Herschel SPIRE FTS spectral mapping calibration
- 162 Downloads
The Herschel SPIRE Fourier transform spectrometer (FTS) performs spectral imaging in the 447–1546 GHz band. It can observe in three spatial sampling modes: sparse mode, with a single pointing on sky, or intermediate or full modes with 1 and 1/2 beam spacing, respectively. In this paper, we investigate the uncertainty and repeatability for fully sampled FTS mapping observations. The repeatability is characterised using nine observations of the Orion Bar. Metrics are derived based on the ratio of the measured intensity in each observation compared to that in the combined spectral cube from all observations. The mean relative deviation is determined to be within 2 %, and the pixel-by-pixel scatter is ∼ 7 %. The scatter increases towards the edges of the maps. The uncertainty in the frequency scale is also studied, and the spread in the line centre velocity across the maps is found to be ∼ 15 km s − 1. Other causes of uncertainty are also discussed including the effect of pointing and the additive uncertainty in the continuum.
KeywordsSPIRE FTS Calibration Repeatability Error
Herschel is an ESA space observatory with science instruments provided by European-led Principal Investigator consortia and with important participation from NASA. SPIRE has been developed by a consortium of institutes led by Cardiff University (UK) and including Univ. Lethbridge (Canada); NAOC (China); CEA, LAM (France); IFSI, Univ. Padua (Italy); IAC (Spain); Stockholm Observatory (Sweden); Imperial College London, RAL, UCL-MSSL, UKATC, Univ. Sussex (UK); and Caltech, JPL, NHSC, Univ. Colorado (USA). This development has been supported by national funding agencies: CSA (Canada); NAOC (China); CEA, CNES, CNRS (France); ASI (Italy); MCINN (Spain); SNSB (Sweden); STFC (UK); and NASA (USA).
- 3.Davis, S.P., Abrams, M.C., Brault, J.W.: Fourier Transform Spectroscopy, p. 149. Academic, San Diego (2001)Google Scholar
- 8.Naylor, D.A., Baluteau, J.-P., Barlow, M.J., et al.: In-orbit performance of the Herschel/SPIRE imaging fourier transform spectrometer. In: Society of Photo-Optical Instrumentation Engineers (SPIE) Conference Series, 7731 (2010)Google Scholar
- 9.Ott, S.: The Herschel data processing system - HIPE and pipelines - up and running since the start of the mission. In: ASP Conference Series, vol. 434 (ADASS XVIII), p. 139 (2010)Google Scholar
- 11.Sánchez-Portal M., et al.: this issue (2013)Google Scholar
- 13.Swinyard B., Polehampton E., Hopwood R., et al.: Calibration of the Herschel SPIRE fourier transform spectrometer. MNRAS, submittedGoogle Scholar
- 14.SPIRE observers manual, version 2.4, available from the Herschel Science Centre (2011)Google Scholar