Measurement of sound pressure and temperature in tissue-mimicking material using an optical fiber Bragg grating sensor
- 1.9k Downloads
The experimental investigation of an optical fiber Bragg grating (FBG) sensor for biomedical application is described. The FBG sensor can be used to measure sound pressure and temperature rise simultaneously in biological tissues exposed to ultrasound. The theoretical maximum values that can be measured with the FBG sensor are 73.0 MPa and 30 °C.
In this study, measurement of sound pressure up to 5 MPa was performed at an ultrasound frequency of 2 MHz. A maximum temperature change of 6 °C was measured in a tissue-mimicking material.
Values yielded by the FBG sensor agreed with those measured using a thermocouple and a hydrophone.
Since this sensor is used to monitor the sound pressure and temperature simultaneously, it can also be used for industrial applications, such as ultrasonic cleaning of semiconductors under controlled temperatures.
KeywordsFBG Viscous heating
This study is supported by the MEXT-Supported Program for the Strategic Research Foundation at Private Universities, 2013–2018.
Compliance with ethical standards
Conflict of interest
The authors declare that there is no conflict of interest.
This article does not contain any studies with human or animal subjects performed by any of the authors.
- 3.Bercoff J, Chaffai S, Tanter M, et al. Ultrafast imaging of beam formed shear waves induced by the acoustic radiation force in soft tissue: application to transient elastography. In: Proceedings of the 2002 IEEE Ultrasonics Symposium. 2002;1899–902.Google Scholar
- 6.Sugimoto T, Ueha S, Itoh K. Tissue hardness measurement using the radiation force of focused ultrasound. In: Proceedings of the 1990 IEEE Ultrasonics Symposium. 1990; 1377–80.Google Scholar
- 12.FDA Guidance for Industry and FDA Staff. Information for manufacturers seeking marketing clearance of diagnostic ultrasound systems and transducers. Rockville: US Department of Health and Human Services, Food and Drug Administration, Center for Devices and Radiological Health; 2008.Google Scholar
- 13.AIUM Clinical Standards Committee. How to interpret the ultrasound output display standard for higher acoustic output diagnostic ultrasound devices: version 2. J Ultrasound Med. 2004;23:723–6.Google Scholar
- 14.Ter Haar G. The Safe use of ultrasound in medical diagnosis. London: Br Inst Radiol; 2000.Google Scholar
- 21.Zhou Z, Ou J. Techniques of temperature compensation for FBG strain sensors used in long-term structural monitoring. In: Proceeding of Asian pacific fundamental problems of opto- and microelectronics, Russia. 2004. p. 465–71.Google Scholar
- 23.Martinez R, Leija L, Vera A. Ultrasonic attenuation in pure water: comparison between through-transmission and pulse-echo techniques. In: Pan American health care exchanges, PAHCE 2010; 2010. p. 81–4.Google Scholar