Repeatability of quantitative assessment of atherosclerotic plaques is important for the accurate detection of high-risk plaques in coronary CT angiography (CTA). We assessed the effect of heart rate (HR) on plaque CT number using a coronary artery model and a cardiac phantom capable of simulating cardiac motion. The coronary artery model with luminal stenosis on a cardiac phantom was imaged with a simulated HR of 0, 50, 60, and 70 beats per minute using a 320-row CT scanner. We reconstructed CT images for cardiac diastolic phases (for 75% R–R interval) using filtered back projection (FBP), hybrid iterative reconstruction (AIDR3D), and model-based iterative reconstruction (FIRST). Two observers measured plaque attenuation in the lesion with 75% stenosis. The coefficient of determination (R2) was obtained to evaluate interobserver agreement. At HR 70, FIRST improved the correlation between two observers compared with FBP and AIDR3D (FIRST: R2 = 0.68, p < 0.05; FBP: R2 = 0.29, p = 0.31; AIDR3D: R2 = 0.22, p = 0.18). These R2 at HR 70 were lower compared with at HR 50 (FIRST: R2 = 0.92, p < 0.05; FBP: R2 = 0.83, p < 0.05; AIDR3D: R2 = 0.87, p < 0.05) and HR 0 (FIRST: R2 = 0.97, p < 0.05; FBP: R2 = 0.89, p < 0.05; AIDR3D: R2 = 0.95, p < 0.05). Higher HR affected plaque measurement repeatability in coronary CTA. FIRST may improve plaque measurement repeatability at the higher HR compared with FBP and AIDR3D.
This is a preview of subscription content, log in to check access.
Buy single article
Instant unlimited access to the full article PDF.
Price includes VAT for USA
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
This is the net price. Taxes to be calculated in checkout.
Computed tomography angiography
Beats per minute
Filtered back projection
Adaptive iterative dose reduction 3D
Forward projected model-based iterative reconstruction solution
Region of interest
Intraclass correlation coefficients
Limits of agreement
Coronary Artery Disease Reporting and Data System
Miller JM, Rochitte CE, Dewey M et al (2008) Diagnostic performance of coronary angiography by 64-row CT. N Engl J Med 359(22):2324–2336
Gu H, Gao Y, Hou Z et al (2017) Prognostic value of coronary atherosclerosis progression evaluated by coronary CT angiography in patients with stable angina. Eur Radiol 28:1066–1076
Ghoshhajra BB, Takx RAP, Staziaki PV et al (2017) Clinical implementation of an emergency department coronary computed tomographic angiography protocol for triage of patients with suspected acute coronary syndrome. Eur Radiol 27(7):2784–2793
Ferencik M, Mayrhofer T, Puchner SB et al (2015) Computed tomography-based high-risk coronary plaque score to predict acute coronary syndrome among patients with acute chest pain—results from the ROMICAT II trial. J Cardiovasc Comput Tomogr 9(6):538–545
Cury RC, Abbara S, Achenbach S et al (2016) CAD-RADS (TM) Coronary artery disease—reporting and data system. An expert consensus document of the Society of Cardiovascular Computed Tomography (SCCT), the American College of Radiology (ACR) and the North American Society for Cardiovascular Imaging (NASCI). Endorsed by the American College of Cardiology. J Cardiovasc Comput Tomogr 10(4):269–281
Tarkin JM, Dweck MR, Evans NR et al (2016) Imaging atherosclerosis. Circ Res 118(4):750–769
Motoyama S, Sarai M, Harigaya H et al (2009) Computed tomographic angiography characteristics of atherosclerotic plaques subsequently resulting in acute coronary syndrome. J Am Coll Cardiol 54(1):49–57
Szilveszter B, Celeng C, Maurovich-Horvat P (2016) Plaque assessment by coronary CT. Int J Cardiovasc Imaging 32(1):161–172
Shaw LJ, Narula J, Chandrashekhar Y (2015) The never-ending story on coronary calcium: is it predictive, punitive, or protective? J Am Coll Cardiol 65(13):1283–1285
Kitagawa T, Yamamoto H, Horiguchi J et al (2011) Effects of statin therapy on non-calcified coronary plaque assessed by 64-slice computed tomography. Int J Cardiol 150(2):146–150
Auscher S, Heinsen L, Nieman K et al (2015) Effects of intensive lipid-lowering therapy on coronary plaques composition in patients with acute myocardial infarction: assessment with serial coronary CT angiography. Atherosclerosis 241(2):579–587
Cademartiri F, Mollet NR, Runza G et al (2005) Influence of intracoronary attenuation on coronary plaque measurements using multislice computed tomography: observations in an ex vivo model of coronary computed tomography angiography. Eur Radiol 15(7):1426–1431
Achenbach S, Boehmer K, Pflederer T et al (2010) Influence of slice thickness and reconstruction kernel on the computed tomographic attenuation of coronary atherosclerotic plaque. J Cardiovasc Comput Tomogr 4(2):110–115
Suzuki S, Furui S, Kuwahara S et al (2006) Accuracy of attenuation measurement of vascular wall in vitro on computed tomography angiography: effect of wall thickness, density of contrast medium, and measurement point. Invest Radiol 41(6):510–515
Kidoh M, Utsunomiya D, Oda S et al (2016) Evaluation of the effect of intracoronary attenuation on coronary plaque measurements using a dual-phase coronary CT angiography technique on a 320-row CT scanner—in vivo validation study. Acad Radiol 23(3):315–320
Tanami Y, Ikeda E, Jinzaki M et al (2010) Computed tomographic attenuation value of coronary atherosclerotic plaques with different tube voltage: an ex vivo study. J Comput Assist Tomogr 34(1):58–63
Dalager MG, Bottcher M, Dalager S et al (2011) Imaging atherosclerotic plaques by cardiac computed tomography in vitro: impact of contrast type and acquisition protocol. Invest Radiol 46(12):790–795
Arbab-Zadeh A, Texter J, Ostbye KM et al (2010) Quantification of lumen stenoses with known dimensions by conventional angiography and computed tomography: implications of using conventional angiography as gold standard. Heart 96(17):1358–1363
Seifarth H, Wienbeck S, Pusken M et al (2007) Optimal systolic and diastolic reconstruction windows for coronary CT angiography using dual-source CT. AJR Am J Roentgenol 189(6):1317–1323
Araoz PA, Kirsch J, Primak AN et al (2009) Optimal image reconstruction phase at low and high heart rates in dual-source CT coronary angiography. Int J Cardiovasc Imaging 25(8):837–845
Funama Y, Utsunomiya D, Hirata K et al (2017) Improved estimation of coronary plaque and luminal attenuation using a vendor-specific model-based iterative reconstruction algorithm in contrast-enhanced ct coronary angiography. Acad Radiol 24(9):1070–1078
Tatsugami F, Higaki T, Sakane H et al (2017) Coronary artery stent evaluation with model-based iterative reconstruction at coronary CT angiography. Acad Radiol 24(8):975–981
Dalager MG, Bottcher M, Andersen G et al (2011) Impact of luminal density on plaque classification by CT coronary angiography. Int J Cardiovasc Imaging 27(4):593–600
Linsen PV, Coenen A, Lubbers MM, Dijkshoorn ML, Ouhlous M, Nieman K (2016) Computed tomography angiography with a 192-slice dual-source computed tomography system: improvements in image quality and radiation dose. J Clin Imaging Sci 6:44
Leipsic J, Labounty TM, Hague CJ et al (2012) Effect of a novel vendor-specific motion-correction algorithm on image quality and diagnostic accuracy in persons undergoing coronary CT angiography without rate-control medications. J Cardiovasc Comput Tomogr 6(3):164–171
Conflict of interest
The authors declare that they have no conflict of interest.
Electronic supplementary material
Below is the link to the electronic supplementary material.
About this article
Cite this article
Kidoh, M., Utsunomiya, D., Funama, Y. et al. The effect of heart rate on coronary plaque measurements in 320-row coronary CT angiography. Int J Cardiovasc Imaging 34, 1977–1985 (2018) doi:10.1007/s10554-018-1415-0
- Coronary computed tomography angiography
- Coronary artery disease
- Coronary artery plaque
- Non-calcified plaque
- Model-based iterative reconstruction