Optical coherence tomography findings of very late stent thrombosis after drug-eluting stent implantation
- 269 Downloads
Previous optical coherence tomography (OCT) studies in patients with drug-eluting stents (DESs)-related very late stent thrombosis (VLST) were scarce. Therefore, we investigated OCT findings of VLST after implantation of DESs. Using OCT, we analyzed the status of stent struts and neointimal characteristics in 18 patients who developed VLST after DES implantation. These results were compared to those in 57 patients with neointimal hyperplasia causing >40% diameter stenosis. Lipid-laden neointima was defined as a region with marked signal attenuation and a diffuse border. Four (22.2%) of 18 patients with VLST had ruptured and lipid-laden neointima inside DESs without uncovered or malapposed stent struts. In the remaining 14 patients who developed VLST without neointimal rupture, uncovered and malapposed struts were observed in nine and seven patients, respectively, and lipid-laden neointima in four patients. Lipid-laden neointima was more frequently observed in four patients with neointimal rupture than in 14 patients without neointimal rupture (100% vs. 28.6%, respectively, P = 0.023). Of 57 patients with neointimal hyperplasia, eight (14.0%) had lipid-laden neointima. Time to OCT study after DES implantation was significantly longer in the eight patients with lipid-laden neointima than in 49 patients without lipid-laden neointima (45.5 ± 17.7 months vs. 11.7 ± 7.2 months, respectively, P < 0.001). Lipid-laden neointima was detected in some patients with neointimal hyperplasia > 1 year after DES implantation. In addition to uncovered or malapposed struts, rupture of lipid-laden neointima inside DESs was identified in some patients with DES-related VLST.
KeywordsOptical coherence tomography Stent Thrombosis
This study was partly supported by grants from the Korea Healthcare Technology R&D Project, Ministry for Health, Welfare & Family Affairs, Republic of Korea (No. A085012 and A102064), the Korea Health 21 R&D Project, Ministry of Health & Welfare, Republic of Korea (No. A085136), and the Cardiovascular Research Center, Seoul, Korea.
Conflict of interest
The authors do not have any potential conflicts of interest associated with this paper.
- 1.Moses JW, Leon MB, Popma JJ, Fitzgerald PJ, Holmes DR, O’Shaughnessy C, Caputo RP, Kereiakes DJ, Williams DO, Teirstein PS, Jaeger JL, Kuntz RE (2003) Sirolimus-eluting stents versus standard stents in patients with stenosis in a native coronary artery. N Engl J Med 349(14):1315–1323PubMedCrossRefGoogle Scholar
- 10.Takano M, Yamamoto M, Inami S, Murakami D, Ohba T, Seino Y, Mizuno K (2009) Appearance of lipid-laden intima and neovascularization after implantation of bare-metal stents extended late-phase observation by intracoronary optical coherence tomography. J Am Coll Cardiol 55:26–32PubMedCrossRefGoogle Scholar
- 19.Stettler C, Wandel S, Allemann S, Kastrati A, Morice MC, Schomig A, Pfisterer ME, Stone GW, Leon MB, de Lezo JS, Goy JJ, Park SJ, Sabate M, Suttorp MJ, Kelbaek H, Spaulding C, Menichelli M, Vermeersch P, Dirksen MT, Cervinka P, Petronio AS, Nordmann AJ, Diem P, Meier B, Zwahlen M, Reichenbach S, Trelle S, Windecker S, Juni P (2007) Outcomes associated with drug-eluting and bare-metal stents: a collaborative network meta-analysis. Lancet 370:937–948PubMedCrossRefGoogle Scholar
- 26.Hassan AK, Bergheanu SC, Stijnen T, van der Hoeven BL, Snoep JD, Plevier JW, Schalij MJ, Wouter Jukema J (2010) Late stent malapposition risk is higher after drug-eluting stent compared with bare-metal stent implantation and associates with late stent thrombosis. Eur Heart J 31:1172–1180PubMedCrossRefGoogle Scholar
- 27.Cook S, Ladich E, Nakazawa G, Eshtehardi P, Neidhart M, Vogel R, Togni M, Wenaweser P, Billinger M, Seiler C, Gay S, Meier B, Pichler WJ, Juni P, Virmani R, Windecker S (2009) Correlation of intravascular ultrasound findings with histopathological analysis of thrombus aspirates in patients with very late drug-eluting stent thrombosis. Circulation 120:391–399PubMedCrossRefGoogle Scholar
- 33.Prati F, Regar E, Mintz GS, Arbustini E, Di Mario C, Jang IK, Akasaka T, Costa M, Guagliumi G, Grube E, Ozaki Y, Pinto F, Serruys PW (2010) Expert review document on methodology, terminology, and clinical applications of optical coherence tomography: physical principles, methodology of image acquisition, and clinical application for assessment of coronary arteries and atherosclerosis. Euro Heart J 31(4):401–415CrossRefGoogle Scholar