Computer-Based Detection and Analysis of Heart Sound and Murmur
To develop a digital algorithm that detects first and second heart sounds, defines the systole and diastole, and characterises the systolic murmur. Heart sounds were recorded in 300 children with a cardiac murmur, using an electronic stethoscope. A Digital algorithm was developed for detection of first and second heart sounds. R-waves and T-waves in the electrocardiography were used as references for detection. The sound signal analysis was carried out using the short-time Fourier transform. The first heart sound detection rate, with reference to the R-wave, was 100% within 0.05–0.2R-R interval. The second heart sound detection rate between the end of the T-wave and the 0.6R-R interval was 97%. The systolic and diastolic phases of the cardiac cycle could be identified. Because of the overlap between heart sounds and murmur a systolic segment between the first and second heart sounds (20–70%) was selected for murmur analysis. The maximum intensity of the systolic murmur, its average frequency, and the mean spectral power were quantified. The frequency at the point with the highest sound intensity in the spectrum and its time from the first heart sound, the highest frequency, and frequency range were also determined. This method will serve as the foundation for computer-based detection of heart sounds and the characterisation of cardiac murmurs.
KeywordsDetection algorithm Signal analysis Short-time Fourier transform Systolic murmur
Abella, M., J. Formolo, and D. G. Penney. Comparison of the acoustic properties of six popular stethoscopes. J. Acoust. Soc. Am.
91(4 Pt 1):2224-2228, 1992.PubMedGoogle Scholar
Balster, D. A., et al.
Digital acoustic analysis of precordial innocent versus ventricular septal defect murmurs in children. Am. J. Cardiol.
79(11):1552–1555, 1997.CrossRefPubMedGoogle Scholar
Baranek, H. L., et al.
Automatic detection of sounds and murmurs in patients with Ionescu–Shiley aortic bioprostheses. Med. Biol. Eng. Comput.
27(5):449–455, 1989.PubMedGoogle Scholar
Bulgrin, J. R., et al.
Comparison of short-time Fourier, wavelet and time-domain analyses of intracardiac sounds. Biomed. Sci. Instrum.
29:465–472, 1993.PubMedGoogle Scholar
Donnerstein, R. L. Continuous spectral analysis of heart murmurs for evaluating stenotic cardiac lesions. Am. J. Cardiol.
64(10):625–630, 1989.CrossRefPubMedGoogle Scholar
Iwata, A., et al.
Algorithm for detecting the first and the second heart sounds by spectral tracking. Med. Biol. Eng. Comput.
18(1):19–26, 1980.PubMedGoogle Scholar
Kindig, J. R., et al.
Acoustical performance of the stethoscope: A comparative analysis. Am. Heart J.
104(2 Pt 1):269–275, 1982.PubMedGoogle Scholar
Lukkarinen, S., et al.
A new phonocardiographic recording system. Computer Cardiol.
24:117–120, 1997.Google Scholar
Maurice, B., E. E. Rappaport, B. Haward, and M. D. Sprague. The acoustic stethoscope and the electrical amplifying stethoscope and stethograph. Am. Heart J.
21(3):257–318, 1940.Google Scholar
Nygards, M., and L. Sornmo. Delineation of the QRS complex using the envelope of the E.C.G. Med. Biol. Eng. Comput.
Sep;21(5):538–547, 1983.PubMedGoogle Scholar
Nygaard, H., et al.
Assessing the severity of aortic valve stenosis by spectral analysis of cardiac murmurs (spectral vibrocardiography). Part I: Technical aspects. J. Heart Valve Dis.
2(4):454–467, 1993.PubMedGoogle Scholar
Nygaard, H., et al.
Assessing the severity of aortic valve stenosis by spectral analysis of cardiac murmurs (spectral vibrocardiography). Part II: Clinical aspects. J. Heart Valve Dis.
2(4):468–475, 1993.PubMedGoogle Scholar
O’Toole, J. D., et al.
The mechanism of splitting of the second heart sound in atrial septal defect. Circulation
56(6):1047–1053, 1977.PubMedGoogle Scholar
Rangayyan, R. M., and R. J. Lehner. Phonocardiogram signal analysis: A review. Crit. Rev. Biomed. Eng.
15(3):211–236, 1987.PubMedGoogle Scholar
Selig, M. B. Stethoscopic and phonoaudio devices: Historical and future perspectives. Am. Heart J.
126(1):262–268, 1993.PubMedGoogle Scholar
Shaver, J. A., and J. D. O’Toole. The second heart sound: Newer concepts. Part I. Normal and wide physiological splitting. Mod. Concepts Cardiovasc. Dis.
46(2):7–12, 1977.Google Scholar
Smythe, J. F., et al.
Initial evaluation of heart murmurs: Are laboratory tests necessary? Pediatrics
86(4):497–500, 1990.PubMedGoogle Scholar
© Biomedical Engineering Society 2005