Rotaphone, a mechanical seismic sensor system for field rotation rate measurements and its in situ calibration
- 219 Downloads
A newly developed mechanical sensor system, called a Rotaphone, for recording rotation rate components is described. The sensor system is based on measurements of the differential motions between paired low-frequency geophones attached to a rigid skeleton. The same differential velocity (and, consequently, the same rotation rate component) is obtained from more than one geophone pair, which allows for in situ calibration and matching of the individual sensors. The calibration method, which is a key point of our methodology, is explained in detail and demonstrated on synthetic examples. The instrument and the calibration technique were also subjected to laboratory testing utilizing a rotational shaking table. Some results of these tests are presented in comparison to the data from a reference sensor (fiber optic gyroscope). The new rotational seismic sensor system is characterized by a flat frequency response over a wide range from 2 to 200 Hz and sensitivity limit of the order of 10 − 8 rad/s. Its advantages are small size, portability, and easy installation and operation in the field. We present several examples of the vertical rotation rate ground motion recorded at Nový Kostel station (Czech Republic): first, a group of records of local micro-earthquakes which occurred in West Bohemia/Vogtland in May 2010 and, second, a record due to an anthropogenic source, a quarry blast at the nearby Vintířov quarry recorded in June 2010. The measured rotation rates are of order between 10 − 6 for the local earthquakes and 10 − 7 rad/s for the blast. These measurements demonstrate that the instrument has much wider application than just prospecting measurements, for which it was originally designed.
KeywordsSeismic rotation Rotational seismometer Inverse problem West Bohemia/Vogtland region
The authors would like to thank John R. Evans and Charles R. Hutt for experimental assistance during testing and for providing reference fiber optic gyro data. This work was supported by the Czech Science Foundation, project no. P210/10/0925. Field measurements were carried out at the WEBNET station supported by the CzechGeo/EPOS project.
- Brokešová J, Málek J (2010) New portable sensor system for rotational seismic motion measurements. Rev Sci Instrum 81. doi: 10.1063/1.3463271
- Brokešová J, Málek J, Štrunc J (2009a) Generator of rotational seismic waves, rotational seismic sensor system and seismic measuring set. Patent CZ 301218Google Scholar
- Brokešová J, Málek J, Štrunc J (2009b) Rotational seismic sensor system, generator of rotational seismic waves and seismic measuring set. Patent CZ 301217Google Scholar
- Igel H, Schreiber U, Flaws A, Schuberth B, Velikoseltsev A, Cochard A. (2005) Rotational motions induced by the M8.1 Tokachi-oki earthquake, September 25, 2003. Geophys Res Lett 32. doi: 10.1029/2004GL022336
- Nigbor RL (1994) Six-degree-of freedom ground-motion measurement. Bull Seismic Soc Am 84:1665–1669Google Scholar
- Schreiber U, Klügel T, Stedman GE (2003) Earth tide and tilt detection by a ring laser gyroscope. J Geophys Res 108. doi: 10.1029/2001JB000569
- Schreiber U, Velikoseltsev A, Rorthacher M, Klügel T, Stedman GE, Wiltshire DL (2004) Direct measurement of diurnal polar motion by ring laser gyroscopes. J Geophys Res 109. doi: 10.1029/2003JB002803
- Tarantola A (1994) Inverse problem theory, methods for data fitting and model parameter estimation. Elsevier, AmsterdamGoogle Scholar
- Teisseyre R, Takeo M, Majewski E (eds) (2006) Earthquake source asymmetry, structural media and rotation effects. Springer, BerlinGoogle Scholar