Biomedical signal acquisition with streaming wireless communication for recording evoked potentials
- First Online:
- 365 Downloads
Commercial electrophysiology systems for recording evoked potentials always connect patients to the acquisition unit via long wires. Wires guarantee timely transfer of signals for synchronization with the stimuli, but they are susceptible to electromagnetic and electrostatic interferences. Though wireless solutions are readily available (e.g. Bluetooth), they introduce high delay variability that will distort the evoked potential traces. We developed a complete wireless acquisition system with a fixed delay.
The system supports up to 4 bipolar channels; each is amplified by 20,000× and digitized to 24 bits. The system incorporates the “driven-right-leg” circuit to lower the common noise. Data are continuously streamed using radio-frequency transmission operating at 915 MHz and then tagged with the stimulus SYNC signal at the receiver. The delay, noise level and transmission error rate were measured. Flash visual evoked potentials were recorded monocularly from both eyes of six adults with normal vision. The signals were acquired via wireless and wired transmissions simultaneously. The recording was repeated on some participants within 2 weeks.
The delay was constant at 20 ms. The system noise was white and Gaussian (2 microvolts RMS). The transmission error rate was about one per million packets. The VEPs recorded with wireless transmission were consistent with those with wired transmission. The VEP amplitudes and shapes showed good intra-session and inter-session reproducibility and were consistent across eyes.
The wireless acquisition system can reliably record visual evoked potentials. It has a constant delay of 20 ms and very low error rate.
KeywordsVEP Wireless EEG ERG ECG Multifocal
- 5.Modarreszadeh M, Schmidt RN (1997) Wireless, 32-channel, EEG and epilepsy monitoring system. Proc Annu Int Conf IEEE Eng Med Biol 3:1157–1160Google Scholar
- 6.Irazoqui-Pastor P, Mody I, Judy JW (2003) In vivo EEG recording using a wireless implantable neural transceiver. In: Proceedings of 1st international conference on IEEE-EMBS conference of neural engineering, pp 622–625Google Scholar
- 8.Steyaert M, Gogaert S, Van Nuland T, Sansen W (1991) A low-power portable telemetry system for eight-channel EMG measurements. Proc Annu Int IEEE-EMBS Conf 13:1711–1712Google Scholar
- 9.Vehkaoja AT, Verho JA, Puurtinen MM, Nöjd NM, Lekkala JO, Hyttinen JA (2005) Wireless head cap for EOG and facial EMG measurements. Proc Annu Int Conf IEEE Eng Med Biol 7:5865–5868Google Scholar
- 15.Ellingson RM, Oken B (2010) Feasibility and performance evaluation of generating and recording visual evoked potentials using ambulatory Bluetooth based system. 2010 Annual international conference of the IEEE engineering in medicine and biology society, EMBC’10, pp 6829–6832Google Scholar
- 16.Webster JG (2009) Medical instrumentation application and design. Wiley, New JerseyGoogle Scholar
- 18.Heckenlively JR, Arden GB (2006) Principles and practice of clinical electrophysiology of vision. MIT Press, CambridgeGoogle Scholar