A brief airway occlusion is sufficient to measure the patient’s inspiratory effort/electrical activity of the diaphragm index (PEI)
Pressure generated by patient’s inspiratory muscles (Pmus) during assisted mechanical ventilation is of significant relevance. However, Pmus is not commonly measured since an esophageal balloon catheter is required. We have previously shown that Pmus can be estimated by measuring the electrical activity of the diaphragm (EAdi) through the Pmus/EAdi index (PEI). We investigated whether PEI could be reliably measured by a brief end-expiratory occlusion maneuver to propose an automated PEI measurement performed by the ventilator. Pmus, EAdi, airway pressure (Paw), and flow waveforms of 12 critically ill patients undergoing assisted mechanical ventilation were recorded. Repeated end-expiratory occlusion maneuvers were performed. PEI was measured at 100 ms (PEI0.1) and 200 ms (PEI0.2) from the start of the occlusion and compared to the PEI measured at the maximum Paw deflection (PEIoccl, reference). PEI0.1 and PEI0.2 tightly correlated with PEIoccl, (p < 0.001, R2 = 0.843 and 0.847). At a patient-level analysis, the highest percentage error was -64% and 50% for PEI0.1 and PEI0.2, respectively, suggesting that PEI0.2 might be a more reliable measurement. After correcting the error bias, the PEI0.2 percentage error was lower than ± 30% in all but one subjects (range − 39 to + 29%). It is possible to calculate PEI over a brief airway occlusion of 200 ms at inspiratory onset without the need for a full patient's inspiratory effort. Automated and repeated brief airway occlusions performed by the ventilator can provide a real time measurement of PEI; combining the automatically measured PEI with the EAdi trace could be used to continuously display the Pmus waveform at the bedside without the need of an esophageal balloon catheter.
KeywordsPressure generated by inspiratory muscles over electrical activity of the diaphragm index Patient spontaneous breathing effort Diaphragm neuromuscular efficiency Patient inspiratory effort
Departmental funds, University of Milan-Bicocca.
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
Informed consent was obtained from all individual participants included in the study, according to the Institutional Ethical Committee recommendations.
Research involving human participants and/or animals
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards.
- 3.Bellani G, Grasselli G, Teggia-Droghi M, Mauri T, Coppadoro A, Brochard L, Pesenti A. Do spontaneous and mechanical breathing have similar effects on average transpulmonary and alveolar pressure? A clinical crossover study. Crit Care. 2016;20(1):142. https://doi.org/10.1186/s13054-016-1290-9.CrossRefPubMedPubMedCentralGoogle Scholar
- 4.Yoshida T, Torsani V, Gomes S, De Santis RR, Beraldo MA, Costa EL, Tucci MR, Zin WA, Kavanagh BP, Amato MB. Spontaneous effort causes occult pendelluft during mechanical ventilation. Am J Respir Crit Care Med. 2013;188(12):1420–7. https://doi.org/10.1164/rccm.201303-0539OC.CrossRefPubMedGoogle Scholar
- 5.ATS/ERS Statement on respiratory muscle testing (2002). Am J Respir Crit Care Med 166 (4):518–624. https://doi.org/10.1164/rccm.166.4.518
- 6.Akoumianaki E, Maggiore SM, Valenza F, Bellani G, Jubran A, Loring SH, Pelosi P, Talmor D, Grasso S, Chiumello D, Guerin C, Patroniti N, Ranieri VM, Gattinoni L, Nava S, Terragni PP, Pesenti A, Tobin M, Mancebo J, Brochard L. The application of esophageal pressure measurement in patients with respiratory failure. Am J Respir Crit Care Med. 2014;189(5):520–31. https://doi.org/10.1164/rccm.201312-2193CI.CrossRefPubMedGoogle Scholar
- 11.Bellani G, Coppadoro A, Pozzi M, Bronco A, Albiero D, Eronia N, Meroni V, Grasselli G, Pesenti A. The Ratio of Inspiratory Pressure Over Electrical Activity of the Diaphragm Remains Stable During ICU Stay and is not Related to Clinical Outcome. Respir Care. 2016;61(4):495–501. https://doi.org/10.4187/respcare.04400.CrossRefPubMedGoogle Scholar
- 16.Jansen D, Jonkman AH, Roesthuis L, Gadgil S, van der Hoeven JG, Scheffer GJ, Girbes A, Doorduin J, Sinderby CS, Heunks LMA. Estimation of the diaphragm neuromuscular efficiency index in mechanically ventilated critically ill patients. Crit Care. 2018;22(1):238. https://doi.org/10.1186/s13054-018-2172-0.CrossRefPubMedPubMedCentralGoogle Scholar
- 17.Bellani G, Coppadoro A, Patroniti N, Turella M, Arrigoni Marocco S, Grasselli G, Mauri T, Pesenti A. Clinical assessment of auto-positive end-expiratory pressure by diaphragmatic electrical activity during pressure support and neurally adjusted ventilatory assist. Anesthesiology. 2014;121(3):563–71. https://doi.org/10.1097/aln.0000000000000371.CrossRefPubMedGoogle Scholar