Numerical and experimental evaluation of RF shimming in the human brain at 9.4 T using a dual-row transmit array
- 523 Downloads
To provide a numerical and experimental investigation of the static RF shimming capabilities in the human brain at 9.4 T using a dual-row transmit array.
Materials and methods
A detailed numerical model of an existing 16-channel, inductively decoupled dual-row array was constructed using time-domain software together with circuit co-simulation. Experiments were conducted on a 9.4 T scanner. Investigation of RF shimming focused on B1 + homogeneity, efficiency and local specific absorption rate (SAR) when applied to large brain volumes and on a slice-by-slice basis.
Numerical results were consistent with experiments regarding component values, S-parameters and B1 + pattern, though the B1 + field was about 25 % weaker in measurements than simulations. Global shim settings were able to prevent B1 + field voids across the entire brain but the capability to simultaneously reduce inhomogeneities was limited. On a slice-by-slice basis, B1 + standard deviations of below 10 % without field dropouts could be achieved in axial, sagittal and coronal orientations across the brain, even with phase-only shimming, but decreased B1 + efficiency and SAR limitations must be considered.
Dual-row transmit arrays facilitate flexible 3D RF management across the entire brain at 9.4 T in order to trade off B1 + homogeneity against power-efficiency and local SAR.
KeywordsRF shimming 9.4 T Transmit array Local SAR RF simulations
- 4.Robitaille PM, Berliner LJ (2006) Ultra high field magnetic resonance imaging, vol 26. Biological Magnetic Resonance, Springer, New YorkGoogle Scholar
- 12.Setsompop K, Alagappan V, Gagoski B, Witzel T, Polimeni J, Potthast A, Hebrank F, Fontius U, Schmitt F, Wald LL, Adalsteinsson E (2008) Slice-selective RF pulses for in vivo B1+ inhomogeneity mitigation at 7 T using parallel RF excitation with a 16-element coil. Magn Reson Med 60(6):1422–1432PubMedPubMedCentralCrossRefGoogle Scholar
- 17.Adriany G, Gozubuyuk A, Auerbach EJ, van de Moortele PF, Andersen P, Vaughan JT, Ugurbil K (2007) A 32 channel Transmit/Receive transmission line head array for 3D RF Shimming. In: proceedings of the 15th scientific meeting, International Society for Magnetic Resonance in Medicine, Berlin, p 166Google Scholar
- 20.Felder J, Hong SM, Celik A, Park J, Geschewski F, Jeong HB, Besançon E, Woo MK, Brenner D, Cho ZH, Shah NJ (2012) RF Coil array for accelerated excitation in three dimensions. In: Proceedings of the 20th scientific meeting, International Society for Magnetic Resonance in Medicine, Melbourne, p 2609Google Scholar
- 22.Bitz AK, Kraff O, Orzada S, Maderwald S, Brote I, Johst S, Ladd ME (2011) Assessment of RF Safety of transmit coils at 7 T by experimental and numerical procedures. In: Proceedings of the 19th scientific meeting, International Society for Magnetic Resonance in Medicine, Montreal, p 490Google Scholar
- 23.Gabriel C (1996) Compilation of the dielectric properties of body tissues at RF and microwave frequencies. Brooks air force technical report AL/OE-TR-1996-0037Google Scholar
- 24.Christ A, Kainz W, Hahn EG, Honegger K, Zefferer M, Neufeld E, Rascher W, Janka R, Bautz W, Chen J, Kiefer B, Schmitt P, Hollenbach HP, Shen J, Oberle M, Szczerba D, Kam A, Guag JW, Kuster N (2010) The virtual family–development of surface-based anatomical models of two adults and two children for dosimetric simulations. Phys Med Biol 55(2):N23–N38PubMedCrossRefGoogle Scholar
- 26.Chishti MH, Wang Z, Yeo DTB (2013) Local SAR prediction errors with variation of electrical properties in the head at 7T. In: Proceedings of the 21th scientific meeting, International Society for Magnetic Resonance in Medicine, Salt Lake City, p 2835Google Scholar
- 27.Kozlov M, Shajan G, Turner R (2013) Effects of tuning condition, head size and position on the SAR of a 9.4T dual row array. In: Proceedings of the 21th scientific meeting, International Society for Magnetic Resonance in Medicine, Salt Lake City, p 2830Google Scholar
- 29.Van de Moortele PF, Ugurbil K (2009) Very fast multi channel B1 calibration at high field in the small flip angle regime. In: Proceedings of the 17th scientific meeting, International Society for Magnetic Resonance in Medicine, Honolulu, p 367Google Scholar
- 33.Kozlov M, Turner R (2011) Analysis of RF transmit performance for a multi-row multi-channel MRI loop array at 300 and 400 MHz. In: Proceedings of the Asia-Pacific Microwave Conference, p 1193Google Scholar
- 36.Hoffmann J, Budde J, Shajan G, Pohmann R (2010) Slice-selective B1 phase shimming at 9.4 T. In: Proceedings of the 18th scientific meeting, International Society for Magnetic Resonance in Medicine, Stockholm, p 1470Google Scholar