Applying the ISO/IEEE 11073 Standards to Wearable Home Health Monitoring Systems
Objective. The goal of this effort was to investigate the feasibility of applying the ISO/IEEE 11073 (a.k.a. X73) standards, originally intended for bedside monitoring in hospital environments, to wearable, multi-sensor monitoring systems designed for home healthcare. Methods. The X73 upper-layer sub-standards (i.e., nomenclature specification, domain information model, application profiles, and vital sign device descriptions) were adopted and implemented on microcontroller-based sensor hardware to provide plug-and-play medical components. Three types of system elements (base stations, data loggers, and sensor units) perform the functionality required in this standards-based home health monitoring system and communicate using Bluetooth wireless modules. The base station incorporates a LabVIEW interface running on a personal computer. Each data logger and sensor unit is implemented on a microcontroller-driven embedded platform. Sensor units include wearable sensors (e.g., electrocardiograph, pulse oximeter) and nearby sensors (e.g., weight scale, ambient environment sensors). Results. The standards-based prototype system with an open architecture achieves plug-and-play performance suitable for a home environment. Each wireless element in the body/home area network can automatically detect other nearby devices, associate with them, and exchange data with them as appropriate. Conclusions. With minor modifications, the X73 standards can be successfully applied to wearable, wireless, point-of-care systems in the home.
Key WordsBluetooth home health care ISO/IEEE 11073 standards plug-and-play interoperability wearable embedded platform wireless X73
Unable to display preview. Download preview PDF.
- 3.ISO/IEEE 11073 Committee. Available at: http://www.ieee1073.org.
- 4.Health Informatics – Point-of-Care Medical Device Communication – Application Profiles – Base Standard: ISO/IEEE 11073 Committee; 2004.Google Scholar
- 5.Standard for Medical Device Communications – Transport Profile – Connection Mode: ISO/IEEE 11073 Committee; 1994.Google Scholar
- 6.Health Informatics – Point-of-Care Medical Device Communication – Device Specialization – ECG Component.: ISO/IEEE 11073 Committee; 1998.Google Scholar
- 7.IEEE Standard for Medical Device Communication – Overview and Framework: ISO/IEEE 11073 Committee; 1996.Google Scholar
- 8.Health Informatics – Point-of-Care Medical Device Communication – Device Specialization – Pulse Oximeter: ISO/IEEE 11073 Committee; 1999.Google Scholar
- 9.IEEE Standard for Medical Device Communications – Transport Profile – IrDA Based – Cable Connected: ISO/IEEE 11073 Committee; 2004.Google Scholar
- 10.IEEE Standard for Medical Device Communications – Physical Layer Interface – Cable Connected: ISO/IEEE 11073 Committee; 2000 (Withdrawn).Google Scholar
- 11.IEEE Standard for Health Informatics – Point-of-Care Medical Device Communication – Nomenclature: ISO/IEEE 11073 Committee; 2002.Google Scholar
- 12.IEEE Standard for Medical Device Communications – Transport Profile-IrDA Based – Infrared Wireless: ISO/IEEE 11073 Committee; 2004.Google Scholar
- 13.IEEE Standard for Health Informatics – Point-of-Care Medical Device Communications – Domain Information Model: ISO/IEEE 11073 Committee; 2005.Google Scholar
- 14.Cooper T. ISO/IEEE 11073 Standards for Medical Device Communication. Paper presented at: 2nd Joint EMBS-BMES Conference, 2002; Houston, TX.Google Scholar
- 15.Wittenber J. Standard Medical Device Data Language (MDDL): Architecture and Applications. Paper presented at: IEEE International Conference on Systems, Man and Cybernetics, 1989; New York, NY.Google Scholar
- 17.Kennelly RJ, Wittenber J. New IEEE Standard Enables Data Collection for Medical Applications. Paper presented at: Proceedings of the 18th Annual Symposium on Computer Applications in Medical Care, 1994; Washington, DC.Google Scholar
- 19.Schrenker R, Cooper T. Building the Foundation for Medical Device Plug-and-Play Interoperability. Medical Electronics Manufacturing; April 2001:10+.Google Scholar
- 20.IEEE Approves Four Point-of-Care Medical Device Communication Standards. IEEE Engineering in Medicine and Biology Society. Available at: http://www.eng.unsw.edu.au/embs/news.cfm.
- 21.Cooper T. BER S/W Update. In: Warren S, Yao J, eds; 2002.Google Scholar
- 22.The Official Bluetooth Membership Site. Bluetooth Special Interest Group. Available at: http://www.bluetooth.org.
- 23.Nutt GJ. Open Systems: Prentice Hall; 1992.Google Scholar
- 24.Jain BN, Agrawala AK. Open Systems Interconnection – Its Architecture and Protocols: McGraw-Hill, Inc.; 1993.Google Scholar
- 25.Infrared Data Association. Available at: http://www.irda.org/.
- 26.Yao J, Schmitz R, Warren S. A Wearable Standards-Based Point-of-Care System for Home Use. Paper presented at: 3rd Joint EMBS-BMES Conference; September 17–21, 2003; Cancun, Mexico.Google Scholar
- 27.IEEE 802.11. Available at: http://grouper.ieee.org/groups/802/11/.
- 28.ZigBee – Wireless Control That Simply Works. ZigBee Alliance. Available at: http://www.zigbee.org/.
- 29.Health Informatics – Point-of-Care Medical Device Communication – Technical Report – Guidelines for the Use of RF Wireless Technologies. ISO/IEEE 11073 Committee. Available at: http://ieee1073.org/standards/11073-00101/11073-00101.html.