Closed Loop Control of an MR-Conditional Robot with Wireless Tracking Coil Feedback
- 121 Downloads
This paper presents a hardware and software system to implement the task space control of an MR-conditional robot by integrating inductively coupled wireless coil based tracking feedback into the control loop. The main motivation of this work is to increase the accuracy performance and address the system uncertainties in the practical scenarios. We present the MR-conditional robot hardware design, wireless tracking method, and custom-designed communication software for real-time tracking data transfer. Based on these working principles, we fabricate the robot platform and evaluate the complete system with respect to various performance indices, i.e. data communication speed, targeting accuracy, tracking coil resolution, image quality, temperature variation, and task space control accuracy for static and dynamic targeting inside MRI scanner. The in-scanner targeting results show that the MR-conditional robot with wireless tracking coil feedback achieves the targeting error of 0.17 ± 0.08 mm, while the error calculated from the joint space optical encoder feedback is 0.68 ± 0.19 mm.
KeywordsMR-conditional robot Wireless tracking Robot control
The authors would like to acknowledge Eric Barth, Robert Webster, and Edward B Welch for their support.
Conflict of interest
All authors declared that they have no conflict of interest.
- 1.Anastasiadis, A. G., M. P. Lichy, U. Nagele, M. A. Kuczyk, A. S. Merseburger, J. Hennenlotter, S. Corvin, K.-D. Sievert, C. D. Claussen, and A. Stenzl. MRI-guided biopsy of the prostate increases diagnostic performance in men with elevated or increasing PSA levels after previous negative TRUS biopsies. Eur. Urol. 50:738–749, 2006.CrossRefGoogle Scholar
- 5.Chen Y., M. E. Poorman, D. B. Comber, E. B. Pitt, C. Liu, I. S. Godage, H. Yu, W. A. Grissom, E. J. Barth and R. J. Webster. Treating epilepsy via thermal ablation: initial experiments with an MRI-guided concentric tube robot. In: 2017 Design of Medical Devices Conference. American Society of Mechanical Engineers, 2017, pp. V001T002A002-V001T002A002Google Scholar
- 10.F2503 A. Standard practice for marking medical devices and other items for safety in the magnetic resonance environment. 2005.Google Scholar
- 11.Fichtinger G., A. Krieger, R. C. Susil, A. Tanacs, L. L. Whitcomb and E. Atalar. Transrectal prostate biopsy inside closed MRI scanner with remote actuation, under real-time image guidance. In: International Conference on Medical Image Computing and Computer-Assisted Intervention. Springer, 2002, pp. 91–98.Google Scholar
- 18.Kwok, K.-W., K.-H. Lee, Y. Chen, W. Wang, Y. Hu, G. C. Chow, H. S. Zhang, W. G. Stevenson, R. Y. Kwong, and W. Luk. Interfacing fast multi-phase cardiac image registration with MRI-based catheter tracking for MRI-guided electrophysiological ablative procedures. Am. Heart Assoc. 130:A18568, 2014.Google Scholar
- 24.Reddy, V. Y., Z. J. Malchano, G. Holmvang, E. J. Schmidt, A. d’Avila, C. Houghtaling, R. C. Chan, and J. N. Ruskin. Integration of cardiac magnetic resonance imaging with three-dimensional electroanatomic mapping to guide left ventricular catheter manipulation: feasibility in a porcine model of healed myocardial infarction. J. Am. Coll. Cardiol. 44:2202–2213, 2004.CrossRefGoogle Scholar
- 30.Sutherland G. R., P. B. McBeth and D. F. Louw. NeuroArm: an MR compatible robot for microsurgery. In: International Congress Series. Elsevier, 2003, pp. 504–508.Google Scholar
- 33.Yang, B., U.-X. Tan, A. McMillan, R. Gullapalli, and J. P. Desai. Design and implementation of a pneumatically-actuated robot for breast biopsy under continuous MRI. New York: Springer, pp. 674–679, 2011.Google Scholar