Spatial distribution of neutron flux in geological larger sample analysis at CDTN/CNEN, Brazil
- 127 Downloads
Iron monitor wire was used to evaluate the neutron flux gradients inside 5 g-geological sample during irradiation. The spatial distribution for thermal flux was about 2 %/cm for radial and axial gradients and about 5 %/cm for radial and 2.5 %/cm for axial gradients for fast neutrons. The results were evaluated using reference material IAEA-SOIL-7 analysed as small and larger samples and tested with calculating E n numbers. This test pointed out that all results were consistent within a 95 % confidence interval. The contribution of the neutron flux gradients may be reduced for larger sample using the standard procedure of the k 0-instrumental neutron activation analysis.
KeywordsNeutron activation analysis k0-instrumental neutron activation analysis Larger sample Neutron flux gradients
This work was partially supported by the International Atomic Energy Agency under Grant BRA-14798, by the Foundation for Research Support of Minas Gerais, FAPEMIG, under Grant APQ-01259-09 and by financial support from the Slovenian Research Agency (ARRS) through programme P1-0143.
- 5.Tzika F, Stamatelatos IE, Kalef-Ezra J, Bode P (2004) Large sample neutron activation analysis: correction for neutron and gamma attenuation. Nukleonika 49:115–121Google Scholar
- 10.Overwater RMW (1994) The physics of big sample instrumental neutron activation analysis, PhD Thesis, TUDelft, DelftGoogle Scholar
- 18.HyperLab (2009) Gamma Spectroscopy Software, HyperLabs Software, Budapest 1998–2013. http://hlabsoft.com/. Accessed 9 June 2013
- 19.Kayzero for Windows (2011) User’s Manual, for reactor neutron activation analysis (NAA) using the k 0 standardisation method, Ver. 2.42. k 0-ware. Heerlen, The NetherlandsGoogle Scholar
- 20.International Standard (2005) Statistical methods for use in proficiency testing by interlaboratory comparisons ISO 13528:2005Google Scholar
- 21.International Atomic Energy Agency (2000) Reference material IAEA-SOIL-7, trace elements in soil. IAEA, ViennaGoogle Scholar
- 22.De Corte F (1987) The k 0 standardisation method: a move to the optimisation of neutron activation analysis. Ryksuniversiteit Ghent, Faculteit Van de Wetenschappen, GentGoogle Scholar