education, Culture, Sports, Science and Technology (1976) Radioactive cesium analysis method. Radioactivity measurement method series 3 (in Japanese) Google Scholar
- 17.Ministry of Education, Culture, Sports, Science and Technology (1982) Sample pre-treatment methods for instrumental analysis using germanium semiconductor detectors etc. Radioactivity measurement method series 13 (in Japanese) Google Scholar
- 22.Yamaguchi H, Ota S, Tomita S (2012) Application of SPE-cesium disks to environmental monitoring of radiocesium in aqueous suspensions. In: SETAC Asia/Pacific conference 2012 proceedings. SETAC, Cleveland, AustraliaGoogle Scholar
- 23.Yamaguchi H, Ota S, Tsukada H (2014). Development of low-level-radiocesium concentration analysis system for irrigation water using solid phase extraction disks. In: 20th world congress of soil science proceedings, ICC, Jeju, KoreaGoogle Scholar
- 24.Yasutaka T, Shin M, Onda Y, Shinano T, Hayashi S, Tsukada Y, Aono T, Iijima K, Eguchi S, Ohno K, Yoshida Y, Kamihigashi H, Kitamura S, Kubota T, Nogawa N, Yoshikawa N, Yamaguchi H, Sueki K, Tsuji H, Miyazu S, Okada Y, Kurihara M, Tarjan S, Matsunami H, Uchida S (2017) Comparison of values analyzed by deferent methods for concentrating low-level radio cesium dissolved in precipitation: evaporation, AMP, solid phase extraction disk, PB filter cartridge, ion-exchange, resins, Japan Society for Analytical Chemistry, the Japan Society of Analytical Chemistry. Bunseki Kagaku 66:299–307 (in Japanese) CrossRefGoogle Scholar
- 25.Yasutaka T, Miyazu S, Kondo Y, Tsuji H, Arita K, Hayashi S, Takahashi A, Kawamoto T, Aoyama M (2016) Development of a copper-substituted, Prussian blue-impregnated, nonwoven cartridge filter to rapidly measure radiocesium concentration in seawater. J Nucl Sci Technol 53:1243–1250CrossRefGoogle Scholar
- 29.The technical manual of pre-treatment and analysis method of radiocesium in water for environmental monitoring, Review Committee of Technical Manual of Monitoring Methods of Radio Cesium in Water. https://unit.aist.go.jp/georesenv/information/20150917/TechRep_Cs_Monitoring_Water.pdf#search=%27. Accessed 16 Jul 2019 (in Japanese)
- 31.International Atomic Energy Agency (2013) IAEA analytical quality in nuclear applications series no. 29, emergency response proficiency test for Japanese laboratories: determination of selected radionuclides in water, soil, vegetation and aerosol filters. IAEA, ViennaGoogle Scholar
- 32.JIS Z 8405 (2018) Statistical methods for use in proficiency testing by interlaboratory comparisons. Japanese Industrial Standards, Tokyo (in Japanese) Google Scholar
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.