Quartz Tuning Fork: Thermometer, Pressure- and Viscometer for Helium Liquids
- First Online:
- 654 Downloads
Commercial quartz oscillators of the tuning-fork type with a resonant frequency of ∼ 32 kHz have been investigated in helium liquids. The oscillators are found to have at best Q values in the range 105–106, when measured in vacuum below 1.5 K. However, the variability is large and for very low temperature operation the sensor has to be preselected. We explore their properties in the regime of linear viscous hydrodynamic response in normal and superfluid 3He and 4He, by comparing measurements to the hydrodynamic model of the sensor.
PACS Numbers47.80.+v 67.90.+z 85.50.−n.
Unable to display preview. Download preview PDF.
- 1.Eernisse E.P., Ward R.W., and Wiggins R.B., IEEE Trans. Ultrason. Ferroelectr. Freq. Control 35, 323 (1988) – a survey of quartz resonator sensor technologies.Google Scholar
- 4.M. Niemetz, H. Kerscher, and W. Schoepe, J. Low Temp. Phys. 126, 287 (2002). [W. Schoepe, Phys. Rev. Lett. 92, 095301 (2004)]–electrically driven magnetically levitated oscillating sphere in 4He-II.Google Scholar
- 5.H. A. Nichol, L. Skrbek, P. C. Hendry, and P. V. E. McClintock, Phys. Rev. Lett. 92, 244501 (2004). [Phys. Rev. E 70, 056307 (2004)]–electrically driven oscillating grid in 4He-II.Google Scholar
- 6.Yano H., Handa A., Nakagawa H., Obara K., Ishikawa O., Hata T., Nakagawa M., J. Low Temp. Phys. 138, 561 (2005)–vibrating wire resonator in 4He-II.Google Scholar
- 7.D. I. Bradley, D. O. Clubb, S. N. Fisher, A. M. Guénault, R. P. Haley, C. J. Matthews, G. R. Pickett, V. Tsepelin, and K. Zaki, Phys. Rev. Lett. 95, 035302 (2005). [ibid. 96, 035301 (2006)].Google Scholar
- 9.J. Martikainen, J. Tuoriniemi, T. Knuuttila, and G. Pickett, J. Low Temp. Phys. 126, 139 (2002)–vibrating wire resonator in 3He – 4He solution.Google Scholar
- 10.K. Karrai and R. D. Grober, Tip-sample distance control for near-field scanning optical microscopes, in Near-Field Optics, M. A. Paesler and P. T. Moyer (ed.), Proc. SPIE 2535, 69 (1995).Google Scholar
- 11.K. Karrai, lecture notes (2000) at http://www.nano.physik.uni-muenchen.de/publikationen/Preprints/p-00-03_Karrai.pdfGoogle Scholar
- 13.L. M. Milne-Thomson, Theoretical hydrodynamics, Dover Publications, New York (1996); H. Lamb, Hydrodynamics, Dover Publications, New York (1945).Google Scholar
- 19.V. D. Arp and R. C. McCarty, The Properties of Critical Helium Gas Technical Report, University Oregon (1998); R. D. McCarty, Thermophysical Properties of Helium-4 from 2 to 1500 K with Pressures to 1000 atm, Technical Note 631, National Bureau of Standards, Gaithersburg, Maryland (1972).Google Scholar
- 20.Frank Pobell, (1996). Matter and Methods at Low Temperatures. Springer, Berlin, 2nd edGoogle Scholar
- 23.The fork used in the measurements at 29 bar pressure was manufactured in Japan and was obtained from RS Company under stock number 472-1161, brand Fox Electronics and manufacturer’s part number NC26. The fork used in the zero pressure measurements was manufactured in Taiwan and was obtained from RS Company under stock number 226–1437, brand C-MAC Frequency Products and manufacturer’s part number A103C.Google Scholar
- 24.A. I. Ahonen, M. Krusius, and M. A. Paalanen, J. Low Temp. Phys. 25, 421 (1976); P. J. Hakonen, M. Krusius, M. M. Salomaa, R. H. Salmelin, J. T. Simola, A. D. Gongadze, G. E. Vachnadze, and G. A. Kharadze, J. Low Temp. Phys. 76, 225 (1989).Google Scholar
- 26.Dobbs E.R., (2000). Helium Three. Oxford University Press, Oxford, UKGoogle Scholar
- 28.In setup 1 the same type of fork was used as in 3He measurements at 29 bar pressure.23 In setup 2 the fork was made in China by Fronter Electronics, product number DT26.Google Scholar
- 30.Donnelly R.J., (1991). Quantized vortices in Helium II. Cambridge University Press, CambridgeGoogle Scholar