Integratable non-clogging microconcentrator based on counter-flow principle for continuous enrichment of CaSki cells sample
- 418 Downloads
This study addresses the need to reduce the risk of clogging when preparing samples for cell concentration, i.e., the CaSki Cell-lines (epidermoid cervical carcinoma cells). Aiming to develop a non-clogging microconcentrator, we proposed a new counter-flow concentration unit characterized by the directions of penetrating flows being at an obtuse angle to the main flow, due to employment of streamlined turbine blade-like micropillars. Based on the optimization results of the counter-flow unit profile, a fractal arrangement for the counter-flow concentration unit was developed. A counter-flow microconcentrator chip was then designed and fabricated, with both the processing layer and collecting layer arranged in terms of the honeycomb structure. Visualized experiments using CaSki cell samples on the microconcentrator chip demonstrated that no cell-clogging phenomena occurred during the test and that no cells were found in the final filtrate. The test results show an excellent concentration performance for the microconcentrator chip, while a concentrating ratio of >4 with the flow rate being below 1.0 ml/min. As only geometrical structure is employed in the passive device, the counter-flow microconcentrator can be easily integrated into advanced microfluidic systems. Owing to the merit of non-clogging and continuous processing ability, the counter-flow microconcentrator is not only suitable for the sample preparation within biomedical field, but also applicable in water-particle separation.
KeywordsCounter-flow Microconcentrator Fractal arrangement Non-clogging Sample preparation
This study was mainly supported by the subcontract of Norwegian Government NFR Funds (Project No.187981 POCNAD), and by Skjema for søknad om FoU-stipend og kompetanseheving på Høgskolen i Vestfold. We are grateful to our colleague, Richard Nelson, for his assistance in proofreading our drafts. The authors appreciated the supports and discussion from Dr. Anja Gulliksen, Dr. Lars Soli, and Dr Hanne Skomedal in NorChip AS.
- Currie IG (1993) Fundamental mechanics of fluids. McGraw-Hill, Inc., New York, USAGoogle Scholar
- Dong T, Yang Z (2008) Measurement and modeling of R141b condensation heat transfer in silicon rectangular microchannels. J Micromech Microeng 18:085012-1–085012-16Google Scholar
- Dong T, Yang Z, Egeland E et al (2009) Clogging failure in microfilter for blood cell separation and its novel improvements IEEE Proc 16th IPFA, 6–10 July, Suzhou, China, pp 759–763. doi: 10.1109/IPFA.2009.5232727
- Dong T, Su Q et al (2010) A smart fully integrated micromachined separator with soft magnetic micro-pillar arrays for cell isolation. J Micromech Microeng 20:115021-1–115021-9Google Scholar
- Ikuta K, Maruo S, Fujisawa T et al (1999) Micro concentrator with opto-sense micro reactor for biochemical IC chip family. Proc of MEMS 99:376–381Google Scholar
- Ji X, Wang W, Lou X, et al. (2007) A centrifugation-enhanced high-efficiency micro-filter with spiral channel. TRANSDUCERS & EUROSENSORS ‘07, the 14th international conference on solid-state sensors, actuators and microsystems, Lyon, France, 10–14 June 2007Google Scholar
- Tran MN, Dong T, Su Q et al (2010) Design and optimization of non-clogging counter-flow microconcentrator for enriching epidermoid cervical carcinoma cells. Biomed Microdevices. doi: 10.1007/s10544-010-9483-5
- van Midwoud PM, Verpoorte E (2008) Implementing sample preconcentration in microfluidic devices. In: Landers JP (ed) Handbook of capillary and microchip electrophoresis and associated microtechniques. CRC Press, Boca Raton, FLGoogle Scholar