Mechanically Stable Intraspinal Microstimulation Implants for Human Translation
The goal of this study was to develop stable intraspinal microstimulation (ISMS) implants for use in humans to restore standing and walking after spinal cord injury. ISMS electrically activates locomotor networks within the lumbar region of the spinal cord. In animals, ISMS produced better functional outcomes than those obtained by other interventions, and recent efforts have focused on translating this approach to humans. This study used domestic pigs to: (1) quantify the movements and length changes of the implant region of the spinal cord during spine flexion and extension movements; and (2) measure the forces leading to the dislodgement of the ISMS electrodes. The displacement of the spinal cord implant region was 5.66 ± 0.57 mm relative to the implant fixation point on the spine. The overall length change of the spinal cord implant region was 5.64 ± 0.59 mm. The electrode dislodgment forces were 60.9 ± 35.5 mN. Based on these results, six different coil types were fabricated and their strain relief capacity assessed. When interposed between the electrodes and the stimulator, five coil types successfully prevented the dislodgement of the electrodes. The results of this study will guide the design of mechanically stable ISMS implants for ultimate human use.
KeywordsIntraspinal implant Dislodgment forces Lumbar spine biomechanics Strain relief Coiled lead wire
Functional electrical stimulation
Foreign body response
Spinal cord injury
- 1.Bamford, J. A., R. M. Lebel, K. Parseyan, and V. K. Mushahwar. The fabrication, implantation and stability of intraspinal microwire arrays in the spinal cord of cat and rat. Trans. Neural Syst. Rehabil. Eng. 2016.Google Scholar
- 10.Hachmann, J. T., J. H. Jeong, P. J. Grahn, G. W. Mallory, L. Q. Evertz, A. J. Bieber, D. A. Lobel, K. E. Bennet, K. H. Lee, and J. L. Lujan. Large animal model for development of functional restoration paradigms using epidural and intraspinal stimulation. Plos One 8:e81443, 2013.CrossRefPubMedPubMedCentralGoogle Scholar
- 12.Holinski, B. J. Restoring walking after spinal cord injury. 2013. <https://era.library.ualberta.ca/public/view/item/uuid:0e698e79-b16a-4519-baf7-552a49dda767/>.
- 22.Nashold, B. S., H. Friedman, J. F. Glenn, J. H. Grimes, W. F. Barry, and R. Avery. Electromicturition in paraplegia. Implantation of a spinal neuroprosthesis. Arch. Surg. Chic. Ill 1960 104:195–202, 1972.Google Scholar
- 27.Purves, D., G. J. Augustine, D. Fitzpatrick, L. C. Katz, A. S. LaMantia, J. O. McNamara, and S. M. Williams. The external anatomy of the spinal cord. In: Neuroscience. Sunderland: Sinauer Associates, 2001. <http://www.ncbi.nlm.nih.gov/books/NBK11160/>
- 32.Smith, J. O. Physical Audio Signal Processing. New York: W3K Publishing, 2010.Google Scholar
- 36.White, A. A., and M. Panjabi. Clinical Biomechanics of the Spine. Philadelphia: Wolters Kluwer, p. 722, 1990.Google Scholar
- 37.WHO Spinal cord injury. <http://www.who.int/mediacentre/factsheets/fs384/en/>