Determination of Commonly Used Excipients in Pharmaceutical Preparations by Microchip Electrophoresis with Conductivity Detection
- 112 Downloads
In this study, a microchip electrophoresis method for simultaneous determination of three frequently used pharmaceutical excipients (methylparaben, propylparaben and erythrosine) was described. Anionic zone electrophoresis (ZE) separations were performed on the microchip with contact conductivity detection using a high pH buffer (9.8). Sufficient resolution of the analytes (≥ 2.6) was achieved by 1 mmol L−1 β-cyclodextrin added to the buffer. Adsorption of the analytes on the walls of a poly(methylmethacrylate) microchip was suppressed by adding 1 mmol L− 1 sulfate to the sample. The LOD values for the excipients ranged from 0.5 to 11.1 µmol L− 1. High repeatability of the migration times of the analytes in model (< 2.9% RSD) and real pharmaceutical (< 4.1% RSD) samples was reached on the microchip. Peak areas of the analytes in all analyzed samples were characterized by RSD values ranging from 0.4 to 5.7%. The ZE method was successfully applied to the analysis of pharmaceutical syrups after a simple pretreatment (centrifugation and dilution), and gave recoveries of the analytes in the range 86.1–96.6%. The results showed a good agreement with those obtained by CZE with UV detection (254 nm) at 95% confidence level.
KeywordsConductivity detection Erythrosine Microchip electrophoresis Parabens Pharmaceutical syrups
This study was funded by the Slovak Research and Development Agency (APVV-0259-12 and APVV-17-0318), Scientific Grant Agency VEGA (1/0787/18), Research and Development Operational Programme funded by the ERDF, project Center of Excellence in Security Research, ITMS: 26240120034, and Central European Exchange Program for University Studies (CIII-RO-0010-12-1718).
Compliance with Ethical Standards
Conflict of interest
All authors declare that they have no conflict of interest.
This article does not contain any studies with human participants or animals performed by any of the authors.
- 1.Rowe RC, Sheskey PJ, Quinn ME (2009) Handbook of pharmaceutical excipients. Pharmaceutical Press and American Pharmacists Association, WashingtonGoogle Scholar
- 2.Farajzadeh MA, Djozan D, Bakhtiyari RF (2010) Use of a capillary tube for collecting an extraction solvent lighter than water after dispersive liquid-liquid microextraction and its application in the determination of parabens in different samples by gas chromatography-flame ionization detection. Talanta 81(4–5):1360–1367. https://doi.org/10.1016/j.talanta.2010.02.035 CrossRefGoogle Scholar
- 4.Núñez L, Turiel E, Martin-Esteban A, Tadeo JL (2010) Molecularly imprinted polymer for the extraction of parabens from environmental solid samples prior to their determination by high performance liquid chromatography-ultraviolet detection. Talanta 80(5):1782–1788. https://doi.org/10.1016/j.talanta.2009.10.023 CrossRefGoogle Scholar
- 9.Blanco E, Casais MdC, Mejuto MdC, Cela R (2009) Combination of off-line solid-phase extraction and on-column sample stacking for sensitive determination of parabens and p-hydroxybenzoic acid in waters by non-aqueous capillary electrophoresis. Anal Chim Acta 647(1):104–111. https://doi.org/10.1016/j.aca.2009.05.024 CrossRefGoogle Scholar
- 10.He S, Zhao Y, Zhu Z, Liu H, Li M, Shao Y, Zhuang Q (2006) Comparative study for the analysis of parabens by micellar electrokinetic capillary chromatography with and without large-volume sample stacking technique. Talanta 69(1):166–171. https://doi.org/10.1016/j.talanta.2005.09.018 CrossRefGoogle Scholar
- 11.Han F, He YZ, Yu CZ (2008) On-line pretreatment and determination of parabens in cosmetic products by combination of flow injection analysis, solid-phase extraction and micellar electrokinetic chromatography. Talanta 74(5):1371–1377. https://doi.org/10.1016/j.talanta.2007.09.007 CrossRefGoogle Scholar
- 12.De Rossi A, Desiderio C (2002) Fast capillary electrochromatographic analysis of parabens and 4-hydroxybenzoic acid in drugs and cosmetics. Electrophoresis 23 (19):3410–3417. https://doi.org/10.1002/1522-2683(200210)23:19%3C3410::AID-ELPS3410%3E3.0.CO;2-I CrossRefGoogle Scholar
- 14.El-Gindy A, Attia KAS, Nassar MW, Abu Seada HH, Shoeib MAS (2013) HPLC method for determination of paracetamol, psedoephedrine, triprolidine, methylparaben, propylparaben, sodium benzoate, and their related substances in pharmaceutical syrup. J Liq Chromatogr Relat Technol 36(9):1251–1263. https://doi.org/10.1080/10826076.2012.685922 Google Scholar
- 15.Louhaichi MR, Jebali S, Kallel M, Adhoum N, Benhamida N, Monser L (2014) Development and validation of a stability-indicating hplc method for the determination of pizotifen, methylparaben, and propylparaben in syrup. J Liq Chromatogr Relat Technol 37(7):988–1003. https://doi.org/10.1080/10826076.2013.765454 CrossRefGoogle Scholar
- 18.Minioti KS, Sakellariou CF, Thomaidis NS (2007) Determination of 13 synthetic food colorants in water-soluble foods by reversed-phase high-performance liquid chromatography coupled with diode-array detector. Anal Chim Acta 583(1):103–110. https://doi.org/10.1016/j.aca.2006.10.002 CrossRefGoogle Scholar
- 19.Ryvolová M, Táborský P, Vrábel P, Krásenský P, Preisler J (2007) Sensitive determination of erythrosine and other red food colorants using capillary electrophoresis with laser-induced fluorescence detection. J Chromatogr A 1141(2):206–211. https://doi.org/10.1016/j.chroma.2006.12.018 CrossRefGoogle Scholar
- 21.Masár M, Kaniansky D (1996) Determination of synthetic dyes in food products by capillary zone electrophoresis in a hydrodynamically closed separation compartment. J Capill Electrophor 3(3):165–171Google Scholar
- 24.Rastogi SD, Dixit S, Tripathi A, Das M (2015) Simultaneous determination of acetaminophen and synthetic color(s) by derivative spectroscopy in syrup formulations and validation by HPLC: exposure risk of colors to children. AAPS PharmSciTech 16(3):505–517. https://doi.org/10.1208/s12249-014-0228-2 CrossRefGoogle Scholar
- 25.Manz A, Harrison DJ, Verpoorte EMJ, Fettinger JC, Paulus A, Lüdi H, Widmer HM (1992) Planar chips technology for miniaturization and integration of separation techniques into monitoring systems. Capillary electrophoresis on a chip. J Chromatogr A 593(1–2):253–258. https://doi.org/10.1016/0021-9673(92)80293-4 CrossRefGoogle Scholar
- 28.Bodor R, Žúborová M, Ölvecká E, Madajová V, Masár M, Kaniansky D, Stanislawski B (2001) Isotachophoresis and isotachophoresis-zone electrophoresis of food additives on a chip with column-coupling separation channels. J Sep Sci 24 (9):802–809. https://doi.org/10.1002/1615-9314(20010901)24:9%3C802::AID-JSSC802%3E3.0.CO;2-1 CrossRefGoogle Scholar
- 33.Batistela VR, Pellosi DS, De Souza FD, Da Costa WF, De Oliveira Santin SM, De Souza VR, Caetano W, De Oliveira HPM, Scarminio IS, Hioka N (2011) PKa determinations of xanthene derivates in aqueous solutions by multivariate analysis applied to UV-Vis spectrophotometric data. Spectrochim Acta Part A Mol Biomol Spectrosc 79(5):889–897. https://doi.org/10.1016/j.saa.2011.03.027 CrossRefGoogle Scholar
- 34.Troška P, Masár M, Pobozy E, Bodor R (2013) Separation of some pharmaceutical additives by microchip and capillary electrophoresis. Chem Listy 107(SI3):S460–S461Google Scholar
- 37.Validation of analytical procedures: Text and methodology Q2(R1) (2005) International conference on harmonization of technical requirements for the registration of pharmaceuticals for human use, Geneva. http://www.gmp-compliance.org/guidemgr/files/Q2(R1).PDF. Accessed 28 Aug 2018
- 38.Scientific Committee on Consumer Safety, Opinion on CI 45430 (Erythrosine) (2010) European commission, directorate-general for health & consumers. http://ec.europa.eu/health/scientific_committees/consumer_safety/docs/sccs_o_030.pdf. Accessed 28 Aug 2018
- 39.Different Preservatives And Concentration Dosage For Pharmaceutical Liquid Preparation (2018) https://www.lfatabletpresses.com/articles/preservatives-concentration-dosage-pharmaceutical-liquid-preparation. Accessed 28 Aug 2018