Fast determination of indoor radon (222Rn) concentration using liquid scintillation counting
- 662 Downloads
The indoor 222Rn radionuclide was directly absorbed in typical 20 ml glass scintillation vials by passing −3 dm3 of ambient air through 16 ml of water-immiscible non-volataile scintillation cocktail Ultima-Gold F for 10 min. The activity of radon and its two α-emitting daughters: 218Po and 214Po, was determined with the BetaScout low-background liquid scintillation counter. The limit of 222Rn detection is 9 Bq/m3, and the quantification limit with 20% relative accuracy is 28 Bq/m3. The results of the indoor Rn measurement in different houses showed good consistency with results obtained from a Sarad EQF 3220 device.
KeywordsIndoor radon Diurnal variations Fast measurements Liquid scintillation
Radon is known to present a risk of lung cancer when it, or rather its decay products, are inhaled . Most of the radon that enters indoor air comes directly from soil and this radionuclide can accumulate there to higher concentrations above 100 Bq/m3. For the practical implementation of a radon protection strategy, the International Commission on Radiological Protection (ICRP) recommended an upper value of the so called derived reference level of 300 Bq/m3 for 222Rn in dwellings . However, on the basis of new epidemiological findings, and consequently the new value of the radon dose conversion coefficient, indoor exposition to 300 Bq/m3 corresponds to a higher annual dose within the range of 15–20 mSv in homes . That range of an effective radiation dose exceeds the total average annual exposure of humans from remaining natural radiation sources by at least ten times, equal to −1.2 mSv. The recent evidences on the risks of very low-level radiation exposure seems to support the LNT (linear no-threshold) model and indicates harmful radiation effects well below 100 mSv. In particular, statistically sound epidemiological studies indicate adverse effects to people exposed to very low doses of −10 mSv, e.g., from medical CT (computer tomography) scans on infants [4, 5], to Chernobyl clean-up workers , and they even reveal adverse effects from background radiation to which all of us are exposed .
Therefore, the Commission strongly encourages national authorities to set a national derived radon reference level as low as reasonably achievable in the range of 100–300 Bq/m3 . A similar approach has been proposed in the European Union (EU) commission’s recommendations: under the national action plan, Member States shall promote action to identify dwellings with radon concentrations (as an annual average) exceeding the reference level .
A major source of uncertainty in radon risk assessment is the radon dose estimate. Methods for radon exposure measurement in homes are one of the factors that affect the risk estimates in a case–control study. Various methods exist to monitor 222Rn in the air. There are several commonly used types of detection, for example: alpha track, activated charcoal adsorption, or instrumental continuous radon monitors. However, there is a need for a fast, relatively simple and cheap method, which can be applied for a preliminary, large scale radon screening in houses with a quantification limit far below 100 Bq/m3. The passive diffusion activated charcoal canister PicoRad technique, commonly used for this purpose, requires detector exposition of at least 24 h, an 8-hour period where a scintillation cocktail is added and the radon is allowed to elute, and correction for temperature fluctuations in the examined rooms .
Among several procedures described in the literature for direct indoor Rn measurements, liquid scintillation plays a very important role. This technique, utilizing the high solubility of the gaseous Rn in aromatic solvents (common scintillation solvents), was introduced by Horrocks five decades ago  and different versions were published until the first decade of this century [12, 13, 14, 15, 16, 17, 18, 19]. Very recently, while preparing this paper, another work using mineral oil as a solvent for radon absorption and 218Po counting was published, but the proposed range of 222Rn determination above 500 Bq/m3 makes that method impractical for usual indoor radon measurements .
The radon partition coefficients between gaseous phase and typical scintillation solvents—o-xylene, toluene and hexane—at 20 °C are equal: 12.75, 13.24 and 16.56, respectively [21, 22]. Therefore, simple air bubbling until the concentration of 222Rn in the liquid scintillator reached a state of equilibrium in the ambient temperature is connected with the evaporation of part of the harmful solvent into the environment. The exposition of the open vial without bubbling can only be applied to high soil gas radon concentrations [13, 14] or with a special counter construction with very low background for α–particle detection [18, 19]. An alternative method, which depends on 10–20 l of air passing through 20 ml of scintillation solvent held at −78 °C in a bath of dry ice and acetone, gave satisfactory radon activities captured in scintillation vials, but its applications for routine large scale screening purposes are limited [16, 17].
In recent decades, a new class of liquid scintillator cocktails based on diisopropylnaphthalene, with very low vapour pressure and excellent detection efficiency, appeared on the market. In conjunction with a new generation of portable liquid scintillation counters, such a combination allows, for example, for simple radon extraction from 10 ml water samples into 10 ml of water-immiscible scintillation cocktail, directly in 20 ml vials and two-phase counting according to the Prichard and Gessler procedure . Using this scintillator and two phase liquid scintillation counting of α-pulses coming from 222Rn and two its daughters: 218Po and 214Po, in the transient radioactive equilibrium, we previously successfully applied that method for low level radon determination in different kinds of groundwater samples .
Based on the Cantaloub data, the estimated theoretical value of the gaseous radon partition coefficient between air and the Utima-Gold F diisopropylnaphthalene scintillator at 20 °C should be around 9, slightly lower than that for toluene . The aim of this study was to check the possibility of the direct absorption of indoor radon in this scintillator cocktail at ambient temperatures in 20 ml scintillation vials and to check the obtained quantification limits using a portable scintillation counter with α/β separation mode.
Materials and methods
Theoretical background of the method
Theoretical calculation of the calibration coefficient for the proposed method
The collected samples were transferred to the laboratory with a delay not exceeding one day. The samples were counted after a minimum 3-h waiting period to ensure the transient radioactive equilibrium between 222Rn and its daughters had been reached. The details of α-radioactivity counting by the BetaScout (PerkinElmer) device and its quality assurance checking is described elsewhere .
Radon fluctuations in two occupied rooms were measured by this method, with 4- to 6-h intervals, and the results were compared with those obtained by the calibrated EQF 3220 (SARAD GmbH) continuous radon monitor.
Results and discussion
The obtained values of K e for the elaborated method ranged from 2530 for 8 °C to 3770 (Bq/m3 cps) for 38 °C. Calculated from the slope of the linear relationship value of the molar enthalpy of radon solubility in Ultima Gold F ΔH s = 9.8 kJ/mol is close to values of ΔH s calculated for the other solvents, hexane and toluene, from the figures presented in Prichard’s paper .
Limits of detection—LD and determining LO20 for the method
The calculated values of the calibration coefficients allow us to determine the detection limit of the method using Currie’s expression . For a counting time of 3600 s and background in the α-region of the BetaScout device equal to 1.39 × 10−3 cps, LD = 9 Bq/m3 at 20 °C. For the same assumptions, the limit of determination with 20% relative accuracy, LO20 = 28 Bq/m3. Therefore, this method allows for fast and quite accurate radon concentration measurements in the majority of dwellings.
Application of the method for intercomparison measurements of radon in the Kowary mines
Comparison of Rn-222 concentration determinations by three methods
Betascout, Inet, cps
Ke (Bq/m3 × cps)
1507 ± 49
1410 ± 130
1316 ± 83
1277 ± 44
1190 ± 120
1172 ± 79
1065 ± 40
947 ± 110
815 ± 64
It is evident from the obtained differences in the radon concentrations are basically in the range of accuracy of simultaneous radon measuerements by two calibrated commercial devices: RAD7 and EQF3020. Relative discrepancies of these three methods used do not exceed the tolerable limit of 20%.
Examination of the daily variations of indoor radon concentrations in two buildings
Short-term radon concentration measurements are recommended for actual exposure assessment caused by diurnal variations of Rn concentrations in residential buildings. In the Central Europe region, outdoor and indoor radon concentrations are subject to high diurnal variations by a factor of as much as 10  with maximums in the early morning hours . This trend is usually caused by temperature inversion conditions in the ground level air layer, when, during the night, its temperature increases with the altitude up to several metres above the ground. These phenomenon additionally causes increase in negative pressure in the basement resulting in suction of the radon from adjacent soil. This process retards the vertical mixing ratio, and radon escaping from the soil surface is, in the early morning, trapped in the ground level layer and intensively diffuses into the houses. Therefore, the proposed method can be used for fast monitoring of actual radon concentrations over the whole day.
The first room was located in the corner of a bungalow-type house used as a hostel in Kalisz, a town in Central Poland, and the observed radon concentrations were in the range of 120–320 Bq/m3.
Room #2 was situated on the first floor of a two-storey residential building in Lodz. In this room, the radon concentrations fluctuated from a relatively low concentration −40 Bq/m3, to a maximal value of 180 Bq/m3. It is worth underlining that for both rooms, these two methods show similar values of the radon concentration for selected samples collection times, and the discrepancies between the corresponding results did not exceed 20%. Therefore, the elaborated liquid scintillation method proved its utility, even for low indoor concentrations −50 Bq/m3. Moreover, in both rooms, the maximal daily radon concentration occurred in the morning hours, in line with expectations, whereas lower values were observed in the late afternoon hours. On the basis of these and other observations [28, 29] for Polish houses, one can assume that a single measurement of radon concentration at morning may be sufficient for preliminary screening of the expected radon level in the house. These measurements were performed in the winter season, and for the evaluation of the annual doses from indoor radon exposure, similar measurements in spring, summer and autumn should be made.
The proposed method based on a simple 10–15 min bubbling of indoor air through 16 ml of water-immiscible scintillator with low vapour pressure in conjunction with a portable liquid scintillation counter with α/β separation option can be applied for large scale screening of indoor radon concentrations. The 222Rn nuclide quantification level LO20 of this method equal to 28 Bq/m3 makes this method extremely convenient for the preliminary selection of buildings with indoor concentrations exceeding the value of 100 Bq/m3 i.e., the expected radon reference levels accepted by the majority of EU countries. The method is very cheap and environmentally friendly, as the scintillation solution can be reused after a one-month.
- 1.World Health Organization (2009) WHO Handbook on indoor Radon: a public health perspective. World Health Organization, GenevaGoogle Scholar
- 9.European Commission (2014) Council Directive 2013/59/EURATOM of 5 December 2013 laying down basic safety standards for protection against the dangers arising from exposure to ionising radiation, and repealing Directives 89/618/Euratom, 90/641/Euratom, 96/29/Euratom, 97/43/Euratom a. doi: 10.3000/19770677.L_2014.013.eng
- 10.Bem H, Bem EM, Chriscielewski W, Skalski H (2000) Temperature calibration of pico-rad detectors for radon measurement. Int J Occup Med Environ Health 13:147–154Google Scholar
- 12.Reimschussel W, Bem H (1973) The application of liquid scintillation techniques for determining leakages in Ra-sources. Postep Fiz Med 8:15–22Google Scholar
- 18.Jonsson G, Theodorsson P, Sigurdsson K (2005) Auto-radon: a new automatic liquid scintillation system for monitoring radon in water and air. In: Chalupnik S, Schönhofer F, Noakes JE (eds) Liq. Scintill. Spectrom. 2005, Proc. Int. Conf. Adv. Liq. Scintill. Spectrom. Katowice, Poland, pp 119–124Google Scholar
- 19.Gudjonsson G, Theodórsson P, Sigurdsson K, et al (2008) A study of the simple open-vial liquid scintillation method for measurement of radon in air. In: Eikenberg J, Jäggi M, Beer H, Baehrle H (eds) Liq. Scintill. Spectrom. 2008, Int. Conf. Adv. Liq. Scintill. Spectrom. Davos, Switzerland, pp 361–366Google Scholar
- 21.Cantaloub MG (2000) Aqueous-organic partition coefficients for Rn-222 and their application to radon analysis by liquid scintillation methods. Oregon State University, CorvallisGoogle Scholar
- 24.Grabowski P, Dlugosz M, Szajerski P, Bem H (2010) A comparison of selected natural radionuclide concentrations in the thermal groundwater of Mszczonów and Cieplice with deep well water from Łódź city, Poland. Nukleonika 55:181–185Google Scholar
- 26.UNSCEAR (1993) UNSCEAR 1993 REPORT; SOURCES AND EFFECTS OF IONIZING RADIATION; United Nations Scientific Committee on the Effects of Atomic Radiation UNSCEAR 1993 Report to the General Assembly, with Scientific AnnexesGoogle Scholar
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.