Evaluation of two novel aluminum-free, zinc-based glass polyalkenoate cements as alternatives to PMMA bone cement for use in vertebroplasty and balloon kyphoplasty
- 398 Downloads
Vertebroplasty (VP) and balloon kyphoplasty (BKP) are now widely used for treating patients in whom the pain due to vertebral compression fractures is severe and has proved to be refractory to conservative treatment. These procedures involve percutaneous delivery of a bolus of an injectable bone cement either directly to the fractured vertebral body, VB (VP) or to a void created in it by an inflatable bone tamp (BKP). Thus, the cement is a vital component of both procedures. In the vast majority of VPs and BKPs, a poly(methyl methacrylate) (PMMA) bone cement is used. This material has many shortcomings, notably lack of bioactivity and very limited resorbability. Thus, there is room for alternative cements. We report here on two variants of a novel, bioactive, Al-free, Zn-based glass polyalkenoate cement (Zn-GPC), and how their properties compare to those of an injectable PMMA bone cement (SIMPL) that is widely used in VP and BKP. The properties determined were injectability, radiopacity, uniaxial compressive strength, and biaxial flexural modulus. In addition, we compared the compression fatigue lives of a validated synthetic osteoporotic VB model (a polyurethane foam cube with an 8 mm-diameter through-thickness cylindrical hole), at 0–2300 N and 3 Hz, when the hole was filled with each of the three cements. A critical review of the results suggests that the performance of each of the Zn-GPCs is comparable to that of SIMPL; thus, the former cements merit further study with a view to being alternatives to an injectable PMMA cement for use in VP and BKP.
KeywordsUniaxial Compressive Strength Bone Cement Fracture Vertebral Body Balloon Kyphoplasty Augmentation Model
The authors thank Mr. Yuan Li and Dr. Si Janna, for their contributions to the compression cyclic fatigue life testing and the financial assistance of Enterprise Ireland for facilitating this work.
- 9.Becker S, Franz K, Wilke I, Ogon M. Injectable bone cements in the treatment of spinal fractures, osteopromotive capacity and surgical considerations. Eur Cells Mater. 2006;11 Suppl 1:1.Google Scholar
- 20.ISO 4049:2000. Dentistry—polymer-based filling, restorative and luting materials. Geneva, Switzerland: International Organization for Standardization (ISO); 2000.Google Scholar
- 21.ISO 9917-1:2003. Dental water based cements. Geneva, Switzerland: International Organization for Standardization; 2003.Google Scholar
- 33.Schwardt J, Slater T, Lee S, Meyer J, Wenz R. KyphOs FSTM calcium phosphate for balloon kyphoplasty: verification of compressive strength and instructions for use. Eur Cells Mater. 2006;11 Suppl 1:28.Google Scholar
- 36.Arlot ME, Roux JP, Boivin G, Perrat B, Tsouderos Y, Delofire P. Effects of a strontium salt (S12911) on both tibial metaphysis and epiphysis in normal growing rats. J Bone Mineral Res. 1995;10:S356.Google Scholar
- 39.Alvarez L, Perez-Higueras A. Percutaneous vertebroplasty in the treatment of osteoporotic fractures. In: Lewandrowski K-U, Wise DL, Trantolo DJ, Yaszemski MJ, White III AA, editors. Advances in spinal fusion: molecular science, biomechanics, and clinical management. New York: Marcel Dekker Inc.; 2004. p. 13–21.Google Scholar
- 40.Lewis G. Putting insoles through the paces. Mech Eng. 1989;111(4):96–8.Google Scholar