Features of the choriocapillaris on four different optical coherence tomography angiography devices
- 25 Downloads
To investigate the features of the choriocapillaris using four different optical coherence tomography angiography (OCTA) devices.
OCTA images of the choriocapillaris from consecutive healthy subjects were obtained with four different OCTA devices (Zeiss PLEX Elite, Topcon DRI OCT-1 Atlantis, Zeiss AngioPlex, and Heidelberg Spectralis OCTA). The 3 × 3 mm OCTA images were processed with ImageJ. The mean vascular density and mean flow void area of the choriocapillaris were compared among devices. Flow voids were analyzed with two different imaging adjustment methods, auto-local threshold with the Phansalkar method and a method using a device-specific threshold value.
The mean vascular density of the choriocapillaris differed among the four devices (all P < 0.001). The mean flow void area as measured with the auto-local threshold method also differed among devices (P < 0.001) and was not correlated among devices (all P > 0.05). Results for mean flow void area measured with a device-specific threshold value using the Plex-Elite and DRI OCT-1 Atlantis were correlated (ß = 2.271, P < 0.001), but there were no correlations among other devices (P > 0.05). For the Plex-Elite and DRI OCT-1 Atlantis, the mean flow void area was positively correlated between the two image adjustment methods (all P < 0.05).
Vascular densities and flow void areas of the choriocapillaris varied according to the device used and the image adjustment method. The characteristics of different devices and the image adjustment method should be considered for analysis of the choriocapillaris.
KeywordsChoriocapillaris Flow voids Optical coherence tomography angiography Vascular density
This study was funded by the National Research Foundation of Korea (Ministry of Science, ICT and Future Planning of the Korean government) (NRF-2017R1C1B5076520).
Compliance with ethical standards
Conflict of interest
Jaeryung Oh is a consultant of Topcon corporation. Other authors declare that they have no conflict of interest.
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards.
Informed consent was obtained from all individual participants included in the study.
- 9.Corvi F, Pellegrini M, Erba S, Cozzi M, Staurenghi G, Giani A (2018) Reproducibility of vessel density, fractal dimension, and foveal avascular zone using 7 different optical coherence tomography angiography devices. Am J Ophthalmol 186:25–31. https://doi.org/10.1016/j.ajo.2017.11.011 CrossRefPubMedGoogle Scholar
- 23.Novais EA, Adhi M, Moult EM et al (2016) Choroidal neovascularization analyzed on ultrahigh-speed swept-source optical coherence tomography angiography compared to spectral-domain optical coherence tomography angiography. Am J Ophthalmol 164:80–88. https://doi.org/10.1016/j.ajo.2016.01.011 CrossRefPubMedPubMedCentralGoogle Scholar
- 26.Choi W, Waheed NK, Moult EM et al (2017) Ultrahigh speed swept source optical coherence tomography angiography of retinal and choriocapillaris alterations in diabetic patients with and without retinopathy. Retina 37:11–21. https://doi.org/10.1097/IAE.0000000000001250 CrossRefPubMedPubMedCentralGoogle Scholar
- 27.Choi W, Moult EM, Waheed NK et al (2015) Ultrahigh-speed, swept-source optical coherence tomography angiography in nonexudative age-related macular degeneration with geographic atrophy. Ophthalmology 122:2532–2544. https://doi.org/10.1016/j.ophtha.2015.08.029 CrossRefPubMedPubMedCentralGoogle Scholar