Wearable Water Content Sensor Based on Ultrasound and Magnetic Sensing
Fluid accumulation in the lower extremities is an early indicator of disease deterioration in cardiac failure, chronic venous insufficiency and lymphedema. At-home wearable monitoring and early detection of fluid accumulation can potentially lead to prompt medical intervention and avoidance of hospitalization. Current methods of fluid accumulation monitoring either suffer from lack of specificity and sensitivity or are invasive and cost-prohibitive to use on a daily basis. Ultrasound velocity in animal and human tissue has been found to change with water content. However, previous prototype fluid monitoring sensors based on ultrasound are cumbersome and not wearable. Hence, in this research a compact water content sensor based on a wearable instrumented elastic band is proposed. A novel integration of magnetic sensing and ultrasonic sensing is utilized, where the magnetic sensor provides distance measurement and the ultrasonic sensor produces time-of-flight measurement. Magnetic field modeling with a Kalman filter and least squares linear fitting algorithms are employed to ensure robust sensor performance on a wearable device. The combination of the two measurements yields ultrasound velocity measurement in tissue. The water content sensor prototype was tested on a tissue phantom, on animal tissue and on a human leg. The error in velocity measurement is shown to be small enough for early detection of tissue edema.
KeywordsUltrasound time-of-flight Magnetic sensing Magnetic field modeling Water content Fluid accumulation Kalman filters Human volunteer tests
This work was partly supported by funds from the National Science Foundation, Division of Information & Intelligent Systems, Grant IIS-1231582, and by the Minnesota Partnership for Biotechnology and Medical Genomics.
Conflict of interest
A portion of the work reported in this paper has been protected through a patent filing. The pending patent will belong to the University of Minnesota and not to the authors of this paper. The authors have no conflict of interest in presenting these results for publication.
- 6.Cleveland Clinic. Heart Failure: Monitoring Your Weight & Fluid Intake. 2016. https://my.clevelandclinic.org/health/diseases/17243-heart-failure-monitoring-your-weight–fluid-intake. Accessed on 6 February 2018.
- 7.Cobbold, R. S. C. Foundations of Biomedical Ultrasound. Oxford: Oxford University Press, 2007.Google Scholar
- 8.Gu, M. D. Investigating a relationship between speed of sound and hydrogel water content via ultrasound for future articular cartilage applications. Cleveland, OH: Case Western Reserve University, p. 2013, 2013.Google Scholar
- 10.Harvard Health. Fluid Retention: What it can Mean for Your Heart. 2014. https://www.health.harvard.edu/heart-health/fluid-retention-what-it-can-mean-for-your-heart. Accessed on 6 February 2018
- 11.Harvard University. A summary of error propagation. Phys. Sci. 2:5, 2007.Google Scholar
- 13.Inan, U. S., and A. S. Inan. Engineering Electromagnetics. Menlo Park: Addison-Wesley, 1999.Google Scholar
- 15.Kumarasinghe, G., and G. Carroll. A guide to peripheral oedema. Med. Today 16:26–34, 2015.Google Scholar
- 16.Maines, M., D. Catanzariti, C. Cemin, C. Vaccarini, and G. Vergara. Usefulness of intrathoracic fluids accumulation monitoring with an implantable biventricular defibrillator in reducing hospitalizations in patients with heart failure: a case-control study. J. Interv. Card. Electrophysiol. 19:201–207, 2007.CrossRefGoogle Scholar
- 17.Marshburn, T., R. Cole, J. Pavela, K. Garcia and A. Sargsyan. Facial Soft Tissue Measurement in Microgravity-Induces Fluid Shifts, 2014.Google Scholar
- 20.Micheau, A. and D. Hoa. Lower extremity: MRI of anatomical atlas. e-Anatomy, 2017. https://www.imaios.com/en/e-Anatomy/Limbs/Lower-extremity-MRI. Accessed on 20th June 2018.
- 23.Onda Corporation. Acoustic Properties of Rubbers. Sunnyvale, CA: Onda Corporation, 2003.Google Scholar
- 24.Paulev, P.-E. Textbook in medical physiology and pathophysiology. Copenhagen: Copenhagen Medical Publishers, 1999.Google Scholar
- 25.Physics Stack Exchange. What is the Strength of the Magnetic Field Required to Penetrate an Average Human Body? 2017. http://physics.stackexchange.com/questions/194312/what-is-the-strength-of-the-magnetic-field-required-to-penetrate-an-average-huma. Accessed on 6th February 2018.
- 26.Physics Stack Exchange. From What Wavelength can Radiation go Through a Human Body Without Very Much Changing? 2017. https://physics.stackexchange.com/questions/239494/from-what-wavelength-can-radiation-go-through-a-human-body-without-very-much-cha. Accessed on 6th February 2018.
- 27.Routson, M. Why it is important to use gel in ultrasound. HubPages, 2016. https://hubpages.com/education/aquasonic_ultrasound_gel. Accessed on 6th February 2018.
- 34.Trayes, K. P., J. S. Studdiford, S. Pickle, and A. S. Tully. Edema: diagnosis and management. Am. Fam. Physician 88:102–110, 2013.Google Scholar
- 35.Ulaby, F. T., and U. Ravaioli. Fundamentals of Applied Electromagnetics. Upper Saddle River, NJ: Pearson, 2015.Google Scholar
- 37.Zhang, S., R. Rajamani, L. Alexander, and A. Serdar Sezen. Note: development of leg size sensors for fluid accumulation monitoring. Rev. Sci. Instrum. 87:1–4, 2016.Google Scholar