# Transmural Ultrasound-based Visualization of Patterns of Action Potential Wave Propagation in Cardiac Tissue

- 687 Downloads
- 4 Citations

## Abstract

The pattern of action potential propagation during various tachyarrhythmias is strongly suspected to be composed of multiple re-entrant waves, but has never been imaged in detail deep within myocardial tissue. An understanding of the nature and dynamics of these waves is important in the development of appropriate electrical or pharmacological treatments for these pathological conditions. We propose a new imaging modality that uses ultrasound to visualize the patterns of propagation of these waves through the mechanical deformations they induce. The new method would have the distinct advantage of being able to visualize these waves deep within cardiac tissue. In this article, we describe one step that would be necessary in this imaging process—the conversion of these deformations into the action potential induced active stresses that produced them. We demonstrate that, because the active stress induced by an action potential is, to a good approximation, only nonzero along the local fiber direction, the problem in our case is actually overdetermined, allowing us to obtain a complete solution. Use of two- rather than three-dimensional displacement data, noise in these displacements, and/or errors in the measurements of the fiber orientations all produce substantial but acceptable errors in the solution. We conclude that the reconstruction of action potential-induced active stress from the deformation it causes appears possible, and that, therefore, the path is open to the development of the new imaging modality.

## Keywords

Action potential imaging Ultrasound Cardiac muscle biomechanics Cardiac electrophysiology## Introduction

Current technology does not allow us to measure large-scale action potential propagation deep inside the walls of the heart. This deficiency has a number of consequences. In the clinical setting, we rely on diagnostics such as the electrocardiogram (ECG) or electrophysiology (EP) studies to provide information about the heart’s electrical state. While these tests do certainly yield a wealth of valuable information, they often do not provide a fundamental understanding of the *dynamics* underlying the patient’s cardiac rhythm. For example, the field of cardiology is currently moving away from EP studies as a diagnostic for determining whether to implant an implantable cardioverter defibrillator (ICD),6 relying instead on the ejection fraction,1,9,20 paradoxically a mechanical rather than electrical measure, as the basis for the decision. This suggests that an improved diagnostic for assessing the patient’s rhythm dynamics is needed. At the research level, this lack of a deep, panoramic view of action potential activity means that we cannot even say with certainty that ventricular fibrillation (VF) is caused by multiple re-entrant action potential waves, although of course it is strongly suspected. Even if VF is a manifestation of multiple waves, the nature of the dynamics of these waves remains controversial, with their induction and maintenance being attributed to either tissue heterogeneity27,33 or steep electrical restitution,11,12,17 and their spatial patterning thought to be composed of either several rotating waves on equal dynamical footing,18,19,22,30 or one dominant “mother rotor” wave driving fibrillatory conduction.4,26 The lack of clarity on these issues has greatly complicated the development of effective therapies for the prevention and treatment of several types of tachyarrhythmias, including VF.

This bewildering landscape of theoretical wave propagation patterns and the mechanisms for their initiation and maintenance are what have led researchers to develop tools that might be used to actually determine what patterns exist during VF and other rapid, abnormal rhythms. Rudy *et al.* 25,29 and others have developed a noninvasive method called ECGi to track action potential propagation on the epicardium using hundreds of electrodes situated on the body surface. This method does not as yet provide transmural information, however. Other approaches, such as plunge electrodes,8,10,24 transillumination,2 and optical tomography15,31 can provide a 3D representation of the electrical activity in the heart. The spatial resolution of the plunge electrode method is, however, relatively low. Furthermore, it is likely that the presence of these electrodes creates electrical heterogeneities that strongly modify the patterns of propagation.3 Transillumination and optical tomography are promising optical techniques; however, they require the use of voltage sensitive dyes that are phototoxic and prevent the application of these techniques *in vivo*. Furthermore, optical tomography cannot achieve sufficient temporal resolution to resolve transient phenomena present during fibrillation.

Magnetic resonance imaging (MRI) and computed tomography (CT) are also capable of noninvasively mapping the 3D contractile motion resulting from electrical activity in the heart. Although these imaging modalities provide very high spatial resolution, neither technology has the time resolution necessary (5–10 ms) to visualize, for non-repeating events, the relatively fast timescales inherent in the propagation of action potential waves.

We can argue that the 3D implementations of ultrasound are among the most promising for investigating the behavior of arrhythmias deep within cardiac tissue. Compared to the aforementioned imaging techniques, ultrasound imaging is attractive because of its relative ease of use, real-time feedback, portability, absence of ionizing radiation, cost effectiveness, and widespread availability in research and clinical applications. We can estimate the required temporal and spatial resolutions to be on the order of 10 ms and 1.5 mm, respectively. These are based on capturing 10 frames per rotation of a spiral wave rotating at 10 Hz, and the characteristic length scale associated with the calcium dynamics (a 10 ms timescale on a wave moving at 15 cm/s). Existing 2D ultrasound machines are easily capable of such resolutions. Indeed, ultrasound techniques have been demonstrated, at least in 2D, to be capable of yielding real-time, highly temporally resolved images of electromechanical wave contractions in humans.23,28

The proposed new imaging modality is based on the ability of ultrasound to detect wave-induced tissue deformation at depth within myocardial tissue. The idea is to extend conventional ultrasound strain imaging, which extracts strain information from the deformations appearing in the ultrasound images that are caused by action potential propagation activity, and convert these strains into the action potential-induced stresses that caused them. These stresses will then serve as markers for the presence of the action potentials, allowing us to create mappings of their propagation as functions of both time and space. A similar ultrasound technique called harmonic motion imaging is already the subject of intense study by the Konofagou and collaborators,16 although their goals, which are principally to assess the normal and pathological states of the cardiac tissue, are quite different from the action potential mapping goals of our research.

Here we report on a mathematically based computational technique that is capable of reconstructing the apparent motion of active stresses induced by the passage of action potentials from the mechanical tissue motion. We will regard this motion as motion that can be extracted from an ultrasound image sequence. The locations of these active stresses as functions of time will then, in many situations, serve as reasonable markers for the locations of the action potentials themselves, thereby allowing the study of propagation patterns of action potentials within cardiac tissue, in two and three spatial dimensions.

## Methods

### Overall Approach

*inverse model*(as shown in Fig. 1). As part of the process we used to test the inverse model algorithm, we also built a

*forward model*that simulates the biomechanical response of the 3D cardiac tissue to a given time-dependent active stress field meant to represent the stress created by a propagating action potential.

*Mathematical foundation for both the forward and inverse models.*Both models were based on the following standard equations:

Here *X* _{ N } and *x* _{ j } (*N*, *j* = 1, 2, 3), are the coordinates in the undeformed and deformed coordinate systems, respectively, *T* _{ MN }(*E*) is the passive second Piola–Kirchhoff stress tensor, which depends on the Lagrange–Green strain tensor \( E \equiv \frac{1}{2}(C - I) \) where the Cauchy–Green deformation tensor \( C \equiv F^{T} F \) and the deformation gradient tensor \( F_{jM} = \partial x_{j} /\partial X_{M} , \) *p* is the local hydrostatic pressure, and \( T_{MN}^{\text{active}} \) is the active second Piola–Kirchhoff stress tensor due to the action potential-induced force generated parallel to the orientation of the myocardial fibers. Sums over repeated indices are implicit. Equation (1) is a statement of force balance and ignores inertia, a standard assumption for the timescales we are considering.32 It also assumes the myocardium is a nonlinear, elastic medium. While the myocardium is perhaps better described as pseudoelastic, the variations of the stretch ratios (strains) for any given stress are no more than 20%,7 so we find this simplification appropriate. In this article, we use the hyperelastic, orthotropic pole-zero model of Nash and Hunter21 for the function *T* _{ MN }(*E*). Equation (2) together with the appearance of the hydrostatic pressure in Eq. (1) expresses the assumption of incompressibility, common in the study of cardiac biomechanics.5,14,21

*p*and the displacements \( \delta x_{P} \equiv x_{P} - X_{P} \) are assumed small (i.e.,

*O*(

*ε*), where

*ε*is a small number). The indices

*M*,

*N*,

*P*, and

*Q*run over the values 1, 2, and 3, with summation again implied over repeated indices. This approximation should also be valid during ventricular fibrillation when the tissue deformations are typically very small; however, for cases of sinus rhythm or external electrical pacing, we expect we will have to use the fully nonlinear equations (1) and (2).

*X*, Δ

*Y*, and Δ

*Z*, were chosen to be uniform in each direction. These spacings may be arbitrarily chosen, as nothing in the resulting equations depends on their values. Weighting functions for Eqs. (3) took the form \( N_{l} ({\mathbf{X}}) = \Upsigma_{e} N_{l}^{e} ({\mathbf{X}}), \) for a given node

*l*, where the sum is over adjacent elements

*e*and \( {\mathbf{X}} \equiv (X_{ 1} , { }X_{ 2} ,\,X_{ 3} )\equiv (X,Y,Z). \) The element shape function \( N_{l}^{e} ({\mathbf{X}}) \) is defined as \( N_{l}^{e} ({\mathbf{X}}) = XYZ/\Updelta X\Updelta Y\Updelta Z \) when the node

*l*is located at \( (\Updelta X,\Updelta Y,\Updelta Z) \) and the element

*e*is \( [0,\Updelta X] \times [0,\Updelta Y] \times [0,\Updelta Z], \) with all other element shape functions defined in analogous fashion. The displacements were defined using trial functions identical to these weighting functions; i.e., \( \delta x_{P} ({\mathbf{X}}) = \delta x_{P}^{l} N_{l} ({\mathbf{X}}), \) where the \( \delta x_{P}^{l} \)’s are constants. In contrast, since

*p*, \( T_{MN}^{\text{active}} \), and \( \partial T_{MN} (0)/\partial E_{PQ} \) appear in Eq. (3) with one less derivative than the displacements, it is appropriate to define them in terms of amplitudes defined on the elements rather than the nodes; e.g., \( p({\mathbf{X}}) = p^{e} \bar{N}_{e} ({\mathbf{X}}), \) etc. Here we use a lower order trial function, defined simply as \( \bar{N}_{e} ({\mathbf{X}}) = 1 \) for

**X**within the element

*e*, and 0 otherwise. This lower order function was also used as the weighting function for Eq. (4). When these finite element representations were substituted into Eqs. (3) and (4), multiplied by the weighting functions, and integrated over

**X**, we obtained,

*t*

^{active}, in the 1–1 direction, before rotation by a unitary matrix

*U*orients the tensor in the local fiber direction. This corresponds to our assumption that, as an action potential propagates by in some arbitrary direction, active contractile stress can only occur in the fiber direction, which we define to be in the “1” direction before rotation. This assumption is not trivial physiologically, as the orientation of fibers along a clear direction can, under some pathological conditions, fail to exist. Sums over

*e*and

*f*are over elements (only a single sum in the middle term), sums over

*l*are over nodes, and sums over

*M*,

*P*, and

*Q*are over 1, 2, and 3. Equation (5) depends on

*N*= 1, 2, 3 and node index

*j*; Eq. (6) just depends on

*e*. Equations (5) and (6) are linear and homogeneous in the displacements defined at the nodes, \( \delta x_{P}^{l} , \) the active stresses along the local fiber direction in each of the elements, \( (t^{\text{active}} )^{e} \), and the pressures in the elements,

*p*

^{ e }. We note that any arbitrary rigid rotation or translation of a solution to Eqs. (5) and (6) is again a solution; thus it was useful to constrain the solution to a particular translation and rotational orientation by adding six equations to Eqs. (5) and (6): three that hold the mean displacement of the nodes to 0 in all three dimensions (which also translates into no net displacement of the center of mass if uniform mass density can be assumed):

*The “forward” model: Determination of the tissue displacements caused by a given active stress field.*This function is easily accomplished by rewriting Eq. (9) as,

Given values for all the active stresses in the column vector *t* ^{active}, we have four sets of equations (i.e., those associated with the first four rows of the matrix on the left) for the four unknown column vectors δ*x* _{1}, δ*x* _{2}, δ*x* _{3}, and *p*. These four sets of equations, when combined with the six constraints defined by the fifth row, have sufficient rank to allow solution using the built-in sparse matrix equation solver in the Matlab programming environment (The Mathworks, Inc.), thus yielding a complete description of the tissue deformation and compensating hydrostatic pressure produced by a given active stress field. Since there are no time derivatives in these equations, the temporal evolution of the deformations induced by a time-dependent active stress field may be created simply by solving Eq. (10) repeatedly for each instant in time for which we have active stress field data.

*The inverse model: Determination of the active stresses produced by action potentials using tissue displacement data.*Two variants of the inverse model were tested. Variant 1: If two of the three direction components (say, the “1” and “3” components) of the displacements can be extracted from the full three-dimensional volume, the model can reconstruct the active stress field that caused the displacements everywhere within the volume. We expect that this method will be useful in future experiments that make use of 3D ultrasound imaging techniques. The appropriate form of Eq. (9) for this method is,

In this case, the four sets of equations plus the six constraints represented by Eq. (11) actually constitute an overdetermined problem for the three unknown fields, δ*x* _{2}, *p*, and *t* ^{active}. We therefore solved for them in the least-squares sense, again using Matlab’s built-in sparse matrix solvers.

Variant 2: If the deformation data come from a two-dimensional imaging plane, say, a plane oriented in the *x* _{1} and *x* _{3} (i.e., the *x* and *z*) directions, embedded within the volume, and consist of the two in-plane directional components of the displacement, then we can estimate, within the same plane, the active stress field that caused it. The solution to this problem is, in principle, relevant to the two-dimensional B-mode ultrasound imaging techniques. In this case, however, there is technically not enough data to uniquely determine the stress field. The problem is that the spatial derivatives in the direction perpendicular to the plane contribute to the determination of the stress. These derivatives cannot be determined solely from displacement data obtained from the plane. One of these derivatives may be obtained from the incompressibility condition; i.e., \( \partial \delta x_{2} /\partial X_{2} = - \partial \delta x_{1} /\partial X_{1} - \partial \delta x_{3} /\partial X_{3} ; \) however, this still leaves the other two normal derivatives, \( \partial \delta x_{1} /\partial X_{2} \) and \( \partial \delta x_{3} /\partial X_{2} \) undetermined. For this initial study, we make the assumption that all planes parallel to this plane contain identical data for the displacement fields δ*x* _{1} and δ*x* _{3}. We then apply Eq. (11). We can expect this assumption to be valid if the variation of mechanical deformation is small in the *x* _{2} (i.e., the *y*) direction.

As with the forward problem, either inverse model can generate the active stress field by simply solving Eq. (11) for each time for which displacement data are available.

## Results

### Tests of the Active Stress Reconstruction Algorithm

*y*-direction on the top surface of the cube-shaped tissue in Fig. 2b, rotated in the clockwise direction as a linear function of

*z*to point in the

*x*-direction in the horizontal plane in the middle of the cube, and continued to rotate linearly clockwise so that they pointed back in the

*y*-direction at the bottom of the cube. For this study, a 10 × 10 × 10 lattice of nodes was employed.

We then used the deformations obtained from the forward model as input into the two variants of the “inverse” model, based on Eq. (11). Variant 1, which takes as input two of the three displacement components expressed as spatially 3D data, was able to reproduce the original active stresses essentially exactly, to within 1 part in 10^{12}, as is readily apparent by comparing Figs. 2c to 2a. Variant 2 was able to do a reasonable job of reconstructing the stress field plane wave in the plane from which the deformation data were taken (Fig. 2d). When random errors were purposely introduced into the fiber direction, both variants of the inverse code were still reasonably successful in reconstructing the active stresses, as illustrated in Figs. 2e and 2f, suggesting that the reconstruction process is not highly sensitive to errors we might make in measuring the fiber direction in the experiment. Here the errors were chosen from a distribution of zero mean and standard deviation of 10°.

Studies of this type were conducted for 10 different orientations of this rotated fiber pattern. Specifically, the ability of Variants 1 and 2 to reconstruct the active stress field of a plane wave propagating in the *x*-direction, in both the presence and absence of fiber direction error, was studied for the cases in which the entire fiber orientation spatial pattern just described was rotated through angles ranging from 0° to 90° around the *z*-axis in increments of 10°. For all 10 orientations, Variant 1 without fiber error reproduced the original active stress field to within 1 part in 10^{12}. For the other three cases (i.e., Variant 2 without fiber errors and both variants with fiber errors), we found that the reconstruction was more accurate in regions in which the fiber orientations lie in the *x*–*z* plane. For example, for the fiber orientation pattern used in all the plots in Fig. 2, this region intersects the displayed cross section in the white rectangle in each panel of Figs. 2d–2f. The higher accuracy within each rectangle is manifested as more red, orange, and yellow cells in the portion of the rectangle inside original non-zero active stress region, and primarily blue cells outside this region (but still inside the rectangle). The position of the region containing *x*–*z* oriented fibers changes in each of the 10 orientations, but in each case, the accuracy was found to be highest within this region. Other than differences in the positioning of this higher accuracy region, the reconstructed active stresses for the 10 cases did not exhibit features substantively different from those shown in Figs. 2c–2f.

We observe that, for the Variant 1 of the inverse model (i.e., the variant that uses 3D data as input), the fraction of cells misclassified is smallest when \( t_{\text{active}}^{\text{c}} = 0.82 \), while for Variant 2, the optimal value for \( t_{\text{active}}^{\text{c}} \) is 0.63 or 0.61, depending on whether fiber error is present or not. Using \( t_{\text{active}}^{\text{c}} = 0.82 \) as the classification criterion for the data shown in the two Variant 1 runs shown in Fig. 2 (i.e., panels c and e), we obtain Figs. 2g and 2i, respectively, where orange represents cells in classified as being inside the active stress region, and blue outside. The agreement is perfect between Figs. 2g and 2a, as expected, since the solution is overdetermined in this case and not compromised by fiber orientation errors. The agreement between Figs. 2i and 2a, when fiber error is present, is not perfect, but is acceptable, in the sense that it provides good insight into how the active stress region (and thus, presumably, the corresponding action potential) is propagating. When \( t_{\text{active}}^{\text{c}} = 0.63 \) was used to classify the Variant 2 runs (e.g., Figs. 2d and 2f), favorable results were again obtained. As shown in Figs. 2h and 2j, a reasonably clear representation of the left-to-right motion of the active stress region is again obtained, despite the presence of a few misclassified cells.

As seen in Fig. 5a, there is significant spread in the calculated values of the active stress, both in the active stress “wave” (red curve) and outside it (blue curve). However, if we classify the pixels as being in the wave when the calculated active stress is greater than 1, and outside the wave otherwise, we found that 8.2% of all the pixels were classified as in the AP when they were not, while 4.8% were classified as not in the AP when they were. The remaining 86.9% of the pixels were classified correctly. It was also interesting to note that the classification of pixels was better near the boundaries (e.g., Fig. 5c) than in the center (e.g., Fig. 5b).

### Active Stresses Cause Tissue Displacement at Long Range

*zz*component of the active stress

*T*

_{ zz }is only nonzero at the origin of a three-dimensional domain of infinite extent, while all other components of the active stress are 0 everywhere. Transformation of Eqs. (3) and (4) into a cylindrical coordinate system \( \left( {R,\varphi ,Z} \right) \) yields the following three equations:

*δR*and

*δZ*are displacements in the

*R*and

*Z*directions, respectively, and the constitutive relationship between strain and active stress is assumed to be isotropic with Young’s modulus

*T*. We find that, when the following forms for the dynamical variables and

*T*

_{ ZZ }are substituted into Eqs. (12–14):

*R*and

*Z*are obtained among the coefficients

*C*

_{ R },

*C*

_{ Z },

*C*

_{ P }, and

*C*

_{ T }indicating that Eqs. (15–18) are eigenmodes of the system of equations (12–14). Here

*K*and

*L*are arbitrary constants, and

*J*

_{0}and

*J*

_{1}are 0th and 1st order Bessel functions of the first kind. Rearrangement of these algebraic relationships allows all of these coefficients to be expressed in terms of

*C*

_{ T }:

*t*

_{0}may be written as a Fourier–Bessel integral13:

*C*

_{ T }is given by Eq. (23). The resulting integral expressions may be evaluated using integral tables.13 Transforming the expressions from cylindrical to spherical (

*r*,θ,φ) coordinates, we obtain the following:

These Green’s function solutions show that the displacements and hydrostatic pressure fall off as 1/*r* ^{2} and 1/*r* ^{3}, respectively, in response to active stress located at a single point. The response to active stress thus behaves like a long-range force not unlike gravity or the electrostatic force, affecting the displacements and pressure far from the actual source. This eliminates the possibility that a technique can simply use the local presence of displacements as a marker for the presence of action potentials activity nearby. Thus, it should not be possible in general to determine the location of the source, i.e., the active stresses, without considering the entire problem, including boundary conditions, within the framework of an electromechanical model. We also note that, while the displacements scale inversely with the Young’s Modulus *T* of the tissue, the pressure, which enforces the incompressibility condition, is independent of this quantity. Also, as expected, all quantities scale linearly with the magnitude of the active stress *t* _{0}.

## Discussion

In this article, we describe key mathematical techniques for visualizing the location and motion of waves of action potential-induced active stress traveling both on the surface and deep within myocardial tissue. As an initial test of these ideas, we have developed the linear version of the algorithm and studied its ability to reconstruct simple plane waves of active stress propagating in a three-dimensional cubic system containing fiber directions that rotate with depth. We find that, when the deformation data consist of two of the three components of the displacement field defined throughout the 3D system, the method is very precise, allowing essentially exact reconstruction of the active stresses. This result is to be expected—the theory shows that, when the active stresses are all a function of a single scalar field (i.e., the active stress along the local fiber direction) and the orientation of these fibers is known throughout the tissue, then the problem is actually overdetermined, and thus will yield the exact solution, when it exists. This calculation is the one relevant to ultrasound machines capable of imaging three-dimensional volumes within the myocardium.

In contrast, when the two components of displacement are only known within a two-dimensional imaging plane, as is the case for B-mode ultrasound imaging, determination of the active stresses within this same plane is an underdetermined problem, requiring reasonable assumptions to be made. As a first attempt, we assumed slow variation of the two in-plane displacements in the direction perpendicular to the imaging plane. With this assumption, we find that the method is still able to do a reliable reconstruction of a plane wave consisting of active stresses along the local fiber direction. This type of assumption may not be valid when the tissue properties are strongly heterogeneous, or when the pattern of wave propagation contains more spatial structure, as would be the case, for example, during ventricular fibrillation. This issue will be a focus of further investigation in future studies.

Even when errors of order 10° were purposely introduced into the fiber directions, both variants of the inverse model performed reasonably well in reconstructing the active stress plane wave. This result suggests that the reconstruction process is not highly sensitive to errors in the measurements of fiber orientation, an important characteristic of the method, as a requirement of precise measurements of the fiber orientations would greatly complicate the use of the method as an imaging modality.

The method also performed reasonably well when white noise of relative amplitude 10% was added to the displacement data. We currently have no predictions about the spatial power spectrum of noise and error we will find in the ultrasound data; however, we believe that the assumption of white noise should be a fairly pessimistic one. It is not too hard to imagine the existence of short-distance positive correlation in the error that results from a tendency for points in the tissue that are in close proximity to move together. Thus, we can reasonably expect that the performance of our inverse algorithm will be even better in practice than is shown here.

One can question whether such an involved inverse calculation is necessary to extract the active stress from the tissue displacements they induce. However, as we demonstrate in the calculations embodied by Eqs. (12–26), the magnitudes of these displacements fall off only as fast as 1/*r* ^{2} with distance, so we believe it is unlikely that active stresses in a particular location can be determined solely from the displacements present in that area.

The method described here does not yield detailed membrane potential profiles of the action potentials. Such profiles are likely to be difficult or impossible to extract from the output of our method, the active stresses, because a clear one-to-one relationship between these stresses and the level of the membrane potential does not exist. This might, for example, be the case in partially refractory tissue or near the centers of spiral wave rotation. Nevertheless, we expect that the spatial and temporal patterning of active stresses yielded by our method should be a reasonable indicator of the location, motion, and spatial patterning of action potentials and their propagation, most of the time over most of the tissue. This level of imaging is, in our opinion, still likely to be very useful in determining the essential dynamics of the action potentials, which we believe will be important in diagnosing the nature of their propagation during clinically important phenomena such as ventricular polymorphic tachycardia and fibrillation.

While we are currently targeting ultrasound imaging as the application for this method, we note that the applicability of the method is not necessarily restricted to the ultrasound modality. Any imaging method that can be translated into mechanical tissue deformations in at least a 2D imaging plane is a potential candidate for use with this method.

## Notes

### Acknowledgments

S. Luther acknowledges support from the Bundesministerium für Bildung und Forschung (FKZ 01EZ0905/6). R. Singh was supported in part by the National Science Foundation, Grant No. 0800793. We thank VisualSonics Inc. (Toronto, Canada) for useful discussions. We appreciate the reviewers’ helpful and useful comments in preparing this article.

### Open Access

This article is distributed under the terms of the Creative Commons Attribution Noncommercial License which permits any noncommercial use, distribution, and reproduction in any medium, provided the original author(s) and source are credited.

## References

- 1.Bardy, G. H., K. L. Lee, D. B. Mark, J. E. Poole, D. L. Packer, R. Boineau, M. Domanski, C. Troutman, J. Anderson, G. Johnson, S. E. McNulty, N. Clapp-Channing, L. D. Davidson-Ray, E. S. Fraulo, D. P. Fishbein, R. M. Luceri, and J. H. Ip. Sudden Cardiac Death in Heart Failure Trial (SCD-HeFT) Investigators. Amiodarone or an implantable cardioverter-defibrillator for congestive heart failure.
*N. Engl. J. Med.*352:225–237, 2005.CrossRefPubMedGoogle Scholar - 2.Baxter, W. T., S. F. Mironov, A. V. Zaitsev, J. Jalife, and A. M. Pertsov. Visualizing excitation waves inside cardiac muscle using transillumination.
*Biophys. J.*80:516–530, 2001.CrossRefPubMedGoogle Scholar - 3.Beaudoin, D. L., and B. J. Roth. Effect of plunge electrodes in active cardiac tissue with curving fibers.
*Heart Rhythm*4:476–481, 2004.Google Scholar - 4.Chen, J., R. Mandapati, O. Berenfeld, A. C. Skanes, and J. Jalife. High-frequency periodic sources underlie ventricular fibrillation in the isolated rabbit heart.
*Circ. Res.*86(1):86–93, 2000.PubMedGoogle Scholar - 5.Costa, K. D., J. W. Holmes, and A. D. McCulloch. Modelling cardiac mechanical properties in three dimensions.
*Philos. Trans. Roy. Soc. Lond.*A35:1233–1250, 2001.CrossRefGoogle Scholar - 6.Daubert, J. P., W. Zareba, W. J. Hall, C. Schuger, A. Corsello, A. R. Leon, M. L. Andrews, S. McNitt, D. T. Huang, and A. J. Moss. Predictive value of ventricular arrhythmia inducibility for subsequent ventricular tachycardia or ventricular fibrillation in Multicenter Automatic Defibrillator Implantation Trial (MADIT) II patients.
*J. Am. Coll. Cardiol.*47:98–107, 2006.CrossRefPubMedGoogle Scholar - 7.Demer, L. L., and F. C. P. Yin. Passive biaxial mechanical properties of isolated canine myocardium.
*J. Physiol.*339:615–630, 1983.PubMedGoogle Scholar - 8.El-Sherif, N., M. Chinushi, E. B. Caref, and M. Restivo. Electrophysiological mechanism of the characteristic electrocardiographic morphology of torsade de pointes tachyarrhythmias in long-QT syndrome: detailed analysis of ventricular tridimensional activation patterns.
*Circulation*96:4392–4399, 1997.PubMedGoogle Scholar - 9.Epstein A. E., J. P. DiMarco, K. A. Ellenbogen, N. A. M. Estes III, R. A. Freedman, L. S. Gettes, A. M. Gillinov, G. Gregoratos, S. C. Hammill, D. L. Hayes, M. A. Hlatky, L. K. Newby, R. L. Page, M. H. Schoenfeld, M. J. Silka, L. W. Stevenson, and M. O. Sweeney. ACC/AHA/HRS 2008 Guidelines for device-based therapy of cardiac rhythm abnormalities: a report of the American College of Cardiology/American Heart Association Task Force on Practice Guidelines (Writing Committee to Revise the ACC/AHA/NASPE 2002 Guideline Update for Implantation of Cardiac Pacemakers and Antiarrhythmia Devices): developed in collaboration with the American Association for Thoracic Surgery and Society of Thoracic Surgeons.
*Circulation*117:e350–e408, 2008.Google Scholar - 10.Frazier, D. W., P. D. Wolf, J. M. Wharton, A. S. Tang, W. M. Smith, and R. E. Ideker. Stimulus-induced critical point. Mechanism for electrical initiation of reentry in normal canine myocardium.
*J. Clin. Invest.*83:1039–1052, 1989.CrossRefPubMedGoogle Scholar - 11.Garfinkel, A., Y. H. Kim, O. Voroshilovsky, Z. Qu, J. R. Kil, M. H. Lee, H. S. Karagueuzian, J. N. Weiss, and P. S. Chen. Preventing ventricular fibrillation by flattening cardiac restitution.
*Proc. Natl Acad. Sci. USA*97:6061–6066, 2000.CrossRefPubMedGoogle Scholar - 12.Gilmour, Jr., R. F., and D. R. Chialvo. Electrical restitution, critical mass, and the riddle of fibrillation.
*J. Cardiovasc. Electrophysiol.*10:1087–1089, 1999.CrossRefPubMedGoogle Scholar - 13.Gradshteyn, I. S., and I. M. Ryzhik. Table of Integrals, Series, and Products: Seventh Edition, edited by A. Jeffrey and D. Zwillinger. Burlington, MA: Academic Press, 2007, pp. 659–753.Google Scholar
- 14.Guccione, J. M., A. D. McCulloch, and L. K. Waldman. Passive material properties of intact ventricular myocardium determined from a cylindrical model.
*ASME J. Biomech. Eng.*113:42–55, 1991.CrossRefGoogle Scholar - 15.Hillman, E. M. C., O. Bernus, E. Pease, M. B. Bouchard, and A. Pertsov. Depth-resolved optical imaging of transmural electrical propagation in perfused heart.
*Opt. Exp.*15(26):17827–17841, 2007.CrossRefGoogle Scholar - 16.Hynynen, K. H., and E. Konofagou. Harmonic Motion Imaging. Patent 6,984,209, 10 Jan 2006.Google Scholar
- 17.Karma, A. Electrical alternans and spiral wave breakup in cardiac tissue.
*Chaos*4:461–472, 1994.CrossRefPubMedGoogle Scholar - 18.Lee, M. H., Z. L. Qu,
*et al*. Patterns of wave break during ventricular fibrillation in isolated swine right ventricle.*Am. J. Physiol. Heart Circ. Physiol.*281(1):H253–H265, 2001.PubMedGoogle Scholar - 19.Moe, G. K. On the multiple wavelet hypothesis of atrial fibrillation.
*Arch. Int. Pharmacodyn. Ther.*140:183–188, 1962.Google Scholar - 20.Moss, A. J., W. Zareba, W. J. Hall, H. Klein, D. J. Wilber, D. S. Cannom, J. P. Daubert, S. L. Higgins, M. W. Brown, and M. L. Andrews. Multicenter Automatic Defibrillator Implantation Trial II Investigators. Prophylactic implantation of a defibrillator in patients with myocardial infarction and reduced ejection fraction.
*N. Engl. J. Med.*346:877–883, 2002.CrossRefPubMedGoogle Scholar - 21.Nash, M. P., and P. J. Hunter. Computational mechanics of the heart.
*J. Elasticity*61:113–141, 2000.CrossRefGoogle Scholar - 22.Panfilov, A. Spiral breakup as a model of ventricular fibrillation.
*Chaos*8:57–64, 1998.CrossRefPubMedGoogle Scholar - 23.Pernot, M., K. Fujikura, S. D. Fung-Kee-Fung, and E. E. Konofagou. ECG-gated, mechanical and electromechanical wave imaging of cardiovascular tissues in vivo.
*Ultrasound Med. Biol.*33(7):1075–1085, 2007.CrossRefPubMedGoogle Scholar - 24.Pogwizd, S. M., and P. B. Corr. Reentrant and nonreentrant mechanisms contribute to arrhythmogenesis during early myocardial ischemia: results using three-dimensional mapping.
*Circ. Res.*61:352–371, 1987.PubMedGoogle Scholar - 25.Ramanathan, C., R. N. Ghanem, P. Jia, K. Ryu, and Y. Rudy. Noninvasive electrocardiographic imaging for cardiac electrophysiology and arrhythmia.
*Nat. Med.*10(4):422–428, 2004.CrossRefPubMedGoogle Scholar - 26.Samie, F. H., O. Berenfeld, J. Anumonwo, S. F. Mironov, S. Udassi, J. Beaumont, S. Taffet, A. M. Pertsov, and J. Jalife. Rectification of the background potassium current: a determinant of rotor dynamics in ventricular fibrillation.
*Circ. Res.*89(12):1216–1223, 2001.CrossRefPubMedGoogle Scholar - 27.Sampson, K. J., and C. Henriquez. Simulation and prediction of functional block in the presence of structural and ionic heterogeneity.
*Am. J. Physiol.*281:H2597–H2603, 2001.Google Scholar - 28.Wang, S., W. N. Lee, J. Provost, J. Luo, and E. E. Konofagou. A composite high-frame-rate system for clinical cardiovascular imaging.
*IEEE Trans. Ultrason. Ferroelectr. Freq. Control*55:2221–2233, 2008.CrossRefPubMedGoogle Scholar - 29.Wang, Y., and Y. Rudy. Applications of the method of fundamental solutions to potential-based inverse electrocardiography.
*Ann. Biomed. Eng.*34(8):1272–1288, 2006.CrossRefPubMedGoogle Scholar - 30.Weiss, J. N., P. S. Chen,
*et al*. Ventricular fibrillation—How do we stop the waves from breaking?*Circ. Res.*87(12):1103–1107, 2000.PubMedGoogle Scholar - 31.Wellner, M., O. Bernus, S. F. Mironov, and A. M. Pertsov. Multiplicative optical tomography of cardiac electrical activity.
*Phys. Med. Biol.*51:4429–4446, 2006.CrossRefPubMedGoogle Scholar - 32.Whiteley, J. P., M. J. Bishop, and D. J. Gavaghan. Soft tissue modeling of cardiac fibres for use in coupled mechano-electric simulations.
*Bull. Math. Biol.*69:2199–2225, 2007.CrossRefPubMedGoogle Scholar - 33.Yan, G. X., W. Shimizu, and C. Antzelevitch. Characteristics and distribution of M cells in arterially perfused canine left ventricular wedge preparations.
*Circulation*98:1921–1927, 1998.PubMedGoogle Scholar